Estimating needle tip deflection in biological tissue from a single transverse ultrasound image: application to brachytherapy
- 272 Downloads
This paper proposes a method to predict the deflection of a flexible needle inserted into soft tissue based on the observation of deflection at a single point along the needle shaft.
We model the needle-tissue as a discretized structure composed of several virtual, weightless, rigid links connected by virtual helical springs whose stiffness coefficient is found using a pattern search algorithm that only requires the force applied at the needle tip during insertion and the needle deflection measured at an arbitrary insertion depth. Needle tip deflections can then be predicted for different insertion depths.
Verification of the proposed method in synthetic and biological tissue shows a deflection estimation error of \(<\)2 mm for images acquired at 35 % or more of the maximum insertion depth, and decreases to 1 mm for images acquired closer to the final insertion depth. We also demonstrate the utility of the model for prostate brachytherapy, where in vivo needle deflection measurements obtained during early stages of insertion are used to predict the needle deflection further along the insertion process.
The method can predict needle deflection based on the observation of deflection at a single point. The ultrasound probe can be maintained at the same position during insertion of the needle, which avoids complications of tissue deformation caused by the motion of the ultrasound probe.
KeywordsNeedle insertion Brachytherapy Ultrasound Guidance Needle-tissue Interaction
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study. Approval for the study granted by the Alberta Cancer Research Ethics Committee under file number 25837.
- 2.Abolhassani N, Patel R, Ayazi F (2007) Needle control along desired tracks in robotic prostate brachytherapy. In: IEEE international conference on systems, man and cybernetics, 2007, ISIC, pp 3361–3366. IEEEGoogle Scholar
- 3.Beer FP, Sanghi S (2012) Mechanics of materials, 6th edn. McGraw-Hill, New York (global ed./adapted by Sanjeev Sanghi edn., includes index)Google Scholar
- 4.Bogdanich W (2009) At VA hospital, a rogue cancer unit. The New York Times, New YorkGoogle Scholar
- 7.Dehghan E, Goksel O, Salcudean SE (2006) A comparison of needle bending models. In: Medical image computing and computer-assisted intervention–MICCAI 2006, pp 305–312. SpringerGoogle Scholar
- 11.Gonzales RC, Woods RE (2002) Digital image processing, 2nd edn. Prentice Hall, Upper Saddle RiverGoogle Scholar
- 12.Haddadi A, Hashtrudi-Zaad K (2011) Development of a dynamic model for bevel-tip flexible needle insertion into soft tissues. In: 2011 annual international conference of the IEEE, engineering in medicine and biology society, EMBC, pp 7478–7482. IEEEGoogle Scholar
- 14.Khadem M, Fallahi B, Rossa C, Sloboda R, Usmani N, Tavakoli M (2015) A mechanics-based model for simulation and control of flexible needle insertion in soft tissue. In: 2015 IEEE international conference on, robotics and automation (ICRA), pp 2264–2269Google Scholar
- 16.Lachaine M, Falco T (2013) Intrafractional prostate motion management with the clarity autoscan system. Med Phys Int 1(1):72–80Google Scholar
- 17.Lehmann T, Rossa C, Usmani N, Sloboda R, Tavakoli M (2015) A virtual sensor for needle deflection estimation during soft-tissue needle insertion. In: 2015 IEEE international conference on, robotics and automation (ICRA), pp 1217–1222Google Scholar
- 18.Mathiassen K, Dall’Alba D, Muradore R, Fiorini P, Elle OJ (2013) Real-time biopsy needle tip estimation in 2d ultrasound images. In: 2013 IEEE international conference on, robotics and automation (ICRA), pp 4363–4369. IEEEGoogle Scholar
- 20.Moreira P, Misra S (2014) Biomechanics-based curvature estimation for ultrasound-guided flexible needle steering in biological tissues. Ann Biomed Eng. doi: 10.1007/s10439-014-1203-5
- 22.Neubach Z, Shoham M (2010) Ultrasound-guided robot for flexible needle steering. Biomed Eng IEEE Trans 57(4):799–805Google Scholar
- 26.Spong MW, Vidyasagar M (2008) Robot dynamics and control. Wiley, HobokenGoogle Scholar
- 27.Vrooijink GJ, Abayazid M, Patil S, Alterovitz R, Misra S (2014) Needle path planning and steering in a three-dimensional non-static environment using two-dimensional ultrasound images. Int J Robot Res 33(10): 0278364914526627Google Scholar
- 28.Waine M, Rossa C, Sloboda R, Usmani N, Tavakoli M (2015) 3D shape visualization of curved needles in tissue from 2D ultrasound images using ransac. In: 2015 IEEE international conference on, robotics and automation, pp 4723–4728. IEEEGoogle Scholar