Evaluation of semi-automatic arterial stenosis quantification
- 47 Downloads
Object: To assess the accuracy and reproducibility of semi-automatic vessel axis extraction and stenosis quantification in 3D contrast-enhanced Magnetic Resonance Angiography (CE-MRA) of the carotid arteries (CA).
Materials and methods: A total of 25 MRA datasets was used: 5 phantoms with known stenoses, and 20 patients (40 CAs) drawn from a multicenter trial database. Maracas software extracted vessel centerlines and quantified the stenoses, based on boundary detection in planes perpendicular to the centerline. Centerline accuracy was visually scored. Semi-automatic measurements were compared with: (1) theoretical phantom morphometric values, and (2) stenosis degrees evaluated by two independent radiologists.
Results: Exploitable centerlines were obtained in 97% of CA and in all phantoms. In phantoms, the software achieved a better agreement with theoretic stenosis degrees (weighted kappa κ w = 0.91) than the radiologists (κ w = 0.69). In patients, agreement between software and radiologists varied from κ w =0.67 to 0.90. In both, Maracas was substantially more reproducible than the readers. Mean operating time was within 1 min/ CA.
Conclusion: Maracas software generates accurate 3D centerlines of vascular segments with minimum user intervention. Semi-automatic quantification of CA stenosis is also accurate, except in very severe stenoses that cannot be segmented. It substantially reduces the inter-observer variability.
KeywordsVascular diseases Carotid artery stenosis Magnetic resonance angiography Three-dimensional image Computer assisted image processing
Unable to display preview. Download preview PDF.
- 1.Lee VS, Doug JM, Krinsky GA, Rofsky NM (2000) Gadolinium-enhanced MR angiography: artifacts and pitfalls. Am J Roentgenol 175:197–205Google Scholar
- 3.Vanninen RL, Manninen HI, Partanen PK, Tulla H, Vainio PA (1996) How should we estimate carotid stenosis using magnetic resonance angiography? Neuroradiol 38:299–305Google Scholar
- 4.Serfaty JM, Chirossel P, Chevallier JM, Ecochard R, Froment JC, Douek PC (2000) Accuracy of three-dimensional gadolinium-enhanced MR Angiography in the assessment of extracranial carotid artery disease. Am J Roentgenol 175: 455–463Google Scholar
- 7.Wong KS, Lam WW, Liang E, Huang YN, Chan YL, Kay R (1996) Variability of magnetic resonance angiography and computed tomography angiography in grading middle cerebral artery stenosis. Stroke 6:1084–1087Google Scholar
- 10.Elgersma OE, Wust AF, Buijs PC, van Der Graaf Y, Eikelboom BC, Mali WP (2000) Multidirectional depiction of internal carotid arterial stenosis: three-dimensional time-of-flight MR angiography versus rotational and conventional digital subtraction angiography. Radiol 216:511–516Google Scholar
- 12.Hernández Hoyos M, Orkisz M, Douek PC, Magnin IE (2005) Assessment of carotid artery stenoses in 3D contrast-enhanced magnetic resonance angiography, based on improved generation of the centerline. Mach Graph Vis 14:349–378Google Scholar
- 14.Mukundan R, Ramakrishnan KR (1998) Moment functions in image analysis, theory and applications. World Scientific Publishing Co. Pte. Ltd., Singapore 150 pGoogle Scholar
- 16.Hoogeveen R, Bakker C, Mali W, Viergever M (1997) diameter measurements in TOF and PC angiography: a need for standardization? In: 5th annual meeting International Society of Magnetic Resonance Medicine, Vancouver, p 1847Google Scholar
- 19.Renaudin CP, Barbier B, Roriz R, Revel D, Amiel M (1994) Coronary arteries: new design for three-dimensional arterial phantom. Radiol 190:579–582Google Scholar
- 21.Douek P, Revel D, Chazel S, Falise B, Villard J, Amiel M (1995) Fast MR angiography of the aortoiliac arteries and arteries of the lower extremity: value of bolus-enhanced, whole-volume subtraction technique. Am J Roentgenol 165:431–437Google Scholar
- 22.Bland J, Altman D (1986) Statistical method for assessing agreement between two methods of clinical measurement. Lancet I:307–310Google Scholar