The potential of inertial measurement units (IMUs) for ergonomics applications appears promising. However, previous IMUs validation studies have been incomplete regarding aspects of joints analysed, complexity of movements and duration of trials. The objective was to determine the technological error and biomechanical model differences between IMUs and an optoelectronic system and evaluate the effect of task complexity and duration. Whole-body kinematics from 12 participants was recorded simultaneously with a full-body Xsens system where an Optotrak cluster was fixed on every IMU. Short functional movements and long manual material handling tasks were performed and joint angles were compared between the two systems. The differences attributed to the biomechanical model showed significantly greater (P ≤ .001) RMSE than the technological error. RMSE was systematically higher (P ≤ .001) for the long complex task with a mean on all joints of 2.8° compared to 1.2° during short functional movements. Definition of local coordinate systems based on anatomical landmarks or single posture was the most influent difference between the two systems. Additionally, IMUs accuracy was affected by the complexity and duration of the tasks. Nevertheless, technological error remained under 5° RMSE during handling tasks, which shows potential to track workers during their daily labour.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
An KN, Chao EYS (1991) Kinematic analysis. In: An KN, Berger RA, Cooney WP III (eds) Biomechanics of the wrist joint. Springer, New York, pp 21–36
Bland JM, Altman DG (1986) Statistical methods for assessing agreement between two methods of clinical measurement. Lancet 1:307–310
Bouvier B, Duprey S, Claudon L, Dumas R, Savescu A (2015) Upper limb kinematics using inertial and magnetic sensors: comparison of sensor-to-segment calibrations. Sensors 15:18813–18833. doi:10.3390/s150818813
Brodie MA, Walmsley A, Page W (2008) Dynamic accuracy of inertial measurement units during simple pendulum motion. Comp Methods Biomech Biomed Eng 11:235–242
Cuesta-Vargas AI, Galan-Mercant A, Williams JM (2010) The use of inertial sensors system for human motion analysis. Phys Ther Rev 15:462–473. doi:10.1179/1743288X11Y.0000000006
Cutti AG, Ferrari A, Garofalo P, Raggi M, Cappello A, Ferrari A (2010) ‘Outwalk’: a protocol for clinical gait analysis based on inertial and magnetic sensors. Med Biol Eng Comput 48:17–25. doi:10.1007/s11517-009-0545-x
Cutti AG, Giovanardi A, Rocchi L, Davalli A, Sacchetti R (2008) Ambulatory measurement of shoulder and elbow kinematics through inertial and magnetic sensors. Med Biol Eng Comput 46:169–178. doi:10.1007/s11517-007-0296-5
de Vries WHK, Veeger HEJ, Baten CTM, van der Helm FCT (2009) Magnetic distortion in motion labs, implications for validating inertial magnetic sensors. Gait Posture 29:535–541
El-Gohary M, McNames J (2012) Shoulder and elbow joint angle tracking with inertial sensors. IEEE Trans Biomed Eng 59(9):2635–2641
El-Zayat BF, Efe T, Heidrich A, Anetsmann R, Timmesfeld N, Fuchs-Winkelmann S, Schofer MD (2013) Objective assessment, repeatability, and agreement of shoulder ROM with a 3D gyroscope. BMC Musculoskelet Disord 14:72. doi:10.1186/1471-2474-14-72
Fantozzi S, Giovanardi A, Magalhaes FA, Di Michele R, Cortesi M, Gatta G (2015) Assessment of three-dimensional joint kinematics of the upper limb during simulated swimming using wearable inertial-magnetic measurement units. J Sports Sci:1–8. doi:10.1080/02640414.2015.1088659
Ferrari A, Cutti AG, Garofalo P, Raggi M, Heijboer M, Cappello A, Davalli A (2010) First in vivo assessment of “Outwalk”: a novel protocol for clinical gait analysis based on inertial and magnetic sensors. Med Biol Eng Comput 48:1–15. doi:10.1007/s11517-009-0544-y
Gamage SS, Lasenby J (2002) New least squares solutions for estimating the average centre of rotation and the axis of rotation. J Biomech 35:87–93
Godwin A, Agnew M, Stevenson J (2009) Accuracy of inertial motion sensors in static, quasistatic, and complex dynamic motion. J Biomech Eng 131:114501. doi:10.1115/1.4000109
Harrington ME, Zavatsky AB, Lawson SE, Yuan Z, Theologis TN (2007) Prediction of the hip joint centre in adults, children, and patients with cerebral palsy based on magnetic resonance imaging. J Biomech 40:595–602. doi:10.1016/j.jbiomech.2006.02.003
Kainz H, Carty CP, Modenese L, Boyd RN, Lloyd DG (2015) Estimation of the hip joint centre in human motion analysis: a systematic review. Clin Biomech 30:319–329. doi:10.1016/j.clinbiomech.2015.02.005
Kim S, Nussbaum MA (2013) Performance evaluation of a wearable inertial motion capture system for capturing physical exposures during manual material handling tasks. Ergonomics 56:314–326. doi:10.1080/00140139.2012.742932
Leardini A, Chiari L, Della Croce U, Cappozzo A (2005) Human movement analysis using stereophotogrammetry. Part 3. Soft tissue artifact assessment and compensation. Gait Posture 21:212–225. doi:10.1016/j.gaitpost.2004.05.002
Luinge HJ, Veltink PH, Baten CT (2007) Ambulatory measurement of arm orientation. J Biomech 40:78–85. doi:10.1016/j.jbiomech.2005.11.011
Palermo E, Rossi S, Marini F, Patanè F, Cappa P (2014) Experimental evaluation of accuracy and repeatability of a novel body-to-sensor calibration procedure for inertial sensor-based gait analysis. Measurement 52:145–155
Picerno P, Cereatti A, Cappozzo A (2008) Joint kinematics estimate using wearable inertial and magnetic sensing modules. Gait Posture 28:588–595
Plamondon A, Delisle A, Larue C, Brouillette D, McFadden D, Desjardins P, Lariviere C (2007) Evaluation of a hybrid system for three-dimensional measurement of trunk posture in motion. Appl Ergon 38:697–712
Reed M, Manary MA, Schneider LW (1999) Methods for measuring and representing automobile occupant posture. SAE Technical Paper
Roetenberg D, Luinge H, Slycke P (2009) Xsens MVN: full 6DOF human motion tracking using miniature inertial sensors. Xsens Motion Technologies BV, TechRep
Roislien J, Skare O, Opheim A, Rennie L (2012) Evaluating the properties of the coefficient of multiple correlation (CMC) for kinematic gait data. J Biomech 45:2014–2018. doi:10.1016/j.jbiomech.2012.05.014
Schall MC, Jr., Fethke NB, Chen H, Oyama S, Douphrate DI (2015) Accuracy and repeatability of an inertial measurement unit system for field-based occupational studies. Ergonomics:1–23. doi:10.1080/00140139.2015.1079335
Schiefer C, Ellegast RP, Hermanns I, Kraus T, Ochsmann E, Larue C, Plamondon A (2014) Optimization of inertial sensor-based motion capturing for magnetically distorted field applications. J Biomech Eng 136:121008
Vignais N, Miezal M, Bleser G, Mura K, Gorecky D, Marin F (2013) Innovative system for real-time ergonomic feedback in industrial manufacturing. Appl Ergon 44:566–574. doi:10.1016/j.apergo.2012.11.008
Weinberg H (2011) Gyro mechanical performance: The most important parameter. Technical article MS-2158 Analog Devices, Inc:1–5
Wu G, Siegler S, Allard P, Kirtley C, Leardini A, Rosenbaum D, Whittle M, D’Lima DD, Cristofolini L, Whittle H, Schmid O, Stokes I, Standardization, Terminology Committee of the International Society of B (2002) ISB recommendation on definitions of joint coordinate system of various joints for the reporting of human joint motion–part I: ankle, hip, and spine. International society of biomechanics. J Biomech 35:543–548
Wu G, van der Helm FC, Veeger HE, Makhsous M, Van Roy P, Anglin C, Nagels J, Karduna AR, McQuade K, Wang X, Werner FW, Buchholz B (2005) ISB recommendation on definitions of joint coordinate systems of various joints for the reporting of human joint motion–Part II: shoulder, elbow, wrist and hand. J Biomech 38:981–992
Zatsiorsky VM (1998) Kinematics of human motion. Human Kinetics, Champaign
Zhang J-T, Novak AC, Brouwer B, Li Q (2013) Concurrent validation of Xsens MVN measurement of lower limb joint angular kinematics. Physiol Meas 34:N63–N69
Zhou H, Hu H, Tao Y (2006) Inertial measurements of upper limb motion. Med Biol Eng Comput 44:479–487
The authors are grateful to the IRSST for financial support of the study through grant (2012-0040) and postdoctoral scholarship program and wish to recognise the technical assistance from Sophie Bellefeuille.
Conflict of interest
The authors declare that they have no conflict of interest.
About this article
Cite this article
Robert-Lachaine, X., Mecheri, H., Larue, C. et al. Validation of inertial measurement units with an optoelectronic system for whole-body motion analysis. Med Biol Eng Comput 55, 609–619 (2017). https://doi.org/10.1007/s11517-016-1537-2
- Inertial sensor
- Task complexity