Compliant self-adjusting compression implants are a novel approach to increase the durability of megaprosthesis fixation. However, there is no report of current implant designs that documents the bone-prosthetic interface of this implant. A well-fixed compliant, self-adjusting distal femoral replacement was retrieved from a patient undergoing revision unrelated to fixation. The prosthesis-bone interface was preserved, embedded in poly(methyl methacrylate), and sectioned into 2–4-mm slices. Slices were then imaged using backscatter electron microscopy, and ongrowth and ingrowth were quantified using imaging software. The average percentage of bony ongrowth from five successive sections was 52.5%, and the average percentage of ingrowth into the porous titanium surface was 13.5%. We found that bone ongrowth on the cortex between anchor plug and spindle averages more than 50% and up to 70% depending upon the slice examined with backscatter electron microscopy. Bone ingrowth was consistently around 13% on every slice examined. This is a new finding compared with prior spindle designs, likely due to the addition of hydroxyapatite-coated porous metal titanium surface on the spindle. This report is an important step in understanding the mechanism of bony fixation generated by this implant and supports its increased use in oncological and complex reconstructive situations.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Abrams GD, Gajendran VK, Mohler DG, Avedian RS. Surgical technique: methods for removing a Compress compliant prestress implant. Clin Orthop Related Res. 2012;470(4):1204–1212. https://doi.org/10.1007/s11999-011-2128-z
Bini SA, Johnston JO, Martin DL. Compliant prestress fixation in tumor prostheses: interface retrieval data. Orthopedics. 2000;23(7):702–707.
Bruns J, Delling G, Gruber H, Lohmann CH, Habermann CR. Cementless fixation of megaprostheses using a conical fluted stem in the treatment of bone tumours. J Bone Joint Surg Br. 2007;89(8), 1084–1087. https://doi.org/10.1302/0301-620X.89B8.19236
Capanna R, Scoccianti G, Frenos F, Vilardi A, Beltrami G, Campanacci DA. What was the survival of megaprostheses in lower limb reconstructions after tumor resections? Clin Orthop Related Res. 2014;473(3):820–830. https://doi.org/10.1007/s11999-014-3736-1
Ferguson PC, Zdero R, Schemitsch EH, Deheshi BM, Bell RS, Wunder JS. A biomechanical evaluation of press-fit stem constructs for tumor endoprosthetic reconstruction of the distal femur. J Arthroplasty. 2011;26(8):1373–1379. https://doi.org/10.1016/j.arth.2010.12.005
Healey JH, Morris CD, Athanasian EA, Boland PJ. Compress knee arthroplasty has 80% 10-year survivorship and novel forms of bone failure. Clin Orthop Related Res. 2013;471(3):774–783. https://doi.org/10.1007/s11999-012-2635-6
Pala E, Trovarelli G, Calabrò T, Angelini A, Abati CN, Ruggieri P. Survival of modern knee tumor megaprostheses: failures, functional results, and a comparative statistical analysis. Clin Orthop Related Res. 2015;473(3):891–899. https://doi.org/10.1007/s11999-014-3699-2
Palumbo BT, Henderson ER, Groundland JS, et al. Advances in segmental endoprosthetic reconstruction for extremity tumors: a review of contemporary designs and techniques. Cancer Control. 2011;18(3):160–170.
Zimel MN, Farfalli GL, Zindman AM, et al. Revision distal femoral arthroplasty with the Compress prosthesis has a low rate of mechanical failure at 10 years. Clin Orthop Related Res. 2016;474(2):528–536. https://doi.org/10.1007/s11999-015-4552-y
Avedian RS, Goldsby RE, Kramer MJ, O’Donnell RJ. Effect of chemotherapy on initial compressive osseointegration of tumor endoprostheses. Clin Orthop Relat Res. 2007;459:48–53.
Conflict of Interest
Alexander B. Christ, MD, and Elexis Baral, BS, declare that they have no conflicts of interest. Timothy M. Wright, PhD, reports research support from Stryker, stock options from Orthobond, intellectual property royalties from Mathys Ltd. and from Exactech, and intellectual property royalties and research support from Lima Corporate, as well as board membership in the Knee Society, outside the submitted work. John H. Healey, MD, reports being a paid consultant to Daiichi Sanyo and Stryker, outside the submitted work.
All procedures followed were in accordance with the ethical standards of the responsible committee on human experimentation (institutional and national) and with the Helsinki Declaration of 1975, as revised in 2013.
Informed consent was waived from the patient for being included in this study.
Required Author Forms:
Disclosure forms provided by the authors are available with the online version of this article.
Work was performed at the Hospital for Special Surgery, New York, NY, and Memorial Sloan Kettering Cancer Center, New York, NY.
About this article
Cite this article
Christ, A.B., Baral, E., Wright, T.M. et al. Osteointegration in Compliant Self-Adjusting Compression Fixation Shown by Backscatter Electron Microscopy: A Case Report. HSS Jrnl 15, 302–306 (2019). https://doi.org/10.1007/s11420-019-09696-w
- compliant self-adjusting compression fixation
- retrieval analysis
- backscatter electron microscopy