Evaluation of particle size distribution (PSD) of multimodal dispersion of nanoparticles is a difficult task due to inherent limitations of size measurement methods. The present work reports the evaluation of PSD of a dispersion of poly(isobutylcyanoacrylate) nanoparticles decorated with dextran known as multimodal and developed as nanomedecine.
The nine methods used were classified as batch particle i.e. Static Light Scattering (SLS) and Dynamic Light Scattering (DLS), single particle i.e. Electron Microscopy (EM), Atomic Force Microscopy (AFM), Tunable Resistive Pulse Sensing (TRPS) and Nanoparticle Tracking Analysis (NTA) and separative particle i.e. Asymmetrical Flow Field-Flow Fractionation coupled with DLS (AsFlFFF) size measurement methods.
The multimodal dispersion was identified using AFM, TRPS and NTA and results were consistent with those provided with the method based on a separation step prior to on-line size measurements. None of the light scattering batch methods could reveal the complexity of the PSD of the dispersion.
Difference between PSD obtained from all size measurement methods tested suggested that study of the PSD of multimodal dispersion required to analyze samples by at least one of the single size particle measurement method or a method that uses a separation step prior PSD measurement.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
Tax calculation will be finalised during checkout.
Atomic force microscopy
Asymmetrical flow field-flow fractionation
Differential centrifugal sedimentation
Dynamic light scattering
Nanoparticle tracking analysis
Photon cross-correlation spectroscopy
Particle size distribution
Particle tracking analysis
Quasi elastic light scattering
Sedimentation field-flow fractionation
Scanning electron microscopy
Static light scattering
Transmission electron microscopy
Tunable resistive pulse sensing
Jung H, Kittelson DB, Zachariah MR. The influence of a cerium additive on ultrafine diesel particle emissions and kinetics of oxidation. Combust Flame. 2005;142:276–88.
Jøgensen B, Kristensen SB, Kunov-Kruse AJ, Fehrmann R, Christensen CH, Riisager A. Gas-phase oxidation of aqueous ethanol by nanoparticle vanadia/anatase catalysts. Top Catal. 2009;52:253–7.
Wissing SA, Müller RH. Cosmetic applications for solid lipid nanoparticles (SLN). Int J Pharm. 2003;254:65–8.
Olivier J. Drug transport to brain with targeted nanoparticles. NeuroRx. 2005;2:118–9.
Cormode DP, Naha PC, Fayad ZA. Nanoparticle contrast agents for computed tomography: a focus on micelles. Contrast Media Mol Imaging. 2014;9(1):37–52.
Neuwelt EA, Varallyay P, Bago AG, Muldoon LL, Nesbit G, Nixon R. Imaging of iron oxide nanoparticles by MR and light microscopy in patients with malignant brain tumours. Neuropathol Appl Neurobiol. 2004;30:456–71.
Perlman O, Weitz IS, Azhari H. Copper oxide nanoparticles as contrast agents for MRI and ultrasound dual-modality imaging. Phys Med Biol. 2015;60(15):5767–83.
Galper MW, Saung MT, Fuster V, Roessl E, Thran A, Proksa R, et al. Effect of computed tomography scanning parameters on gold nanoparticle and iodine contrast. Invest Radiol. 2012;47(8):475–81.
Liu CJ, Wang CH, Chen ST, Chen HH, Leng WH, Chien CC, et al. Enhancement of cell radiation sensitivity by pegylated gold nanoparticles. Phys Med Biol. 2010;55(4):931–45.
Seaton A, Tran L, Aitken R, Donaldson K. Nanoparticles, human health hazard and regulation. J R Soc Interface. 2009;7:119–29.
Li C. Structure controlling and process scale-up in the fabrication of nanomaterials. Front Chem Eng China. 2010;4:18–25.
Organisation for Economic Co-operation and Development (OCDE), Regulatory frameworks for nanotechnology in foods and medical products: summary results of a survey activity, DSTI/STP/NANO(2012)22/FINAL, 21 March 2013. Available from: http://search.oecd.org/officialdocuments/publicdisplaydocumentpdf/?cote=DSTI/STP/NANO%282012%2922/FINAL&docLanguage=En (consulted on November 2015). Available from.
Draft guidance from FDA, Considering Whether an FDA-Regulated Product Involves the Application of Nanotechnology, 14 June 2011. Available from:http://www.fda.gov/RegulatoryInformation/Guidances/ucm257698.htm (consulted on November 2015).
Reflection paper on the data requirements for intravenous liposomal products developed with reference to an innovator liposomal product, EMA/CHMP/806058/2009/Rev 02, 21 February 2013. Available from: http://www.ema.europa.eu/docs/en_GB/document_library/Scientific_guideline/2013/03/WC500140351.pdf (consulted on November 2015).
Joint MHLW/EMA reflection paper on the development of block copolymer micelle medicinal products, EMA/CHMP/13099/2013, 17 January 2013. Available from:http://www.ema.europa.eu/docs/en_GB/document_library/Scientific_guideline/2013/02/WC500138390.pdf (consulted on November 2015).
Report of the Joint Regulator -Industry Ad Hoc Working Group: Currently Available Methods for Characterization of Nanomaterials, 17 June 2011. Available from: http://ec.europa.eu/consumers/sectors/cosmetics/files/pdf/iccr5_char_nano_en.pdf (consulted on November 2015).
Organization for Economic Co-operation and Development (OCDE), Guidance manual for the testing of manufactured nanomaterials: OECD’s sponsorship programme; First revision ENV/JM/MONO(2009)20/REV, 2 June 2010. Available from: http://search.oecd.org/officialdocuments/displaydocumentpdf/?cote=env/jm/mono%282009%2920/rev&doclanguage=en (consulted on November 2015).
FDA advisory committee for pharmaceutical science and clinical pharmacology meeting Topic 2 Nanotechnology - Update on FDA Activities, 9 August 2012. Available from: http://www.fda.gov/downloads/AdvisoryCommittees/CommitteesMeetingMaterials/Drugs/AdvisoryCommitteeforPharmaceuticalScienceandClinicalPharmacology/UCM314585.pdf (consulted on November 2015).
Gaumet M, Vargas A, Gurny R, Delie F. Nanoparticles for drug delivery: the need for precision in reporting particle size parameters. Eur J Pharm Biopharm. 2008;69:1–9.
Shekunov BY, Chattopadhyay P, Tong HHY, Chow AHL. Particle size analysis in pharmaceutics: principles, methods and applications. Pharm Res. 2006;24(2):203–27.
Varenne F, Botton J, Merlet C, Beck-Broichsitter M, Legrand F-X, Vauthier C. Standardization and validation of a protocol of size measurements by dynamic light scattering for monodispersed stable nanomaterial characterization. Colloid Surf A. 2015;486:124–38.
Braun A, Couteau O, Franks K, Kestens V, Roebben G, Lamberty A, et al. Validation of dynamic light scattering and centrifugal liquid sedimentation methods for nanoparticle characterisation. Adv Powder Technol. 2011;22:766–70.
Woodward RC, Heeris J, St Pierre TG, Saunders M, Gilbert EP, Rutnakornpituk M, et al. A comparison of methods for the measurement of the particle-size distribution of magnetic nanoparticles. J Appl Crystallogr. 2007;40:495–500.
Elizalde O, Leal GP, Leiza JR. Particle size distribution measurements of polymeric dispersions: a comparative study. Part Part Syst Charact. 2000;17:236–43.
Fielding LA, Mykhaylyk OO, Armes SP, Fowler PW, Mittal V, Fitzpatrick S. Correcting for a density distribution: particle size analysis of core-shell nanocomposite particles using disk centrifuge photosedimentometry. Langmuir. 2012;28:2536–44.
Bell NC, Minelli C, Tompkins J, Stevens MM, Shard AG. Emerging techniques for submicrometer particle sizing applied to Stoeber silica. Langmuir. 2012;28:10860–72.
Linsinger T, Roebben G, Gilliland D, Calzolai L, Rossi F, Gibson N, et al, Requirements on measurements for the implementation of the European Commission definition of the term “nanomaterial”. JRC Reference Reports. 2012.
Powers KW, Brown SC, Krishna VB, Wasdo SC, Moudgil BM, Roberts SM. Research strategies for satefy evaluation of nanomaterials. Part VI. Characterization of nanoscale particles for toxicological evaluation. Toxicol Sci. 2006;90(2):296–303.
Sowerby SJ, Broom MF, Petersen GB. Dynamically resizable nanometre-scale apertures for molecular sensing. Sensors Actuators B. 2007;123:325–30.
Willmott GR, Vogel R, Yu SSC, Groenewegen LG, Roberts GS, Kozak D, et al. Use of tunable nanopore blockade rates to investigate colloidal dispersions. J Phys-Condens Mat. 2010;22(45):1–11.
Vogel R, Willmott G, Kozak D, Roberts GS, Anderson W, Groenewegen L, et al. Quantitative sizing of nano/microparticles with a tunable elastomeric pore sensor. Anal Chem. 2011;83:3499–506.
Lespes G, Gigault J. Hyphenated analytical techniques for multidimensional characterisation of submicron particles: a review. Anal Chim Acta. 2011;692:26–41.
Anderson W, Kozak D, Coleman VA, Jämting ÅK, Trau M. A comapartive study of submicron particle sizing platforms: accuracy, precision and resolution analysis of polydisperse particle size distributions. J Colloid Interface Sci. 2013;405:322–30.
Sokolova V, Ludwig A-K, Hornung S, Rotan O, Horn PA, Epple M, et al. Characterisation of exosomes derived from human cells by nanoparticle tracking analysis and scanning electron microscopy. Coilloid Surf B. 2011;87:146–50.
Pace HE, Rogers NJ, Jarolimek C, Coleman VA, Gray EP, Higgins CP, et al. Single particle inductively coupled plasma-mass spectrometry: a performance evaluation and method comparison in the determination of nanoparticle size. Environ Sci Technol. 2012;46:12272–80.
van der Pol E, Coumans FAW, Grootemaat AE, Gardiner C, Sargent IL, Harrison P, et al. Particle size distribution of exosomes and microvesicles determined by transmission electron microscopy, flow cytometry, nanoparticle tracking analysis, and resistive pulse sensing. J Thromb Haemost. 2014;12:1182–92.
Cascio C, Gilliland D, Rossi F, Calzolai L, Contado C. Critical experimental evaluation of Key methods to detect, size and quantify nanoparticulate silver. Anal Chem. 2014;86:12143–51.
Calzolai L, Gilliland D, Garcìa CP, Rossi F. Separation and characterization of gold nanoparticle mixtures by flow-field-flow fractionation. J Chromatogr A. 2011;1218:4234–9.
Ingebrigtsen L, Brandl M. Determination of the size distribution of liposomes by SEC fractionation, and PCS analysis and enzymatic assay of lipid content. AAPS Pharm Sci Tech. 2002;3(2):9–15.
Sitar S, Kejžar A, Pahovnik D, Kogej K, Tušek-Žnidarič M, Lenassi M, et al. Size characterization and quantification of exosomes by asymmetrical-flow field-flow fractionation. Anal Chem. 2015;87:9225–33.
Gun’ko VM, Klyueva AV, Levchuk YN, Leboda R. Photon correlation spectroscopy investigations of proteins. Adv Colloid Interface. 2003;105:201–328.
ISO/TS 10797:2012: Nanotechnologies - Characterization of single-wall carbon nanotubes using transmission electron microscopy.
ISO 13322-1:2004 Particle size analysis - Image analysis methods - Part 1: Static image analysis, methods.
Vauthier C, Persson B, Lindner P, Cabane B. Protein adsorption and complement activation for di-block copolymer nanoparticles. Biomaterials. 2011;32:1646–56.
Rasband W. ImageJ (Computer Program), National Institute of Health, 2013.
Cybernetics M. Image-Pro Plus (Computer Program), Roper Industries, 2013.
Binnig G, Quate CF, Gerber C. Atomic force microscope. Phys Rev Lett. 1986;56:930–3.
Meyer G, Amer NM. Novel optical approach to atomic force microscopy. Appl Phys Lett. 1988;53(12):1045–7.
Couteau O, Roebben G. Measurement of the size of spherical nanoparticles by means of atomic force microscopy. Meas Sci Technol. 2001;22(6):65101–8.
Filipe V, Hawe A, Jiskoot W. Critical evaluation of Nanoparticle Tracking Analysis (NTA) by NanoSight for the measurement of nanoparticles and protein aggregates. Pharm Res. 2010;27:796–810.
ISO 22 412:2008(E): Particle size analysis - dynamic light scattering (DLS).
Cho TJ, Hackley VA. Fractionation and characterization of gold nanoparticles in aqueous solution: asymmetric-flow field flow fractionation with MALS, DLS, and UV–vis detection. Anal Bioanal Chem. 2010;398:2003–18.
Rbii K, Violleau F, Guedj S, Surel O. Analysis of aged gelatin by AFlFFF-MALS: Identification of high molar mass components and their influence on solubility. Food Hydrocoll. 2009;23:1024–30.
ACKNOWLEDGMENTS AND DISCLOSURES
This work was supported by BpI France (Project NICE). The authors acknowledge the Région Ile-de-France (“Equipement mi-lourd 2012” program, DIM Malinf) and the JPK Company for their active support. The authors acknowledge all persons who performed measurement with different instruments: Camille Roesch (Izon Science Europe Ltd, Magdalen Centre, The Oxford Science Park, Oxford, UK), Pierre Peotta (Malvern, Parc club de l’Université, Orsay, France), Philippe Violle (Sympatec, Orsay, France), Serge Réteaud (Beckman Coulter, Villepinte, France), Caroline Ferré and Alain Jalocha (Cilas, Orléans, France). The present work has benefited from the facilities and expertise of the Electron Micoscopy facilities of Imagerie-Gif (http://www.i2bc.paris-saclay.fr/spip.php?article282). This core facility is member of the Infrastructures en Biologie Santé et Agronomie (IBiSA), and is supported by the French national Research Agency under Investments for the Future programs “France-BioImaging”, and the Labex “Saclay Plant Science” (ANR-10-INSB-04-01 and ANR-11-IDEX-0003-02, respectively).
About this article
Cite this article
Varenne, F., Makky, A., Gaucher-Delmas, M. et al. Multimodal Dispersion of Nanoparticles: A Comprehensive Evaluation of Size Distribution with 9 Size Measurement Methods. Pharm Res 33, 1220–1234 (2016). https://doi.org/10.1007/s11095-016-1867-7
- light scattering
- nanoparticle tracking analysis
- particle size distribution
- tunable resistive pulse sensing