Design and Expected Performance of GISMO-2, a Two Color Millimeter Camera for the IRAM 30 m Telescope
- 119 Downloads
We present the main design features for the GISMO-2 bolometer camera, which we build for background-limited operation at the IRAM 30 m telescope on Pico Veleta, Spain. GISMO-2 will operate simultaneously in the 1 and 2 mm atmospherical windows. The 1 mm channel uses a \(32 \times 40\) TES-based backshort under grid (BUG) bolometer array, the 2 mm channel operates with a \(16 \times 16\) BUG array. The camera utilizes almost the entire full field of view provided by the telescope. The optical design of GISMO-2 was strongly influenced by our experience with the GISMO 2 mm bolometer camera, which is successfully operating at the 30 m telescope. GISMO is accessible to the astronomical community through the regular IRAM call for proposals.
KeywordsTransition Edge Sensor Noise Equivalent Power Silicon Monoxide Bolometer Array Thermal Blocker
This work was supported through NSF ATI grants 1020981 and 1106284.
- 2.J.G. Staguhn, C.A. Allen, D.J. Benford, J.A. Chervenak, D.T. Chuss, T.M. Miller, S.H. Moseley, E.J. Wollack, Characterization of TES bolometers used in 2-dimensional backshort-under-grid (BUG) arrays for far-infrared astronomy. Nuclear Instrum. Methods Phys. Res. A 559, 545–547 (2006)ADSCrossRefGoogle Scholar
- 3.J.G. Staguhn, D.J. Benford, C.A. Allen, S.F. Maher, E.H. Sharp, T.J. Ames, R.G. Arendt, D.T. Chuss, E. Dwek, D.J. Fixsen, T.M. Miller, S.H. Moseley, S. Navarro, A. Sievers, E.J. Wollack, Instrument performance of GISMO, a 2 millimeter TES bolometer camera used at the IRAM 30 m Telescope, vol. 7020E, in SPIE, (2008) p. 3.Google Scholar
- 4.P.L. Capak, D. Riechers, N.Z. Scoville, C. Carilli, P. Cox, R. Neri, B. Robertson, M. Salvato, E. Schinnerer, L. Yan, G.W. Wilson, M. Yun, F. Civano, M. Elvis, A. Karim, B. Mobasher, J.G. Staguhn, A massive protocluster of galaxies at a redshift of z \(\sim \)5.3. Nature 470, 233–235 (2011)ADSCrossRefGoogle Scholar
- 7.J.G. Staguhn, A. Kovacs, R.G. Arendt, D.J. Benford, R. Decarli, E. Dwek, D.J. Fixsen, G.C. Hilton, K.D. Irwin, C.A. Jhabvala, A. Karim, S. Leclercq, S.F. Maher, T.M. Miller, S.H. Moseley, E.H. Sharp, F. Walter, E.J. Wollack, The GISMO 2-millimeter deep field in GOODS-N, arXiv:1311.1485 (2013).Google Scholar