This paper describes a method for estimating the oxygen enhanced end-tidal fraction of oxygen (FetOe), the end-tidal fraction of oxygen (FetO2) that is raised by administering supplemental oxygen. The paper has two purposes: the first is to evaluate the method’s accuracy on the bench and in volunteers; the second purpose is to demonstrate how to apply the method to compare two techniques of oxygen administration. The method estimates FetOe by analyzing expired oxygen as oxygen washes out of the lung. The method for estimating FetOe was first validated using a bench simulation in which tracheal oxygen was measured directly. Then it was evaluated in 30 healthy volunteers and compared to the bench simulation. Bland–Altman analysis compared calculated and observed FetOe/FetO2 measurements. After the method was evaluated, it was implemented to compare the FetOe obtained when administering oxygen using two different techniques (pulsed and continuous flow). A total of eighteen breath washout conditions were evaluated on the bench. FetOe estimates and tracheal FetO2 had a mean difference of − 0.016 FO2 with 95% limits of agreement from − 0.048 to 0.016 FO2. Thirteen breath washouts per volunteer were analyzed. Extrapolated and observed FetO2 had a mean difference of − 0.001 FO2 with 95% limits of agreement from − 0.006 to 0.004 FO2. Pulsed flow oxygen (PFO) achieved the same FetOe values as continuous flow oxygen (CFO) using 32.1% ± 2.27% (mean ± SD) of the CFO rate. This paper has demonstrated that the method estimates FetO2 enhanced by administering supplemental oxygen with clinically insignificant differences. This paper has also shown that PFO can obtain FetO2 similar to CFO using approximately one-third of the oxygen volume. After evaluating this method, we conclude that the method provides useful estimates of nasal FetO2 enhanced by supplemental oxygen administration.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Weingarten M. Respiratory monitoring of carbon dioxide and oxygen: a ten-year perspective. J Clin Monit. 1990;6:217–25.
Edmark L, Kostova-Aherdan K, Enlund M, Hedenstierna G. Optimal oxygen concentration during induction of general anesthesia. Anesthesiology. 2003;98:28–33.
Miner JR, Heegaard W, Plummer D. End-tidal carbon dioxide monitoring during procedural sedation. Acad Emerg Med. 2002;9:275–80.
Linko K, Paloheimo M. Monitoring of the inspired and end-tidal oxygen, carbon dioxide, and nitrous oxide concentrations: clinical applications during anesthesia and recovery. J Clin Monit. 1989;5:149–56.
Folke M, Cernerud L, Ekström M, Hök B. Critical review of non-invasive respiratory monitoring in medical care. Med Biol Eng Comput. 2003;41:377–83.
Morrison JE, Friesen RH, Koyle MA. The oxygram: an unappreciated technique for assessing hypoventilation in paediatric anaesthesia. Pediatr Anesth. 1994;4:39–43.
Keidan I, Gravenstein D, Berkenstadt H, Ziv A, Shavit I, Sidi A. Supplemental oxygen compromises the use of pulse oximetry for detection of apnea and hypoventilation during sedation in simulated pediatric patients. Pediatrics. 2008;122:293–8.
Fu ES, Downs JB, Schweiger JW, Miguel RV, Smith RA. Supplemental oxygen impairs detection of hypoventilation by pulse oximetry. Chest. 2004;126:1552–8.
Downs JS. Prevention of hypoxemia: the simple, logical, but incorrect solution. J Clin Anesth. 1994;6:180–1.
Chang K-C, Orr J, Hsu W-C, Yu L, Tsou M-Y, Westenskow DR, Ting C-K. Accuracy of CO2 monitoring via nasal cannulas and oral bite blocks during sedation for esophagogastroduodenoscopy. J Clin Monit Computg. 2016;30:169–73.
Ebert TJ, Novalija J, Uhrich TD, Barney JA. The effectiveness of oxygen delivery and reliability of carbon dioxide waveforms: a crossover comparison of 4 nasal cannulae. Anesth Analg. 2015;120:342–8.
Yanagidate F, Dohi S. Modified nasal cannula for simultaneous oxygen delivery and end-tidal CO2 monitoring during spontaneous breathing. Eur J Anaesthesiol. 2006;23:257–60.
Ortega R, Connor C, Kim S, Djang R, Patel K. Monitoring ventilation with capnography. N Engl J Med. 2012;367:e27–7.
Myles P, Heap M, Langley M. Agreement between end-tidal oxygen concentration and the alveolar gas equation: pre and post cardiopulmonary bypass. Anaesth Intensive Care. 1993;21:240–1.
Chang K-C, Orr J, Hsu W-C, Yu L, Tsou M-Y, Westenskow DR, Ting C-K. Accuracy of CO2 monitoring via nasal cannulas and oral bite blocks during sedation for esophagogastroduodenoscopy. J Clin Monit Comput. 2016;30:169–73.
Banner MJ, Downs JB, Kirby RR, Smith RA, Boysen PG, Lampotang S. Effects of expiratory flow resistance on inspiratory work of breathing. Chest. 1988;93:795–9.
Chen JZ, Katz IM, Pichelin M, Zhu K, Caillibotte G, Noga ML, Finlay WH, Martin AR. Comparison of pulsed versus continuous oxygen delivery using realistic adult nasal airway replicas. Int J Chron Obstr Pulm Dis. 2017;12:2559.
Groombridge C, Chin CW, Hanrahan B, Holdgate A. Assessment of common preoxygenation strategies outside of the operating room environment. Acad Emerg Med. 2016;23:342–6.
Hayes-Bradley C, Lewis A, Burns B, Miller M. Efficacy of nasal cannula oxygen as a preoxygenation adjunct in emergency airway management. Ann Emerg Med. 2016;68:174–80.
Bliss PL, McCoy RW, Adams AB. A bench study comparison of demand oxygen delivery systems and continuous flow oxygen. Respir Care. 1999;44:925–31.
Bland JM, Altman D. Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986;327:307–10.
Fuhrman C, Chouaid C, Herigault R, Housset B, Adnot S. Comparison of four demand oxygen delivery systems at rest and during exercise for chronic obstructive pulmonary disease. Respir Med. 2004;98:938–44.
Yaeger ES, Goodman S, Hoddes E, Christopher KL. Oxygen therapy using pulse and continuous flow with a transtracheal catheter and a nasal cannula. Chest. 1994;106:854–60.
Yuan LC, Jun Z, Min LP. Clinical evaluation of pulse-dose and continuous-flow oxygen delivery. Respir Care. 1995;40:811–4.
Braun SR, Spratt G, Scott GC, Ellersieck M. Comparison of six oxygen delivery systems for COPD patients at rest and during exercise. Chest. 1992;102:694–8.
Roberts C, Bell J, Wedzicha J. Comparison of the efficacy of a demand oxygen delivery system with continuous low flow oxygen in subjects with stable COPD and severe oxygen desaturation on walking. Thorax. 1996;51:831–4.
Zhou S, Chatburn RL. Effect of the anatomic reservoir on low-flow oxygen delivery via nasal cannula: constant flow versus pulse flow with portable oxygen concentrator. Respir Care. 2014;59(8):1199–209.
Wettstein RB, Shelledy DC, Peters JI. Delivered oxygen concentrations using low-flow and high-flow nasal cannulas. Respir Care. 2005;50:604–9.
Bower JS, Brook CJ, Zimmer K, Davis D. Performance of a demand oxygen saver system during rest, exercise, and sleep in hypoxemic patients. Chest. 1988;94:77–80.
Rinow ME, Saltzman AR. Effectiveness of a new oxygen demand valve in chronic hypoxemia. Chest. 1986;90:204–7.
Vlessides M. (2017) Prototype oxygen flowmeter delivers supplemental O2 at variable rate. May 17, 2017. Anesthesiology News, pp. 19.
Orr J, Burk K, Evaluation of end-tidal CO2 measurement during supplemental oxygen delivery using a prototype intelligent oxygen flowmeter. American Society of Anesthesiologists Annual Meeting 2016:A1104.
Miguel-Montanes R, Hajage D, Messika J, Bertrand F, Gaudry S, Rafat C, Labbé V, Dufour N, Jean-Baptiste S, Bedet A. Use of high-flow nasal cannula oxygen therapy to prevent desaturation during tracheal intubation of intensive care patients with mild-to-moderate hypoxemia. Crit Care Med. 2015;43:574–83.
Simon M, Wachs C, Braune S, de Heer G, Frings D, Kluge S. High-flow nasal cannula oxygen versus bag-valve-mask for preoxygenation before intubation in subjects with hypoxemic respiratory failure. Respir Care. 2016;61:1160–7.
This work was supported by a Utah Space Grant Consortium Graduate Research Fellowship (K.M.B.). The authors would like to thank the reviewers for their attentive review and insightful comments.
This study was partially funded by Dynasthetics, LLC.
Conflict of interest
Kyle Burk: Dynasthetics LLC. Kai Kuck: No Conflict of Interest. Joseph Orr: Dynasthetics LLC.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable standards.
Informed consent was obtained from all individual participants included in the study.
Research involving human and animal rights
This article does not contain any studies with animals performed by any of the authors.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Burk, K.M., Kuck, K. & Orr, J.A. Evaluation and application of a method for estimating nasal end-tidal O2 fraction while administering supplemental O2. J Clin Monit Comput 33, 1071–1080 (2019). https://doi.org/10.1007/s10877-019-00264-5
- Nasal end-tidal O2 fraction measurement
- Mathematical modeling
- Pulsed flow oxygen
- Continuous flow oxygen