A Portable Dilution Refrigerator
- 50 Downloads
Refrigerating units of two dilution cryostats manufactured in the form of inserts for a commercial portable nitrogen-free cryostat for liquid helium are described. The 3He circulation in both apparatuses is achieved by its evacuation via its condensation on a wall at a temperature of 0.35–0.5 K, which is cooled by the sorption-based evacuation of 3He from a separate bath. One of the apparatuses operates periodically. A temperature below 0.1 K is maintained in it for several hours, after which the sorption pump needs to be regenerated for ∼0.5 h. The other apparatus includes an additional bath with liquid 3He that maintains its circulation during the sorption-pump regeneration thanks to the large specific heat of this liquid. The minimum temperature reached in experiments was 0.035 K. The refrigerators operate at a circulation rate of ∼2–10 µmol/s and consume 1–2 l/day of liquid helium for maintaining their operation. The refrigerators are designed to cool photodetectors or samples in experiments that do not require a high refrigerating capacity.
KeywordsPhysical Chemistry Helium Minimum Temperature Liquid Helium Circulation Rate
Unable to display preview. Download preview PDF.
- 1.Wollmann, D.A., Irvine, K.D., Hilton, G.C., et al., J.Microsc. (Oxford), 1997, vol. 188, no.3, p. 196.Google Scholar
- 3.Akimov, D.Yu., Prib. Tekh. Eksp., 2001, no. 5, p. 6 [Instrum. Exp. Tech. (Engl. Transl.), 2001, no. 5, p. 575].Google Scholar
- 4.www.oxinstr.com.Google Scholar
- 5.Pobell, F., Matter and Methods at Low Temperatures, Berlin: Springer, 1996.Google Scholar
- 6.Edel'man, V.S., Prib. Tekh. Eksp., 2002, no. 3, p. 139 [Instrum. Exp. Tech. (Engl. Transl.), 2002, no. 3, p. 420].Google Scholar
- 8.Edel'man, V.S., Prib. Tekh. Eksp., 1972, no. 4, p. 229.Google Scholar
- 10.Radebaugh, R., NBS Technical Note, 1967, no. 362.Google Scholar
- 11.Keller, W.F., Helium-3 and Helium-4, New York: Plenum, 1969.Google Scholar