Mechanical ventricular support with left ventricular assist device (LVAD) has emerged as a durable and safe therapy, both as bridge-to-transplant (BTT) or destination therapy (DT), in patients with advanced heart failure (HF). However, the occurrence of pump thrombosis (PT) still represents a serious complication, especially when LVADs of first or second generation are implanted. During the latest years, some investigations have recognized the occurrence of cavitation, evidenced through transthoracic echocardiography (TTE), as a potential early and indirect sign of PT. In the present manuscript, we reviewed the available data on the occurrence of cavitation in LVAD patients as an early potential marker of PT, also presenting the hemodynamic mechanisms involved.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
Tax calculation will be finalised during checkout.
High-intensity transient signals
Left ventricular assist device
Left ventricular end diastolic diameter
Li X, Kondray V, Tavri S, Ruhparwar A, Azeze S, Dey A, Partovi S, Rengier F (2019) Role of imaging in diagnosis and management of left ventricular assist device complications. Int J Cardiovasc Imaging 35:1365–1377. https://doi.org/10.1007/s10554-019-01562-4
Rigatelli G, Santini F, Faggian G (2012) Past and present of cardiocirculatory assist devices: a comprehensive critical review. J Geriatr Cardiol 9:389–400
Han JJ, Acker MA, Atluri P (2018) Left ventricular assist devices. Circulation 138:2841–2851
Prinzing A, Herold U, Berkefeld A, Krane M, Lange R, Voss B (2016) Left ventricular assist devices-current state and perspectives. J Thorac Dis 8:E660–E666
Uriel N, Han J, Morrison KA, Nahumi N, Yuzefpolskaya M, Garan AR, Duong J, Colombo PC, Takayama H, Thomas S, Naka Y, Jorde UP (2014) Device thrombosis in HeartMate II continuous-flow left ventricular assist devices: a multifactorial phenomenon. J Heart Lung Transplant 33:51–59
Girod G, Jaussi A, Rosset C, De Werra P, Hirt F, Kappenberger L (2002) Cavitation versus degassing: in vitro study of the microbubble phenomenon observed during echocardiography in patients with mechanical prosthetic cardiac valves. Echocardiography 19:531–536
Johansen P (2004) Mechanical heart valve cavitation. Expert Rev Med Devices 1:95–104
Johansen P, Andersen TS, Hasenkam JM, Nygaard H, Paulsen PK (2014) Mechanical heart valve cavitation in patients with bileaflet valves. Conf Proc IEEE Eng Med Biol Soc:5655–5658
Davis MK, Ha R, Banerjee D (2014) The presence of air bubbles in the aorta of a patient with a HeartMate II left ventricular assist device: a novel sign of outflow graft obstruction. ASAIO J 60:600–602
Raina A, Agarwal R, Benza RL (2014) Spontaneous microbubbles in the aortic root and thrombosis of a continuous-flow left ventricular assist device. J Heart Lung Transplant 33:550–551
Nguyen TN, Tri Nguyen NM, Truong VT, Vo VM, Rigatelli G (2018) Cavitation phenomenon creating bubbles and their explosion in the coronary arteries causes damage to the endothelium and start the atherosclerotic process. JACC 71:A269. https://doi.org/10.1016/S0735-1097(18)30810-6
Andersen TS, Johansen P, Christensen BO, Paulsen PK, Nygaard H, Hasenkam JM (2006) Intraoperative and postoperative evaluation of cavitation in mechanical heart valve patients. Ann Thorac Surg 81:34–41
Zong YJ, Liu G,Long J, Zhou D, Zhang L, Feng Y, Wan M (2017) Focused ultrasound cavitation induced injury and localized atherosclerosis plaques of rabbit abdominal arterial endothelium. Conference: 2017 IEEE International Ultrasonics Symposium (IUS). 1-4. https://doi.org/10.1109/ULTSYM.2017.8092766
Zong YJ, Wang R, Zhang L, Liu G, Zou X, Feng Y, Wan M (2015) Local cavitation induced vessel wall injury and its potential application in developing atherosclerosis model, 2015 IEEE International Ultrasonics Symposium (IUS) 1-4 https://doi.org/10.1109/ULTSYM.2015.0226
Graf T, Fischer H, Reul H, Rau G (1991) Cavitation potential of mechanical heart valve prostheses. Int J Artif Organs 14:169–174
Kafesjian R, Howanec M, Ward GD, Diep L, Wagstaff LS, Rhee R (1994) Cavitation damage of pyrolytic carbon in mechanical heart valves. J Heart Valve 3:2–7
Barak M, Katz Y (2005) Microbubbles: pathophysiology and clinical implications. Chest 128:2918–2932
Dhillon SS, Khoo NS, Quinonez L, Buchholz H (2013) Spontaneous endogenous microbubbles in a child with Berlin heart ventricular assist device. ASAIO J 59:181–182
Malik AB, Johnson A, Tahamont MV (1982) Mechanisms of lung vascular injury after intravascular coagulation. Ann N Y Acad Sci 384:213–234
Kolla KR, Maltais S, Pereira NL, Chaliki HP (2018) Microbubbles in the aorta and left ventricle of a patient with a left ventricular assist device: a unique presentation of pump thrombosis leading to urgent surgery. Cureus 10:e2463
Palaniswamy C, Garg J, Dutta T, Shah A, Gass A, Lanier GM (2014) Cavitation phenomenon: a novel echocardiographic finding in pump thrombosis. J Card Fail 20:874–875
Dimitrov K, Riebandt J, Haberl T, Wiedemann D, Simon P, Laufer G, Schima H, Zimpfer D (2016) High-intensity transient signals in the outflow graft and thrombosis of a HeartWare left ventricular assist device. Ann Thorac Surg 101:e83–e85
Letarte L, Sears-Rogan P, Boyce S, Tyson M, Wang Z (2015) Spontaneous microbubble formation is an indicator of LVAD pump thrombosis. J Heart Lung Transplant 34:S110–S111
Letarte L, Wang Z, Rodrigo ME, Sheikh FH, Hofmeyer M, Boyce SW, Najjar SS, Majure DT (2016) Microbubbles in patients with LVADs: echocardiographic markers for LVAD pump thrombosis? J Heart Lung Transplnat 35:S241
Stainback RF, Estep JD, Agler DA, Birks EJ, Bremer M, Hung J, Kirkpatrick JN, Rogers JG, Shah NR, American Society of Echocardiography (2015) Echocardiography in the management of patients with left ventricular assist devices: recommendations from the American Society of Echocardiography. J Am Soc Echocardiogr 28:853–909
Ammar KA, Umland MM, Kramer C, Sulemanjee N, Jan MF, Khandheria BK, Seward JB, Paterick TE (2012) The ABCs of left ventricular assist device echocardiography: a systematic approach. Eur Heart J Cardiovasc Imaging 13:885–899
Kato TS, Colombo PC, Nahumi N, Kitada S, Takayama H, Naka Y, Di Tullio MR, Homma S, Mancini D, Jorde UP, Uriel N (2014) Value of serial echo-guided ramp studies in a patient with suspicion of device thrombosis after left ventricular assist device implantation. Echocardiography 31:E5–E9
Imamura T, Jeevanandam V, Kim G, Raikhelkar J, Sarswat N, Kalantari S, Smith B, Rodgers D, Besser S, Chung B, Nguyen A, Narang N, Ota T, Song T, Juricek C, Mehra M, Costanzo MR, Jorde UP, Burkhoff D, Sayer G, Uriel N (2019) Optimal hemodynamics during left ventricular assist device support are associated with reduced readmission rates. Circ Heart Fail 12:e005094
Yang F, Kormos RL, Antaki JF (2015) High-speed visualization of disturbed pathlines in axial flow ventricular assist device under pulsatile conditions. J Thorac Cardiovasc Surg 150:938–944
Chiu AH, Haluszkiewicz E, McAuliffe W (2015) High-speed visualization of disturbed pathlines in axial flow ventricular assist device under pulsatile conditions. J Thorac Cardiovasc Surg 150:938–944
Rigatelli G, Bacich D, Zuin M, Braggion G, Dell'Avvocata F (2019) Cardiac pump-induced platypnea-orthodeoxia. Eur Heart J Cardiovasc Imaging. https://doi.org/10.1093/ehjci/jey228
Ferns J, Dowling R, Bhat G (2001) Evaluation of a patient with left ventricular assist device dysfunction. ASAIO J 47:696–698
Conflict of interest
None of the authors have conflict of interest to declare
This article does not contain any studies with human participants or animals performed by any of the authors.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Zuin, M., Rigatelli, G., Braggion, G. et al. Cavitation in left ventricular assist device patients: a potential early sign of pump thrombosis. Heart Fail Rev 25, 965–972 (2020). https://doi.org/10.1007/s10741-019-09884-0
- Left ventricular assist device
- Pump thrombosis