Feasibility of a single-beat prospective ECG-gated cardiac CT for comprehensive evaluation of aortic valve disease using a 256-detector row wide-volume CT scanner: an initial experience
The aim of our study was to investigate the feasibility of single-beat prospective electrocardiogram (ECG)-gated cardiac computed tomography (CT) using a 256-detector row wide-volume CT scanner for functional and anatomical evaluation of the aortic valve (AV) and coronary arteries in patients with AV disease. A total of 50 patients who underwent cardiac CT scan with a wide-volume 256-detector row CT scanner for the evaluation of AV and aorta were retrospectively enrolled. Cardiac CT was performed using the prospective ECG-gated acquisition mode, and AV image quality was analyzed using a four-point grading system. Severity of aortic stenosis (AS) and aortic regurgitation (AR) were assessed by CT and correlated to that assessed by transthoracic echocardiography (TTE) based on kappa statistics (κ). Estimated radiation exposure was assessed. Among 50 patients, 44 underwent cardiac CT with single-beat acquisition. The median image quality score of AV was 3.0 on the systolic phase and 4.0 on the diastolic phase. Severity of AS and AR by CT showed moderate agreement with TTE. The mean effective radiation dose was 3.75 ± 1.43 mSv for CT angiography. Using 256-detector row wide-volume CT, the single-beat cardiac CT is feasible for evaluation of AV disease and the coronary arteries, with acceptable image quality and a low radiation dose of 3.75 mSv.
KeywordsAortic valve Multidetector computed tomography Echocardiography Aortic valve stenosis Aortic valve insufficiency
Aortic valve area
Regurgitant orifice area
Compliance with ethical standards
Conflict of interest
All authors declare that there are no conflicts of interest.
- 4.Achenbach S, Delgado V, Hausleiter J, Schoenhagen P, Min JK, Leipsic JA (2012) SCCT expert consensus document on computed tomography imaging before transcatheter aortic valve implantation (TAVI)/transcatheter aortic valve replacement (TAVR). J Cardiovasc Comput Tomogr 6(6):366–380CrossRefPubMedGoogle Scholar
- 17.Halpern EJ, Mallya R, Sewell M, Shulman M, Zwas DR (2009) Differences in aortic valve area measured with CT planimetry and echocardiography (continuity equation) are related to divergent estimates of left ventricular outflow tract area. AJR Am J Roentgenol 192(6):1668–1673CrossRefPubMedGoogle Scholar
- 20.Kurlansky PA, Williams DB, Traad EA, Carrillo RG, Zucker M, Ebra G (2007) Surgical management of aortic valve disease in elderly patients with and without coronary artery disease: influence on quality of life. J Cardiovasc Surg 48(2):215–226Google Scholar
- 22.Latif MA, Sanchez FW, Sayegh K et al (2016) Volumetric single-beat coronary computed tomography angiography: relationship of image quality, heart rate, and body mass index. Initial patient experience with a new computed tomography scanner. J Comput Assist Tomogr 40(5):763–772CrossRefPubMedGoogle Scholar