Portable low-power thermal cycler with dual thin-film Pt heaters for a polymeric PCR chip
- 145 Downloads
Polymerase chain reaction (PCR) has been widely used for major definite diagnostic tool, but very limited its place used only indoor such as hospital or diagnosis lab. For the rapid on-site detection of pathogen in an outdoor environment, a low-power cordless polymerase chain reaction (PCR) thermal cycler is crucial module. At this point of view, we proposed a low-power PCR thermal cycler that could be operated in an outdoor anywhere. The disposable PCR chip was made of a polymeric (PI/PET) film to reduce the thermal mass. A dual arrangement of the Pt heaters, which were positioned on the top and bottom of the PCR chip, improved the temperature uniformity. The temperature sensor, which was made of the same material as the heater, utilized the temperature dependence of the Pt resistor to ensure simple fabrication of the temperature sensor. Cooling the PCR chip using dual blower fans enabled thermal cycling to operate with a lower power than that of a Peltier element with a high power consumption. The PCR components were electrically connected to a control module that could be operated with a Li-ion battery (12 V), and the PCR conditions (temperature, time, cycle, etc.) were inputted on a touch screen. For 30 PCR cycles, the accumulated power consumption of heating and cooling was 7.3 Wh, which is easily available from a compact battery. Escherichia coli genomic DNA (510 bp) was amplified using the proposed PCR thermal cycler and the disposable PCR chip. A similar DNA amplification capability was confirmed using the proposed portable and low-power thermal cycler compared with a conventional thermal cycler.
KeywordsPCR chip Thermal cycler Pt heater Pt RTD Dual arrangement Low-power Battery-powered
This research was partially supported by BioNano Health-Guard Research Center funded by the Ministry of Science and ICT(MSIT) of Korea as Global Frontier Project (Grant number H-GUARD_2015M3A6B2063547, H-GUARD_2013M3A6B2078950), and also partially supported by the Biomedical Integrated Technology Research Project through a grant provided by GIST in 2017 and Chonnam National University Hospital Biomedical Research Institute (CRI 16-072-3). We thank Dr. Kyoungsook Park (H-GUARD, Korea) and Dr. Byeong Il Kim (BIOTNS Inc., Korea) for their technical supports in the sample preparation and the PCR chip fabrication, respectively.
- R. Aschenbrenner, R. Miessner, H. Reichl, 1st Polymeric Electronics Packaging on IEEE International Symposium, 86–94 (1997)Google Scholar
- W. Chung, N. Kak, C. Park, W. Jung, S. Jung, K. H. Kim, J. Shim, K. Y. Hwang, H. Lim, J. H. Kim, H. Nam, 16th International Conference on Miniaturized Systems for Chemistry and Life Sciences 1399–1401 (2012)Google Scholar
- D.S. Lee, S.H. Park, K.Y. Chung, H.B. Pyo, Sensors 8, 558–564 (2008)Google Scholar
- S. Sinha, S. Roy, C. K. Sarkar, International Symposium on Devices MEMS, Intelligent Systems & Communication, 26–31 (2011)Google Scholar