MouthLab: A Tricorder Concept Optimized for Rapid Medical Assessment
The goal of rapid medical assessment (RMA) is to estimate the general health of a patient during an emergency room or a doctor’s office visit, or even while the patient is at home. Currently the devices used during RMA are typically “all-in-one” vital signs monitors. They require time, effort and expertise to attach various sensors to the body. A device optimized for RMA should instead require little effort or expertise to operate and be able to rapidly obtain and consolidate as much information as possible. MouthLab is a battery powered hand-held device intended to acquire and evaluate many measurements such as non-invasive blood sugar, saliva and respiratory biochemistry. Our initial prototype acquires standard vital signs: pulse rate (PR), breathing rate (BR), temperature (T), blood oxygen saturation (SpO2), blood pressure (BP), and a three-lead electrocardiogram. In our clinical study we tested the device performance against the measurements obtained with a standard patient monitor. 52 people participated in the study. The measurement errors were as follows: PR: −1.7 ± 3.5 BPM, BR: 0.4 ± 2.4 BPM, T: −0.4 ± 1.24 °F, SpO2: −0.6 ± 1.7%. BP systolic: −1.8 ± 12 mmHg, BP diastolic: 0.6 ± 8 mmHg. We have shown that RMA can be easily performed non-invasively by patients with no prior training.
KeywordsTricorder Vital signs monitoring Medical assessment
We thank Sunny Smith, Charles Della Santina, and Lani Swarthout for material support, helpful discussions, and contribution to the logistics associated with prototype development and human studies. We thank Divya Kernik, Lucas Fridman, and Alexandra Della Santina for their contribution to the early prototype designs.
- 2.Caduff, A., M. S. Talary, M. Mueller, F. Dewarrat, J. Klisic, M. Donath, L. Heinemann, and W. A. Stahel. Non-invasive glucose monitoring in patients with Type 1 diabetes: a multisensor system combining sensors for dielectric and optical characterisation of skin. Biosens. Bioelectron. 24(9):2778–2784, 2009.PubMedCrossRefGoogle Scholar
- 8.Haykin, S. Neural Networks A Comprehensive Foundation. New York: IEEE Press, 1994.Google Scholar
- 12.Ramsay, M. A., M. Usman, E. Lagow, M. Mendoza, E. Untalan, and V. E. De. The accuracy, precision and reliability of measuring ventilatory rate and detecting ventilatory pause by rainbow acoustic monitoring and capnometry. Anesth. Analg. 117(1):69–75, 2013. doi: 10.1213/ANE.0b013e318290c798.PubMedCrossRefGoogle Scholar
- 14.Webster, J. G. Medical Instrumentation Application and Design (4th ed.). Hoboken: Wiley, 2009.Google Scholar
- 16.Zloczower, M., A. Z. Reznick, R. O. Zouby, and R. M. Nagler. Relationship of flow rate, uric acid, peroxidase, and superoxide dismutase activity levels with complications in diabetic patients: can saliva be used to diagnose diabetes? Antioxid. Redox Signal. 9(6):765–773, 2007.PubMedCrossRefGoogle Scholar