A novel MR-compatible sensor to assess active medical device safety: stimulation monitoring, rectified radio frequency pulses, and gradient-induced voltage measurements
- 21 Downloads
To evaluate the function of an active implantable medical device (AIMD) during magnetic resonance imaging (MRI) scans. The induced voltages caused by the switching of magnetic field gradients and rectified radio frequency (RF) pulse were measured, along with the AIMD stimulations.
Materials and methods
An MRI-compatible voltage probe with a bandwidth of 0–40 kHz was designed. Measurements were carried out both on the bench with an overvoltage protection circuit commonly used for AIMD and with a pacemaker during MRI scans on a 1.5 T (64 MHz) MR scanner.
The sensor exhibits a measurement range of ± 15 V with an amplitude resolution of 7 mV and a temporal resolution of 10 µs. Rectification was measured on the bench with the overvoltage protection circuit. Linear proportionality was confirmed between the induced voltage and the magnetic field gradient slew rate. The pacemaker pacing was recorded successfully during MRI scans.
The characteristics of this low-frequency voltage probe allow its use with extreme RF transmission power and magnetic field gradient positioning for MR safety test of AIMD during MRI scans.
KeywordsLow-frequency sensor Induced voltage Active medical device MR safety Malfunction
Authors would like to thank the F.E.D.E.R. as well as the Région Lorraine for financial support and Sorin CRM SAS for the devices lending.
Study conception and design: Thérèse Barbier, Sarra Aissani, Nicolas Weber. Acquisition of data: Thérèse Barbier, Sarra Aissani. Analysis and interpretation of data: Thérèse Barbier, Sarra Aissani. Drafting of manuscript: Sarra Aissani, Jacques Felblinger. Critical revision: Cédric Pasquier.
Compliance with ethical standards
Conflict of interest
Authors declare they have no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
- 8.New PF, Rosen BR, Brady TJ, Buonanno FS, Kistler JP, Burt CT, Hinshaw WS, Newhouse JH, Pohost GM, Taveras JM (1983) Potential hazards and artifacts of ferromagnetic and nonferromagnetic surgical and dental materials and devices in nuclear magnetic resonance imaging. Radiology 147:139–148CrossRefPubMedGoogle Scholar
- 10.Shellock FG (2009) Reference manual for magnetic resonance safety: 2016 edition. Biomedical Research Publishing Group, Los AngelesGoogle Scholar
- 11.ISO/TS 10974:2012(E) (2012) Assessment of the safety of magnetic resonance imaging for patients with an active implantable medical deviceGoogle Scholar
- 18.ASTM F2182-11A (2011) Measurement of radio frequency induced heating on or near passive implants during magnetic resonance imaging. ASTM, West ConshohockenGoogle Scholar
- 19.International Electrotechnical Commission (2010) Medical equipment—IEC 60601-2-33: particular requirements for the safety of magnetic resonance equipment, 3rd edn. International Electrotechnical Commission, GenevaGoogle Scholar
- 24.Ferreira AM, Costa F, Tralhão A, Marques H, Cardim N, Adragão P (2014) MRI-conditional pacemakers: current perspectives. Med Devices Auckl NZ 7:115–124Google Scholar
- 27.Stevenson RA (1998) Feedthrough EMI filter with ground isolation for cardiac pacemakers and implantable cardioverter defibrillators. In: Proceedings of the 20th annual international conference of IEEE Engineering in Medicine and Biology Society, vol 20 Biomedical Engineering Year 2000 Cat No. 98CH36286, vol 6, p 3319–3323Google Scholar