Variable slice thickness (VAST) EPI for the reduction of susceptibility artifacts in whole-brain GE-EPI at 7 Tesla
- 147 Downloads
A new technique for 2D gradient-recalled echo echo-planar imaging (GE-EPI) termed ‘variable slice thickness’ (VAST) is proposed, which reduces signal losses caused by through-slice susceptibility artifacts, while keeping the volume repetition time (TR) manageable. The slice thickness is varied across the brain, with thinner slices being used in the inferior brain regions where signal voids are most severe.
Materials and methods
Various axial slice thickness schemes with identical whole-brain coverage were compared to regular EPI, which may either suffer from unfeasibly long TR if appropriately thin slices are used throughout, or signal loss if no counter-measures are taken. Evaluation is based on time-course signal-to-noise (tSNR) maps from resting state data and a statistical group-level region of interest (ROI) analysis on breath-hold fMRI measurements.
The inferior brain region signal voids with static B0 inhomogeneities could be markedly reduced with VAST GE-EPI in contrast to regular GE-EPI. ROI-averaged event-related signal changes showed 48% increase in VAST compared to GE-EPI with regular “thick” slices. tSNR measurements proved the comparable signal robustness of VAST in comparison to regular GE-EPI with thin slices.
A novel acquisition strategy for functional 2D GE-EPI at ultrahigh magnetic field is presented to reduce susceptibility-induced signal voids and keep TR sufficiently short for whole-brain coverage.
KeywordsfMRI Echo-planar imaging Susceptibility artifact Repetition time Slice thickness 7 Tesla ultrahigh field MRI
S Brunheim: project development, programming, data collection, data analysis, manuscript writing. S Johst: project development, programming, data collection, manuscript editing. V Pfaffenrot: data collection, data analysis, manuscript editing. S Maderwald: data collection, manuscript editing. HH Quick: project supervision, manuscript editing. BA Poser: project supervision, project development, programming, data analysis, manuscript editing.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflicts of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki Declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 10.Mansfield P, Maudsley AA (1977) Planar spin imaging by NMR. J Magn Reson 27:101–119Google Scholar
- 15.Weiskopf N, Hutton C, Josephs O, Deichmann R, Gottfried JA, Hutton C, Turner R, Weiskopf N, Hutton C, Josephs O, Deichmann R (2006) Optimal EPI parameters for reduction of susceptibility-induced BOLD sensitivity losses: a whole-brain analysis at 3 T and 1.5 T. Neuroimage 33:493–504CrossRefPubMedGoogle Scholar
- 26.Guérin B, Stockmann JP, Baboli M, Torrado-Carvajal A, Stenger AV, Wald LL (2016) Robust time-shifted spoke pulse design in the presence of large B0 variations with simultaneous reduction of through-plane dephasing, B1+ effects, and the specific absorption rate using parallel transmission. Magn Reson Med 76:540–554CrossRefPubMedGoogle Scholar
- 35.Fautz H, Vogel M, Gross P (2008) B1 mapping of coil arrays for parallel transmission. Intl Soc Mag Reson Med, Proc, p 1247Google Scholar
- 42.Polimeni JR, Bhat H, Witzel T, Benner T, Feiweier T, Inati SJ, Renvall V, Heberlein K, Wald LL (2016) Reducing sensitivity losses due to respiration and motion in accelerated echo planar imaging by reordering the autocalibration data acquisition. Magn Reson Med 75:665–679CrossRefPubMedGoogle Scholar
- 52.Talairach J, Tournoux P (1988) Co-planar stereotaxic atlas of the human brain. 3-Dimensional proportional system: an approach to cerebral imaging. Thieme, New YorkGoogle Scholar
- 58.Evans AC, Collins DL, Mills SR, Brown ED, Kelly RL, Peters TM (1993) 3D statistical neuroanatomical models from 305 MRI volumes. In: Proc. IEEE - Nuclear Science Symposium and Medical Imaging Conference, pp 1813–1817Google Scholar
- 67.Vu AT, Jamison K, Glasser MF, Smith SM, Coalson T, Moeller S, Auerbach EJ, Ugurbil K, Yacoub E (2017) Tradeoffs in pushing the spatial resolution of fMRI for the 7T human connectome project. Neuroimage 154:23–32Google Scholar
- 68.Poser BA, Kemper VG, Ivanov D, Uludag K, Kannengiesser SA, Barth M (2013) CAIPIRINHA-accelerated 3D EPI for high temporal and/or spatial resolution EPI acquisitions. In: Proc. to ESMRMB Congr. Toulouse, Fr, p 287Google Scholar
- 73.Pfaffenrot V, Brunheim S, Rietsch SHG, Ernst TM, Orzada S, Quick HH (2017) An 8/15-channel Tx/Rx head neck RF coil combination with semi-dynamic B1 shimming for improved fMRI of the cerebellum at 7 T. In: Proc Intl Soc Mag Reson Med, p 1126Google Scholar