Application of the limited-memory quasi-Newton algorithm for multi-dimensional, large flip-angle RF pulses at 7T
- 282 Downloads
Ultrahigh field MRI provides great opportunities for medical diagnostics and research. However, ultrahigh field MRI also brings challenges, such as larger magnetic susceptibility induced field changes. Parallel-transmit radio-frequency pulses can ameliorate these complications while performing advanced tasks in routine applications. To address one class of such pulses, we propose an optimal-control algorithm as a tool for designing advanced multi-dimensional, large flip-angle, radio-frequency pulses. We contrast initial conditions, constraints, and field correction abilities against increasing pulse trajectory acceleration factors.
Materials and methods
On an 8-channel 7T system, we demonstrate the quasi-Newton algorithm with pulse designs for reduced field-of-view imaging with an oil phantom and in vivo with scans of the human brain stem. We used echo-planar imaging with 2D spatial-selective pulses. Pulses are computed sufficiently rapid for routine applications.
Our dataset was quantitatively analyzed with the conventional mean-square-error metric and the structural-similarity index from image processing. Analysis of both full and reduced field-of-view scans benefit from utilizing both complementary measures.
We obtained excellent outer-volume suppression with our proposed method, thus enabling reduced field-of-view imaging using pulse trajectory acceleration factors up to 4.
KeywordsMRI RF pulse design Parallel transmit Ultrahigh field MRI
MSV and TV acknowledge support from the Danish Council for Strategic Research, the Danish National Research Foundation (DNRF59), the Ministry of Higher Education and Science (AU-2010-612-181), the Lundbeck Foundation, Aarhus University Research Foundation, and the Programme Commission on Strategic Growth Technologies, Innovation Fund Denmark (0603-00439B). IIM, SV, and DS thank Deutsche Forschungsgemeinschaft grant (SU 192/32-1) for support of this work, and Dr. Daniel Edelhoff and Mr. Raphael Mocek for assistance in the experiments.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Informed consent was obtained from all individual participants included in the study.
- 4.Guérin B, Stockmann JP, Baboli M, Torrado-Carvajal A, Stenger AV, Wald LL (2015) Robust time-shifted spoke pulse design in the presence of large B0 variations with simultaneous reduction of through-plane dephasing, B1+ effects, and the specific absorption rate using parallel transmission. Magn Reson Med. doi: 10.1002/mrm.25902 Google Scholar
- 6.Vinding MS, Guérin B, Vosegaard T, Nielsen N Chr (2015) Local SAR, global SAR, and power-constrained large-flip-angle pulses with optimal control and virtual observation points. Magn Reson Med. doi: 10.1002/mrm.26086
- 7.Bryson AE, Ho Y-C (1975) Applied optimal control: optimization, estimation, and control. Hemisphere publishing corporation, Washington, DCGoogle Scholar
- 21.Press WH, Teukolsky SA, Vetterling WT, Flannery BP (1993) Numerical recipes in C: the art of scientific computing. Cambridge University Press, Cambridge, New York, Port Chester, Melbourne, SydneyGoogle Scholar
- 29.Pauly JM, Nishimura DG, Macovski A (1989) A linear class of large-tip-angle selective excitation pulses. J Magn Reson 82:571–587Google Scholar
- 33.Ndajah P, Kikuchi H, Yukawa M, Watanabe H, Muramatsu S (2010) SSIM image quality metric for denoised images. In: Mastorakis NE, Mladenov V (eds) Proceedings of the 3rd WSEAS international conference on Visualization, imaging and simulation (VIS '10), World Scientific and Engineering Academy and Society (WSEAS), Stevens Point, Wisconsin, pp 53–57Google Scholar
- 35.Brenner D, Tse DHY, Ledden PJ, Neumann C, Stöcker T (2015) Rapid and accurate PTX B1 mapping using 3DREAM with dual interferometry. In: Proceedings of the international society for magnetic resonance in medicine scientific meeting and exhibition, 0109Google Scholar
- 36.Brenner D, Tse DHY, Pracht ED, Feiweier T, Stirnberg R, Stöcker T (2014) 3DREAM—a three-dimensional variant of the DREAM sequence. In: Proceedings of the international society for magnetic resonance in medicine scientific meeting and exhibition, 1455Google Scholar