This study aimed to evaluate in vitro performance of near-infrared laser transillumination (NIR-LTI) for detecting early occlusal caries in permanent teeth and compare it with quantitative light-induced fluorescence (QLF), DIAGNOdent Pen (DDPen), and conventional radiography (CR). Ninety-four occlusal surfaces presenting International Caries Detection and Assessment System (ICDAS) scores ranging from 0 to 3 were selected. For the NIR-LTI examination, images were captured using a prototype, which consists of a laser beam (808 nm) and an infrared CCD camera. One occlusal site on each tooth was assessed twice by two examiners. The teeth were prepared histologically and assessed for the presence of early caries. The intraexaminer correlation showed no difference between the NIR-LTI, DDPen, and QLF analytical methods, but all these methods differed from CR. Interexaminer reproducibility was moderate for NIR-LTI, which showed sensitivity (0.68), specificity (0.85), accuracy (0.73), and area under the receiver-operating characteristic (ROC) curve (0.76) similar to those of the fluorescence method and different from those of the CR. In conclusion, the performance of NIR-LTI was comparable to that of DDPen and QLF and may therefore be considered a valid and reliable alternative for the diagnosis of incipient lesions on the occlusal surface of permanent teeth.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Price excludes VAT (USA)
Tax calculation will be finalised during checkout.
Bader JD, Shugars DA, Bonito AJ (2001) Systematic review of selected dental caries diagnostic and management methods. J Dent Educ 65:960–968
Hall A, Girkin JM (2004) A review of potential new diagnostic modalities for caries lesions. J Dent Res 83:C89–C94. doi:10.1177/154405910408301S18
Pretty IA (2006) Caries detection and diagnosis: novel technologies. J Dent 34:727–739. doi:10.1016/j.jdent.2006.06.001
Gomez J, Zakian C, Salsone S, Pinto SC, Taylor A, Pretty IA, Ellwood R (2013) In vitro performance of different methods in detecting occlusal caries lesions. J Dent 41:180–186. doi:10.1016/j.jdent.2012.11.003
Ferreira Zandoná A, Santiago E, Eckert G, Fontana M, Ando M, Zero DT (2010) Use of ICDAS combined with quantitative light-induced fluorescence as a caries detection method. Caries Res 44:317–322. doi:10.1159/000317294
Kühnisch J, Ifland S, Tranaeus S, Hickel R, Stösser L, Heinrich-Weltzien R (2007) In vivo detection of non-cavitated caries lesions on occlusal surfaces by visual inspection and quantitative light-induced fluorescence. Acta Odontol Scand 65:183–188. doi:10.1080/00016350701291685
De Benedetto MS, Morais CC, Novaes TF, de Almeida Rodrigues J, Braga MM, Mendes FM (2011) Comparing the reliability of a new fluorescence camera with conventional laser fluorescence devices in detecting caries lesions in occlusal and smooth surfaces of primary teeth. Lasers Med Sci 26:157–162. doi:10.1007/s10103-010-0757-1
Huth KC, Neuhaus KW, Gygax M, Bücher K, Crispin A, Paschos E, Hickel R, Lussi A (2008) Clinical performance of a new laser fluorescence device for detection of occlusal caries lesions in permanent molars. J Dent 36:1033–1040. doi:10.1016/j.jdent.2008.08.013
Diniz MB, Boldieri T, Rodrigues JA, Santos-Pinto L, Lussi A, Cordeiro RC (2012) The performance of conventional and fluorescence-based methods for occlusal caries detection: an in vivo study with histologic validation. J Am Dent Assoc 143:339–350. doi:10.14219/jada.archive.2012.0176
Darling CL, Huynh GD, Fried D (2006) Light scattering properties of natural and artificially demineralized dental enamel at 1310 nm. J Biomed Opt 11:34023. doi:10.1117/1.2204603
Bühler C, Ngaotheppitak P, Fried D (2005) Imaging of occlusal dental caries (decay) with near-IR light at 1310-nm. Opt Express 13:573–582. doi:10.1364/OPEX.13.000573
Fried D, Featherstone JD, Darling CL, Jones RS, Ngaotheppitak P, Bühler CM (2005) Early caries imaging and monitoring with near-infrared light. Dent Clin N Am 49:771–793. doi:10.1016/j.cden.2005.05.008
Diniz MB, Sciasci P, Rodrigues JA, Lussi A, Cordeiro RC (2011) Influence of different professional prophylactic methods on fluorescence measurements for detection of occlusal caries. Caries Res 45:264–268. doi:10.1159/000326110
Souza JF, Boldieri T, Diniz MB, Rodrigues JA, Lussi A, Cordeiro RC (2013) Traditional and novel methods for occlusal caries detection: performance on primary teeth. Lasers Med Sci 28:287–295. doi:10.1007/s10103-012-1154-8
Francescut P, Zimmerli B, Lussi A (2006) Influence of different storage methods on laser fluorescence values: a two-year study. Caries Res 40:181–185. doi:10.1159/000092223
Diniz MB, Lima LM, Eckert G, Zandona AG, Cordeiro RC, Pinto LS (2011) In vitro evaluation of ICDAS and radiographic examination of occlusal surfaces and their association with treatment decisions. Oper Dent 36:133–142. doi:10.2341/10-006-L
Rodrigues JA, Hug I, Diniz MB, Lussi A (2008) Performance of fluorescence methods, radiographic examination and ICDAS II on occlusal surfaces in vitro. Caries Res 42:297–304. doi:10.1159/000148162
Lussi A, Hack A, Hug I, Heckenberger H, Megert B, Stich H (2006) Detection of approximal caries with a new laser fluorescence device. Caries Res 40:97–103. doi:10.1159/000091054
Diniz MB, Rodrigues JA, de Paula AB, Cordeiro RCL (2009) In vivo evaluation of laser fluorescence performance using different cut-off limits for occlusal caries detection. Lasers Med Sci 24:295–300. doi:10.1007/s10103-008-0547-1
de Josselin de Jong E, Sundström F, Westerling H, Tranaeus S, ten Bosch JJ, Angmar-Månsson B (1995) A new method for in vivo quantification of changes in initial enamel caries with laser fluorescence. Caries Res 29:2–7. doi:10.1159/000262032
Alammari MR, Smith PW, de Josselin de Jong E, Higham SM (2013) Quantitative light-induced fluorescence (QLF): a tool for early occlusal dental caries detection and supporting decision making in vivo. J Dent 41:127–132. doi:10.1016/j.jdent.2012.08.013
Landis JR, Koch GG (1977) The measurement of observer agreement for categorical data. Biometrics 33:159–174. doi:10.2307/2529310
Ewoldsen N, Koka S (2010) There are no clearly superior methods for diagnosing, predicting, and noninvasively treating dental caries. J Evid Based Dent Pract 10:16–17. doi:10.1016/j.jebdp.2009.11.008
Maia AM, Karlsson L, Margulis W, Gomes AS (2011) Evaluation of two imaging techniques: near-infrared transillumination and dental radiographs for the detection of early approximal enamel caries. Dentomaxillofac Radiol 40:429–433. doi:10.1259/dmfr/32702114
Pretty IA, Hall AF, Smith PW, Edgar WM, Higham SM (2002) The intra- and inter-examiner reliability of quantitative light-induced fluorescence (QLF) analyses. Br Dent J 193:105–109. doi:10.1038/sj.bdj.4801496
Kühnisch J, Bücher K, Henschel V, Hickel R (2007) Reproducibility of DIAGNOdent 2095 and DIAGNOdent Pen measurements: results from an in vitro study on occlusal sites. Eur J Oral Sci 15:206–211. doi:10.1111/j.1600-0722.2007.00441.x
Kühnisch J, Heinrich-Weltzien R (2004) Quantitative light-induced fluorescence (QLF)—a literature review. Int J Comput Dent 7:325–338
Lussi A, Hellwig E (2006) Performance of a new laser fluorescence device for the detection of occlusal caries in vitro. J Dent 34:467–471. doi:10.1016/j.jdent.2005.11.002
Bamzahim M, Shi XQ, Angmar-Månsson B (2002) Occlusal caries detection and quantification by DIAGNOdent and Electronic Caries Monitor: in vitro comparison. Acta Odontol Scand 60:360–364. doi:10.1080/000163502762667397
Hintze H, Wenzel A, Frydenberg M (2002) Accuracy of caries detection with four storage phosphor systems and E-speed radiographs. Dentomaxillofac Radiol 31:170–175. doi:10.1038/sj/dmfr/4600686
Dias da Silva PR, Martins Marques M, Steagall W Jr, Medeiros Mendes F, Lascala CA (2010) Accuracy of direct digital radiography for detecting occlusal caries in primary teeth compared with conventional radiography and visual inspection: an in vitro study. Dentomaxillofac Radiol 39:362–367. doi:10.1259/dmfr/22865872
Kayipmaz S, Sezgin ÖS, Saricaoğlu ST, Çan G (2011) An in vitro comparison of diagnostic abilities of conventional radiography, storage phosphor, and cone beam computed tomography to determine occlusal and approximal caries. Eur J Radiol 80:478–482. doi:10.1016/j.ejrad.2010.09.011
Teo TK, Ashley PF, Louca C (2014) An in vivo and in vitro investigation of the use of ICDAS, DIAGNOdent pen and CarieScan PRO for the detection and assessment of occlusal caries in primary molar teeth. Clin Oral Investig 18:737–744. doi:10.1007/s00784-013-1021-4
Jones R, Fried D (2002) Attenuation of 1310-nm and 1550-nm laser light through sound dental enamel. Proc SPIE 4610:187–190. doi:10.1117/12.469324
Salsone S, Taylor A, Gomez J, Pretty I, Ellwood R, Dickinson M, Lombardo G, Zakian C (2012) Histological validation of near-infrared reflectance multispectral imaging technique for caries detection and quantification. J Biomed Opt 17:076009. doi:10.1117/1.JBO.17.7.076009
Fried D, Staninec M, Darling CL, Lee C, Kang H, Chan KH (2011) In vivo near-IR imaging of occlusal lesions at 1310-nm. Proc Soc Photo Opt Instrum Eng 7884(78840B). doi:10.1117/12.878888
The authors thank São Paulo Research Foundation (FAPESP) for its financial support (PIPE-FAPESP 2010/50479-2), and DMC Equipamentos for the use of its prototype.
Conflict of interest
The authors certify that they have no commercial or associative interest that represents a conflict of interest in connection with the manuscript.
About this article
Cite this article
Bussaneli, D.G., Restrepo, M., Boldieri, T. et al. Assessment of a new infrared laser transillumination technology (808 nm) for the detection of occlusal caries—an in vitro study. Lasers Med Sci 30, 1873–1879 (2015). https://doi.org/10.1007/s10103-014-1704-3