Nephron-sparing surgery was performed in a porcine model with a 1.92-μm fiber laser dissection device in comparison to a standard high-frequency dissection device. In nine pigs, general anesthesia and a median laparotomy were performed to expose both kidneys. On six kidneys (three HF and three laser) a partial renal parenchyma resection of the lower pole without opening of the renal pelvis was performed (group A). On 12 kidneys (four HF and eight laser), a hemi nephrectomy with opening of the renal pelvis was performed (group B). Total resection time including hemostasis of the remaining tissue was 501 ± 394 s in group “A-laser ” vs. 176 ± 139 s in group “A-HF”. For the group “B”, the total resection time was 1174 ± 501 s (B laser) vs. 960 ± 407 s (B-HF). Blood loss was 28 ± 22 ml in group “A laser ” vs. 15 ± 15 ml in group “A-HF”. In group “B”, the blood loss was 98 ± 73 ml (B laser) vs. 137 ± 118 ml (B-HF). No ischemic time for the kidneys was needed in group “A” for both dissection devices. In group “B”, ischemia of the kidneys was performed three times during the eight laser procedures (420 ± 60 s) and only once at the four HF procedures (1,260 s). Healing process was observed over 4–6 weeks, survival rate was 100%, and no renal fistulas were found after the survival period. In conclusion, no significant differences were found between the compared dissection devices. However, the laser system with the flexible transmission fiber may have an advantage for a laparoscopic approach by steerable instruments.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Price excludes VAT (USA)
Tax calculation will be finalised during checkout.
Jemal A, Siegel R, Ward E, Hao Y, Xu J, Thun MJ (2009) Cancer statistics, 2009. CA Cancer J Clin 59:225–249
Kunkle DA, Egleston BL, Uzzo RG (2008) Excise, ablate or observe: the small renal mass dilemma—a meta-analysis and review. J Urol 179:1227–1233, discussion 1233-4
Thompson RH, Frank I, Lohse CM, Saad IR, Fergany A, Zincke H, Leibovich BC, Blute ML, Novick AC (2007) The impact of ischemia time during open nephron-sparing surgery on solitary kidneys: a multi-institutional study. J Urol 177:471–476
Zimmermann R, Janetschek G (2008) Complications of laparoscopic partial nephrectomy. World J Urol 26:531–537
Benway BM, Bhayani SB, Rogers CG, Dulabon LM, Patel MN, Lipkin M, Wang AJ, Stifelman MD (2009) Robot assisted partial nephrectomy versus laparoscopic partial nephrectomy for renal tumors: a multi-institutional analysis of perioperative outcomes. J Urol 182:866–872
Breda A, Finelli A, Janetschek G, Porpiglia F, Montorsi F (2009) Complications of laparoscopic surgery for renal masses: prevention, management, and comparison with the open experience. Eur Urol 55:836–850
Msezane LP, Katz MH, Gofrit ON, Shalhav AL, Zorn KC (2008) Hemostatic agents and instruments in laparoscopic renal surgery. J Endourol 22:403–408
Wenk S, Fuerst S, Danicke V, Theisen-Kunde D (2007) "Design and Technical Concept of a Tm Laser Scalpel for clinical investigation based on a 60 W, 1.92 μm Tm fibre laser system," presented at 3. Remagener Physiktage, Remagen
Theisen-Kunde D, Tedsen S, Danicke V, Keller R, Brinkmann R (2007) "Laser-Scalpel for kidney resection based on 1,94 μm fibre laser system," presented at Advances in Medical Engineering
Theisen-Kunde D, Tedsen S, Herrmann K, Danicke V, Brinkmann R (2007) "Partial kidney resection based on 1.94 μm fibre laser system," presented at SPIE-OSA, Munich
Lange B, Brendel T, Hüttmann G (2002) Temperature dependence of light absorption in water at holmium and thulium laser wavelength. Appl Opt 41:5797–5803
Theisen-Kunde D, Danicke V, Wendt M, Brinkmann R (2008) "Temperature dependence of water absorption for wavelengths at 1920 nm and 1940 nm," presented at ECIFMBE, Antverpen
Jackman SV, Cadeddu JA, Chen RN, Micali S, Bishoff JT, Lee BR, Moore RG, Kavoussi LR (1998) Utility of the harmonic scalpel for laparoscopic partial nephrectomy. J Endourol 12:441–444
Honeck P, Wendt-Nordahl G, Bolenz C, Peters T, Weiss C, Alken P, Michel MS, Hacker A (2008) Hemostatic properties of four devices for partial nephrectomy: a comparative ex vivo study. J Endourol 22:1071–1076
Lotan Y, Gettman MT, Ogan K, Baker LA, Cadeddu JA (2002) Clinical use of the holmium: YAG laser in laparoscopic partial nephrectomy. J Endourol 16:289–292
Lotan Y, Gettman MT, Lindberg G, Napper CA, Hoopman J, Pearle MS, Cadeddu JA (2004) Laparoscopic partial nephrectomy using holmium laser in a porcine model. J Laparoendosc Surg 8:51–55
Ogan K, Wilhelm D, Lindberg G, Lotan Y, Napper C, Hoopman J, Pearle MS, Cadeddu JA (2002) Laparoscopic partial nephrectomy with a diode laser: porcine results. J Endourol 16:749–753
Hindley RG, Barber NJ, Walsh K, Petersen A, Poulsen J, Muir GH (2006) Laparoscopic partial nephrectomy using the potassium titanyl phosphate laser in a porcine model. Urology 67:1079–1083
Liu M, Rajbabu K, Zhu G, Petersen A, Muir GH, Poulson J (2006) Laparoscopic partial nephrectomy with saline-irrigated KTP laser in a porcine model. J Endourol 20:1096–1100
Bui MH, Breda A, Gui D, Said J, Schulam P (2007) Less smoke and minimal tissue carbonization using a thulium laser for laparoscopic partial nephrectomy without hilar clamping in a porcine model. J Endourol 21:1107–1111
Gruschwitz T, Schubert J, Wunderlich H (2008) Laser-supported partial nephrectomy for renal cell carcinoma. Urologe A 47:1173–1174
Mattioli S, Munoz R, Recasens R, Berbegal C, Teichmann H (2008) What does Revolix laser contribute to partial nephrectomy? Arch Esp Urol 61:1126–1129
Theisen-Kunde D, Danicke V, Brinkmann R (2009) "Comparison of two CW Infrared Laser Systems emitting wavelengths at 1.92 ìm and 2.01 ìm for tissue dissection in liver surgery," presented at IFMBE, Munich
Anderson JK, Baker MR, Lindberg G, Cadeddu JA (2007) Large-volume laparoscopic partial nephrectomy using the potassium-titanyl-phosphate (KTP) laser in a survival porcine model. Eur Urol 51:749–754
Benderev TV, Schaeffer AJ (1985) Efficacy and safety of the Nd:YAG laser in canine partial nephrectomy. J Urol 133:1108–1111
This study was supported by the Land Schleswig-Holstein, Germany (ZIP-Program 2005) and StarMedTec GmbH, Starnberg, Germany.
Rights and permissions
About this article
Cite this article
Theisen-Kunde, D., Tedsen, S., Doehn, C. et al. Comparison between a 1.92-μm fiber laser and a standard HF-dissection device for nephron-sparing kidney resection in a porcine in vivo study. Lasers Med Sci 26, 509–514 (2011). https://doi.org/10.1007/s10103-010-0873-y
- Renal cell carcinoma
- Nephron-sparing surgery
- Tumor ablation