Outflow graft anastomosis site design could be correlated to aortic valve regurgitation under left ventricular assist device support
- 67 Downloads
Aortic valve regurgitation (AR) is a critical complication during circulatory support with a left ventricular assist device (LVAD). The time-course of AR and related factors, including outflow graft anastomosis site design, were investigated. Twenty-three patients who had continuous-flow LVAD implantation and were supported for more than 6 months were investigated. AR grade (none, 0; trivial, 0.5; mild, 1; mild-moderate, 1.5; moderate, 2; moderate-severe, 2.5; severe, 3) and aortic valve opening were evaluated with echocardiography. Computed tomography was performed to all the patients postoperatively. The angle of the outflow graft to the aorta (O-A angle, parallel 0; tangent 90°, 0–180°), aortic diameter at the anastomosis site, sino-tubular junction (STJ) diameter, distance between the STJ and the anastomosis site, and distance between the anastomosis site and the brachiocephalic artery were measured. The patients’ age was 38 ± 11 years. Support duration was 686 ± 354 days. Mean AR grade after continuous-flow LVAD implantation was increased to around mild and was maintained thereafter. No patient needed any intervention to the aortic valve. The aortic valves of 82.6% of patients were closed continuously. The O-A angle (83 ± 14) was positively correlated with maximum AR grade (p = 0.0095). The O-A angle was significantly smaller in patients with maximum AR grade of 1 or less (77 ± 9°) than in those with 1.5 or greater (94 ± 15°, p = 0.021). The other CT measurements had no correlation with AR grade. In conclusion, the O-A angle was correlated with AR grade progression. The O-A angle appears to be one of the important factors related to AR under continuous-flow LVAD support.
KeywordsAortic valve regurgitation Left ventricular assist device Heart failure Continuous-flow
Compliance with ethical standards
Conflict of interest
Kenji Yamazaki is a remunerated consultant of SunMedical Technology Research Co. All other authors declare no conflict of interest.
- 3.Hatano M, Kinugawa K, Shiga T, Kato N, Endo M, Hisagi M, et al. Less frequent opening of the aortic valve and a continuous flow pump are risk factors for postoperative onset of aortic insufficiency in patients with a left ventricular assist device. Circ J. 2011;75:1147–55.CrossRefPubMedGoogle Scholar
- 6.Benk C, Mauch A, Beyersdorf F, Klemm R, Russe M, Blanke P, et al. Effect of cannula position in the thoracic aorta with continuous left ventricular support: four-dimensional flow-sensitive magnetic resonance imaging in an in vitro model. Eur J Cardio-thoracic Surg. 2013;44:551–8.CrossRefGoogle Scholar