The aim of this study was to investigate the in-vitro performance and fracture resistance of a temporary computer-aided designed and computer-aided manufactured polymethylmethacrylate (CAD/CAM-PMMA) material as implant or tooth-supported single crown with respect to the clinical procedure (permanently bonded/temporarily cemented).
Materials and methods
Sixty-four crowns were fabricated on implants or human molar teeth simulating (a) labside procedure on prefabricated titanium-bonding base ([TiBase] implant crown bonded in laboratory, screwed chairside), (b) labside procedure ([LAB] standard abutment and implant crown bonded in laboratory, screwed chairside), (c) chairside procedure ([CHAIR] implant crown bonded to abutment), and (d) reference ([TOOTH] crowns luted on prepared human teeth). Crowns were made of a CAD/CAM-PMMA temporary material (TelioCAD, Ivoclar-Vivadent). For investigating the influence of fixation, half of the crowns were permanently (P) or temporarily (T) bonded. Combined thermal cycling and mechanical loading (TCML) was performed simulating a 5-year clinical situation. Fracture force was determined. Data were statistically analyzed (Kolmogorov-Smirnov test, one-way ANOVA; post hoc Bonferroni, α = 0.05).
All restorations survived TCML without visible failures. Fracture results varied between 3034.3 (Tooth-P) and 1602.9 N (Tooth-T) [TOOTH], 1510.5 (TiBase-P) and 963.6 N (TiBase-T) [TiBase], 2691.1 (LAB-P) and 2064.5 N (LAB-T) [LAB], and 1609.4 (Chair-P) and 1253.0 N (Chair-T) [CHAIR]. Tested groups showed significantly (p < 0.001) different fracture values. Failure pattern was characterized by fractures in mesial-distal, buccal-oral, or mixed (mesial-distal/buccal-oral) directions, with differences for the individual groups.
Temporary CAD/CAM crowns showed no different in-vitro performance but provided different fracture results that depended on cementation, screw channel, and type of abutment.
All bonded and screwed PMMA crowns were in a range where clinical application seems not restricted.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Gratton DG, Aquilino SA (2004) Interim restorations. Dent Clin N Am 48:487–497
Rayyan MM, Aboushelib M, Sayed NM, Ibrahim A, Jimbo R (2015) Comparison of interim restorations fabricated by CAD/CAM with those fabricated manually. J Prosthet Dent 114:414–419
Lodding DW (1997) Long-term esthetic provisional restorations in dentistry. Curr Opin Cosmet Dent 4:16–21
Wiegand A, Stucki L, Hoffmann R, Attin T, Stawarczyk B (2015) Repairability of CAD/CAM high-density PMMA- and composite-based polymers. Clin Oral Investig 19:2007–2013
Keul C, Müller-Hahl M, Eichberger M, Liebermann A, Roos M, Edelhoff D, Stawarczyk B (2014) Impact of different adhesives on work of adhesion between CAD/CAM polymers and resin composite cements. J Dent 42:1105–1114
Attia A, Abdelaziz KM, Freitag S, Kern M (2006) Fracture load of composite resin and feldspathic all-ceramic CAD/CAM crowns. J Prosthet Dent 95:117–123
Wimmer T, Ender A, Roos M, Stawarczyk B (2013) Fracture load of milled polymeric fixed dental prostheses as a function of connector cross-sectional areas. J Prosthet Dent 110:288–295
Stawarczyk B, Özcan M, Trottmann A, Schmutz F, Roos M, Hämmerle C (2013) Two-body wear rate of CAD/CAM resin blocks and their enamel antagonists. J Prosthet Dent 109:325–332
Seiss M, Nitz S, Kleinsasser N, Buters JT, Behrendt H, Hickel R, Reichl FX (2007) Identification of 2,3-epoxymethacrylic acid as an intermediate in the metabolism of dental materials in human liver microsomes. Dent Mater 23:9–16
Bähr N, Keul C, Edelhoff D, Eichberger M, Roos M, Gernet W, Stawarczyk B (2013) Effect of different adhesives combined with two resin composite cements on shear bond strength to polymeric CAD/CAM materials. Dent Mater J 32:492–501
Keul C, Kohen D, Eichberger M, Roos M, Gernet W, Stawarczyk B (2015) The effect of different pretreatment methods of PMMA-based crowns on the long-term tensile bond strength to dentin abutments. Clin Oral Investig 19:35–43
Stawarczyk B, Basler T, Ender A, Roos M, Ozcan M, Hämmerle C (2012) Effect of surface conditioning with airborne-particle abrasion on the tensile strength of polymeric CAD/CAM crowns luted with self-adhesive and conventional resin cements. J Prosthet Dent 107:94–101
Güth JF, Almeida E, Silva JS, Ramberger M, Beuer F, Edelhoff D (2012) Treatment concept with CAD/CAM-fabricated high-density polymer temporary restorations. J Esthet Restor Dent 24:310–318
Güth JF, Almeida E, Silva JS, Beuer FF, Edelhoff D (2012) Enhancing the predictability of complex rehabilitation with a removable CAD/CAM-fabricated long-term provisional prosthesis: a clinical report. J Prosthet Dent 107:1–6
Hebel KS, Gajjar RC (1997) Cement-retained versus screw-retained implant restorations: achieving optimal occlusion and esthetics in implant dentistry. J Prosthet Dent 77:28–35
Weber HP, Kim DM, Ng MW, Hwang JW, Fiorellini JP (2006) Peri-implant soft-tissue health surrounding cement- and screw-retained implant restorations: a multi-center, 3-year prospective study. Clin Oral Implants Res 17:375–379
Chu KM, Tredwin CJ, Setchell DJ, Hems E (2005) Effect of screw hole filling on retention of implant crowns. Eur J Prosthodont Restor Dent 13:154–158
Lewis S (1995) Anterior single-tooth implant restorations. Int J Periodontics Restorative Dent 15:30–41
Saba S (1997) Anatomically correct soft tissue profiles using fixed detachable provisional implant restorations. J Can Dent Assoc 63(767–768):770
Breeding LC, Dixon DL (1996) Transfer of gingival contours to a master cast. J Prosthet Dent 75:341–343
Neale D, Chee WW (1994) Development of implant soft tissue emergence profile: a technique. J Prosthet Dent 71:364–368
Capp NJ (1985) The diagnostic use of provisional restorations. Restor Dent 1(92):94–98
Moscovitch MS, Saba S (1996) The use of a provisional restoration in implant dentistry: a clinical report. Int J Oral Maxillofac Implants 11:395–399
Rosentritt M, Hmaidouch R, Behr M, Handel G, Schneider-Feyrer S (2011) Fracture resistance of zirconia FPDs with adhesive bonding versus conventional cementation. Int J Prosthodont 24:168–171
Rosentritt M, Behr M, Gebhard R, Handel G (2006) Influence of stress simulation parameters on the fracture strength of all-ceramic fixed-partial dentures. Dent Mater 22:176–182
Rosentritt M, Behr M, van der Zel JM, Feilzer AJ (2009) Approach for valuating the influence of laboratory simulation. Dent Mater 25:348–352
Rosentritt M, Siavikis G, Behr M, Kolbeck C, Handel G (2008) Approach for valuating the significance of laboratory simulation. J Dent 36:1048–1053
Menini M, Conserva E, Tealdo T, Bevilacqua M, Pera F, Signori A, Pera P (2013) Shock absorption capacity of restorative materials for dental implant prostheses: an in vitro study. Int J Prosthodont 26:549–556
Pjetursson BE, Brägger U, Lang NP, Zwahlen M (2007) Comparison of survival and complication rates of tooth-supported fixed dental prostheses (FDPs) and implant-supported FDPs and single crowns (SCs). Clin Oral Implants Res 18(Suppl 3):97–113
Shembish FA, Tong H, Kaizer M, Janal MN, Thompson VP, Opdam NJ, Zhang Y (2016) Fatigue resistance of CAD/CAM resin composite molar crowns. Dent Mater 32:499–509
de Kok P, Kleverlaan CJ, de Jager N, Kuijs R, Feilzer AJ (2015) Mechanical performance of implant-supported posterior crowns. J Prosthet Dent 114:59–66
Conserva E, Menini M, Tealdo T, Bevilacqua M, Ravera G, Pera F, Pera P (2009) The use of a masticatory robot to analyze the shock absorption capacity of different restorative materials for prosthetic implants: a preliminary report. Int J Prosthodont 22:53–55
Jung RE, Pjetursson BE, Glauser R, Zembic A, Zwahlen M, Lang NP (2008) A systematic review of the 5-year survival and complication rates of implant-supported single crowns. Clin Oral Implants Res 19:119–130
Bijjargi S, Chowdhary R (2013) Stress dissipation in the bone through various crown materials of dental implant restoration: a 2-D finite element analysis. J Investig Clin Dent 4:172–177
Trulsson M, Gunne HS (1998) Food-holding and -biting behavior in human subjects lacking periodontal receptors. J Dent Res 77:574–582
Silva GC, Cornacchia TM, de Magalhães CS, Bueno AC, Moreira AN (2014) Biomechanical evaluation of screw- and cement-retained implant-supported prostheses: a nonlinear finite element analysis. J Prosthet Dent 112:1479–1488
Wittneben JG, Millen C, Bragger U (2014) Clinical performance of screw- versus cement-retained fixed implant-supported reconstructions - a systematic review. Int J Oral Maxillofac Implants 29(Suppl):84–98
Torrado E, Ercoli C, Al Mardini M, Graser GN, Tallents RH, Cordaro L (2004) A comparison of the porcelain fracture resistance of screw-retained and cement-retained implant-supported metal-ceramic crowns. J Prosthet Dent 91:532–537
Gressler May L, Kelly JR, Bottino MA, Hill T (2015) Influence of the resin cement thickness on the fatigue failure loads of CAD/CAM feldspathic crowns. Dent Mater 31:895–900
Rosentritt M, Hahnel S, Engelhardt F, Behr M, Preis V (2016) In vitro performance and fracture resistance of CAD/CAM-fabricated implant supported molar crowns. Clin Oral Investig. doi:10.1007/s00784-016-1898-9
Preis V, Kammermeier A, Handel G, Rosentritt M (2016) In vitro performance of two-piece zirconia implant systems for anterior application. Dent Mater 32:765–774
Varga S, Spalj S, Lapter Varga M, Anic Milosevic S, Mestrovic S, Slaj M (2011) Maximum voluntary molar bite force in subjects with normal occlusion. Eur J Orthod 3:427–433
We would like to thank the manufacturers for providing the materials.
Conflict of interest
Prof. Dr. Martin Rosentritt declares that he has third-party research projects with all dental companies.
Dr. Philipp Raab declares that he has no conflict of interest.
Prof. Dr. Sebastian Hahnel declares that he has no conflict of interest.
Dr. Matthias Stöckle declares that he has no conflict of interest.
PD Dr. Verena Preis declares that she has no conflict of interest.
This article does not contain any studies with human participants or animals performed by any of the authors.
For this type of study, formal consent is not required.
About this article
Cite this article
Rosentritt, M., Raab, P., Hahnel, S. et al. In-vitro performance of CAD/CAM-fabricated implant-supported temporary crowns. Clin Oral Invest 21, 2581–2587 (2017). https://doi.org/10.1007/s00784-017-2057-7
- CAD/CAM composites
- PMMA, temporary
- Provisional, chewing simulation
- Implant crown
- Fracture resistance