Pedicle screw insertion with patient-specific 3D-printed guides based on low-dose CT scan is more accurate than free-hand technique in spine deformity patients: a prospective, randomized clinical trial
Screw misplacement incidence can be as high as 15–30% in spine deformity surgery, with possible devastating consequences. Some technical solutions to prevent misplacement require expensive devices. MySpineTM comprises a low-dose CT scan of the patient’s spine to build a virtual model of the spine to plan the screw trajectories and a 3D-printed patient-specific guide system to prepare the screw trajectories and to implant the screws in the vertebrae in order to increase reproducibility and safety of the implants. The aim of this open-label, single-center, prospective randomized clinical trial with independent evaluation of outcomes was to compare the accuracy of free-hand insertion of pedicle screws to MySpineTM 3D-printed patient-specific guides.
Twenty-nine patients undergoing surgical correction for spinal deformity were randomized to Group A (pedicle screws implantation with MySpineTM) or Group B (free-hand implantation). Group A received 297 pedicle screws, and Group B 243 screws. Forty-three screws in Group A crossed over to free-hand implantation. Screw position was graded according to Gertzbein in grades 0, A, B or C, with grades 0 or A considered as “safe area.” Total fluoroscopy dose and time were compared in six patients of each group.
Comparing the two study groups, we observed a statistically significant difference between the two groups (p < 0.05), with 96.1% of screws in the “safe area” in Group A versus a 82.9% in Group B. Group-A patients had a mean effective dose of 0.23 mSv compared to 0.82 mSv in Group B. Patient-specific, 3D-printed pedicle screw guides increase safety in a wide spectrum of deformity conditions. In addition, the total radiation dose is reduced, even considering the need of a low-dose preoperative CT for surgical planning.
Level of evidence
KeywordsScoliosis Navigation 3D printing Accuracy Spine deformity
This study was financed with an unrestricted research Grant of 22,975 Euros from Medacta SA, CH.
Compliance with ethical standards
Conflict of interest
Authors RC, PB, AZ, MD and CL are consultants for Medacta SA.
- 2.Sciubba DM, Yurter A, Smith JS, Kelly MP, Scheer JK, Goodwin CR, Lafage V, Hart RA, Bess S, Kebaish K, Schwab F, Shaffrey CI, Ames CP, International Spine Study Group (ISSG) (2015) A comprehensive review of complication rates after surgery for adult deformity: a reference for informed consent. Spine Deform 3(6):575–594CrossRefGoogle Scholar
- 13.Hu Y, Yuan ZS, Spiker WR, Dong WX, Sun XY, Yuan JB, Zhang J, Zhu B (2016) A comparative study on the accuracy of pedicle screw placement assisted by personalized rapid prototyping template between pre- and post-operation in patients with relatively normal mid-upper thoracic spine. Eur Spine J 25(6):1706–1715CrossRefGoogle Scholar
- 17.Radermacher K, Bliem R, Hennecke C, Staudte HW, Rau G (1996) A desktop image processing system for computer-assisted orthopedic surgery (DISOS). Stud Health Technol Inform 29:675–680Google Scholar