Total mesorectal excision using a soft and flexible robotic arm: a feasibility study in cadaver models
- 556 Downloads
Sponsored by the European Commission, the FP7 STIFF-FLOP project aimed at developing a STIFFness controllable Flexible and Learn-able manipulator for surgical operations, in order to overcome the current limitations of rigid-link robotic technology. Herein, we describe the first cadaveric series of total mesorectal excision (TME) using a soft and flexible robotic arm for optic vision in a cadaver model.
TME assisted by the STIFF-FLOP robotic optics was successfully performed in two embalmed male human cadavers. The soft and flexible optic prototype consisted of two modules, each measuring 60 mm in length and 14.3 mm in maximum outer diameter. The robot was attached to a rigid shaft connected to an anthropomorphic manipulator robot arm with six degrees of freedom. The controller device was equipped with two joysticks. The cadavers (BMI 25 and 28 kg/m2) were prepared according to the Thiel embalming method. The procedure was performed using three standard laparoscopic instruments for traction and dissection, with the aid of a 30° rigid optics in the rear for documentation.
Following mobilization of the left colonic flexure and division of the inferior mesenteric vessels, TME was completed down to the pelvic floor. The STIFF-FLOP robotic optic arm seemed to acquire superior angles of vision of the surgical field in the pelvis, resulting in an intact mesorectum in both cases. Completion times of the procedures were 165 and 145 min, respectively. No intraoperative complications occurred. No technical failures were registered.
The STIFF-FLOP soft and flexible robotic optic arm proved effective in assisting a laparoscopic TME in human cadavers, with a superior field of vision compared to the standard laparoscopic vision, especially low in the pelvis. The introduction of soft and flexible robotic devices may aid in overcoming the technical challenges of difficult laparoscopic procedures based on standard rigid instruments.
KeywordsSoft Flexible Robotic surgery Total mesorectal excision
The video shows the first cadaveric series of Total Mesorectal Excision (TME) using a soft and flexible robotic arm for optic vision in a cadaver model. (MP4 1035811 kb)
- 2.Guillou PJ, Quirke P, Thorpe H, Walker J, Jayne DG, Smith AM, Heath RM, Brown JM, MRC CLASICC trial group (2005) Short-term endpoints of conventional versus laparoscopic-assisted surgery in patients with colorectal cancer (MRC CLASICC trial): multicentre, randomised controlled trial. Lancet 365:1718–1726CrossRefPubMedGoogle Scholar
- 4.Veldkamp R, Kuhry E, Hop WC, Jeekel J, Kazemier G, Bonjer HJ, Haglind E, Påhlman L, Cuesta MA, Msika S, Morino M, Lacy AM, Colon Cancer Laparoscopic or Open Resection Study Group (COLOR) (2005) Laparoscopic surgery versus open surgery for colon cancer: short-term outcomes of a randomised trial. Lancet Oncol 6:477–484CrossRefPubMedGoogle Scholar
- 7.van der Pas MH, Haglind E, Cuesta MA, Fürst A, Lacy AM, Hop WC, Bonjer HJ (2013) Colorectal cancer laparoscopic or open resection II (COLOR II) study group. Laparoscopic versus open surgery for rectal cancer (COLOR II): short-term outcomes of a randomised, phase 3 trial. Lancet Oncol 14(3):210–218CrossRefPubMedGoogle Scholar
- 8.Bonjer HJ, Deijen CL, Abis GA, Cuesta MA, van der Pas MH, de Lange-de Klerk ES, Lacy AM, Bemelman WA, Andersson J, Angenete E, Rosenberg J, Fuerst A, Haglind E, COLOR II Study Group (2015) A randomized trial of laparoscopic versus open surgery for rectal cancer. N Engl J Med 372(14):1324–1332CrossRefPubMedGoogle Scholar
- 9.Fleshman J, Branda M, Sargent DJ, Boller AM, George V, Abbas M, Peters WR Jr, Maun D, Chang G, Herline A, Fichera A, Mutch M, Wexner S, Whiteford M, Marks J, Birnbaum E, Margolin D, Larson D, Marcello P, Posner M, Read T, Monson J, Wren SM, Pisters PW, Nelson H (2015) Effect of laparoscopic-assisted resection versus open resection of stage II or III rectal cancer on pathologic outcomes: the ACOSOG Z6051 randomized clinical trial. JAMA 314(13):1346–1355CrossRefPubMedPubMedCentralGoogle Scholar
- 10.Stevenson AR, Solomon MJ, Lumley JW, Hewett P, Clouston AD, Gebski VJ, Davies L, Wilson K, Hague W, Simes J, ALaCaRT Investigators (2015) Effect of laparoscopic-assisted resection versus open resection on pathological outcomes in rectal cancer: the ALaCaRT randomized clinical trial. JAMA 314(13):1356–1363CrossRefPubMedGoogle Scholar
- 15.Jayne D (2015) MRC/EME ROLARR trial; the first results. In: 23rd international congress of the European association for endoscopic surgery (E.A.E.S.), Bucharest June 3–June 6Google Scholar
- 16.Jiang A, Ataollahi A, Althoefer K, Dasgupta P, Nanayakkara T (2012) A variable stiffness joint by granular jamming. In: ASME 2012 international design engineering technical conferences and computers and information in engineering conference, pp 267–275Google Scholar
- 17.Jiang A, Xynogalas G, Dasgupta P, Althoefer K, Nanayakkara T (2012) Design of a variable stiffness flexible manipulator with composite granular jamming and membrane coupling. In: IEEE/RSJ international conference on intelligent robots and systems (IROS), pp 2922–2927Google Scholar
- 18.Stilli A, Wurdemann HA, Althoefer K (2014) Shrinkable, stiffness-controllable soft manipulator based on a bio-inspired antagonistic actuation principle. In: IEEE/RSJ international conference on intelligent robots and systems (IROS), pp 2476–2481Google Scholar
- 19.Degani A, Choset H, Wolf A, Zenati MA (2006) Highly articulated robotic probe for minimally invasive surgery. In: IEEE international conference on robotics and automation proceedings, pp 4167–4172Google Scholar
- 20.Wei W, Kai X, Simaan N (2006) A compact two-armed slave manipulator for minimally invasive surgery of the throat. In: IEEE/RAS-EMBS international conference on biomedical robotics and biomechatronics, pp 769–774Google Scholar
- 21.Bajo A, Dharamsi LM, Netterville JL, Garrett CG, Simaan N (2013) Robotic-assisted micro-surgery of the throat: the trans-nasal approach. In: IEEE international conference on robotics and automation (ICRA), pp 232–238Google Scholar
- 24.Bajo A, Goldman RE, Long W, Fowler D, Simaan N (2012) Integration and preliminary evaluation of an insertable robotic effectors platform for single port access surgery. In: IEEE international conference on robotics and automation (ICRA), pp 3381–3387Google Scholar
- 25.Shang J, Noonan DP, Payne C, Clark J, Sodergren MH, Darzi A, Yang GZ (2011) An articulated universal joint based flexible access robot for minimally invasive surgery. In: IEEE international conference robotics and automation (ICRA), pp 1147–1152Google Scholar
- 28.Loeve A, Breedveld P, Dankelman J (2010). Scopes too flexible…and too stiff. Pulse, IEEE 1(3):26–41Google Scholar
- 31.Fras J, Czarnowski J, Macias M, Glowka J, Cianchetti M, Menciassi A (2015) New STIFF-FLOP module construction idea for improved actuation and sensing. In: IEEE international conference on robotics and automation, 2015 SeattleGoogle Scholar
- 32.Małota Z, Nawrat Z, Sadowski W (2014) Minimally invasive surgery simulation. Eng Biomater 126(17):2–11Google Scholar
- 33.Miller K, Clavel R (1992) The lagrange-based model of delta-4 robot dynamics. Robotersysteme 8:49–54Google Scholar