Eyelid motion analysis can provide important information about ophthalmic, neurologic, and systemic diseases. Routine assessment of eyelid function is currently based mainly on clinical examination estimating Levator Function and static palpebral fissure measurements. Most clinical tools developed to date are cumbersome expensive and difficult to operate. Currently there is no widely available, affordable device providing user friendly precision based evaluation of eyelid kinematics. Our goal is to develop a novel device for evaluation of eyelid kinematics providing rapid defined diagnosis of diseases involving eyelid movement.
A real-time prototype eyelid motion monitoring system was designed based on magnetic field sensors detecting movement of a tiny magnet located on the upper eyelid. Motion is recorded and analyzed using specially developed hardware and software, respectively, enabling both real-time and off-line data presentation. The Eyelid Motion Monitor correlates between blinking characteristics of eyelid movement and the output voltages produced by the system. Blink detection is defined as peak in voltage, caused by eyelid closure or opening. The device was tested on 20 healthy volunteers with normal clinical blinking patterns.
The Eyelid Motion Monitor succeeded in detecting full blink motion. The system easily extracts different parameters of eyelid kinetics.
An inexpensive prototype novel device was developed for monitoring and analyzing eyelid motion characteristics, including the inter-blink interval, eye closing/opening duration and entire blink duration. The device should allow early objective non- invasive diagnosis and follow-up of disease progression. It could be of great potential value in many ophthalmic, neurologic, and systemic diseases.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Guitton D, Simard R, Codère F (1991) Upper eyelid movements measured with a search coil during blinks and vertical saccades. Invest Ophthalmol Vis Sci 32(13):3298–3305
van Essen TA, van der Giessen RS, Koekkoek SK et al (2010) Anti-malaria drug mefloquine induces motor learning deficits in humans. Front Neurosci 4:191. doi:10.3389/fnins.00191
Smit AE, van der Geest JN, Vellema M et al (2008) Savings and extinction of conditioned eyeblink responses in fragile X syndrome. Genes Brain Behav 7(7):770–777. doi:10.1111/j.1601-183X.2008.00417.x
Koekkoek SK, Yamaguchi K, Milojkovic BA et al (2005) Deletion of FMR1 in Purkinje cells enhances parallel fiber LTD, enlarges spines, and attenuates cerebellar eyelid conditioning in fragile X syndrome. Neuron 47(3):339–352
Garcia DM, Barbosa JC, Pinto CT, Cruz AA (2013) Estimation of spontaneous blinking main sequence in normal subjects and patients with graves’ upper eyelid retraction. Invest Ophthalmol Vis Sci 54(2):1434–1442. doi:10.1167/iovs.12-11452
Bracha V, Nilaweera W, Zenitsky G, Irwin K (2003) Video recording system for the measurement of eyelid movements during classical conditioning of the eyeblink response in the rabbit. J Neurosci Methods 125(1–2):173–181
Malbouisson JM, E Cruz AA, Messias A, Leite LV, Rios GD (2005) Upper and lower eyelid saccades describe a harmonic oscillator function. Invest Ophthalmol Vis Sci 46(3):857–862
Koekkoek SK, Den Ouden WL, Perry G, Highstein SM, De Zeeuw CI (2002) Monitoring kinetic and frequency-domain properties of eyelid responses in mice with magnetic distance measurement technique. J Neurophysiol 88(4):2124–2133
Gruart A, Schreurs BG, del Toro ED, Delgado-García JM (2000) Kinetic and frequency-domain properties of reflex and conditioned eyelid responses in the rabbit. J Neurophysiol 83(2):836–852
GoPro HERO3+ (2013) User Manual and Specifications, http://cbcdn2.gp-static.com/uploads/product_manual/file/195/UM_H3PlusSilver_ENG_REVB_WEB.pdf. Accessed 20 Apr 2016
Vandermeer S, Amsel A (1952) Work and rest factors in eyelid conditioning. J Exp Psychol 43(4):261–273
Kennard DW, Glaser GH (1964) An analysis of eyelid movements. J Nerv Ment Dis 139:31–48
Evinger C, Shaw MD, Peck CK, Manning KA, Baker R (1984) Blinking and associated eye movements in humans, guinea pigs, and rabbits. J Neurophysiol 52(2):323–339
Collewijn H, van der Steen J, Steinman RM (1985) Human eye movements associated with blinks and prolonged eyelid closure. J Neurophysiol 54(1):11–27
Doane MG (1980) Interactions of eyelids and tears in corneal wetting and the dynamics of the normal human eyeblink. Am J Ophthalmol 89(4):507–516
Hung G, Hsu F, Stark L (1977) Dynamics of the human eyeblink. Am J Optom Physiol Optic 54(10):678–690
Franks CM, Withers WC (1955) Photoelectric recording of eyelid movements. Am J Psychol 68(3):467–471
Berman SM, Bullimore MA, Jacobs RJ, Bailey IL, Gandhi N (1994) An objective measure of discomfort glare. J Illum Eng Soc 23(2):40–49
Ruhhammer J, Herbstritt T, Ruh D et al (2014) Magnetic sensor for arterial distension and blood pressure monitoring. Biomed Microdevices 16(6):815–827. doi:10.1007/s10544-014-9885-x
The research is supported by the High Speed Digital Systems Laboratory (HS-DSL), Technion – Israel Institute of Technology and Emek Medical Center. Special thanks to Alex Kinko from the HS-DSL for guiding the software team (Boaz Blankrot, Leon Karabchevsky) and to Oz Levinkron (Technion Faculty of Medicine) and Adi Orbach (Emek Medical Center) for performing the clinical measurements.
No funding was received for this research.
Conflict of interest
All authors certify that they have no affiliations with or involvement in any organization or entity with any financial interest (such as honoraria; educational grants; participation in speakers’ bureaus; membership, employment, consultancies, stock ownership, or other equity interest; and expert testimony or patent-licensing arrangements), or non-financial interest (such as personal or professional relationships, affiliations, knowledge or beliefs) in the subject matter or materials discussed in this manuscript.
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
Additional informed consent was obtained from all individual participants for whom identifying information is included in this article.
Informed consent was obtained from all individual participants included in the study.
About this article
Cite this article
Hanuka, A., Itzhak, M., Berger, A. et al. A novel eyelid motion monitor. Graefes Arch Clin Exp Ophthalmol 255, 1811–1817 (2017). https://doi.org/10.1007/s00417-017-3712-z
- Eyelid motion
- Blink detection
- Monitor device
- Magnetic sensor