To examine the outcomes with the AdVance XP and AdVance slings in the management of post-prostatectomy incontinence (PPI).
Eighty patients were treated with the AdVance resp. AdVance XP at one tertiary reference centre. The initial patients were implanted with the AdVance male sling with no associated surgery. Following the introduction of the AdVance XP sling, subsequent patients were implanted with the AdVance XP slings. Measurements included: daily pad usage, 24-h pad weight test, International Quality of Life Questionnaire, International Consultation on Incontinence Questionnaire Short Form and the Patient Global Impression of Improvement. Adverse events were recorded.
Follow-up was available for 39 and 41 patients treated with the AdVance and AdVance XP slings, respectively. At a median follow-up of 24.7 months, 69.3 % of patients could be classified as cured or improved in the AdVance group and at a median follow-up of 11.9 months, 90.3 % of AdVance XP-treated patients. At 3-month follow-up, 96.8 and 97.3 % of patients could be classed as cured or improved in the AdVance and AdVance XP groups, respectively (not significant). There were no perioperative complications. Two AdVance and three AdVance XP serious post-operative complications were reported, which were resolved following treatment. QOL was improved in both treatment groups.
Both the AdVance and the AdVance XP slings are effective and safe for the treatment of PPI.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Rehder P, Gozzi C (2007) Transobturator sling suspension for male urinary incontinence including post-radical prostatectomy. Eur Urol 52:860–866
Davies TO, Bepple JL, McCammon KA (2009) Urodynamic changes and initial results of the AdVance male sling. Urology 74:354–357
Soljanik I, Becker AJ, Stief CG, Gozzi C, Bauer RM (2011) Urodynamic parameters after retrourethral transobturator male sling and their influence on outcome. Urology 78:708–712
Rehder P, Mitterberger MJ, Pichler R, Kerschbaumer A, Glodny B (2010) The 1 year outcome of the transobturator retroluminal repositioning sling in the treatment of male stress urinary incontinence. BJU Int 106:1668–1672
McCammon K, Haab F (2011) AdVance® male sling: surgical technique and post-operative patient management. Eur Urol 10(suppl 4):395–400
Bushnell DM, Martin ML, Summers KH, Svihra J, Lionis C, Patrick DL (2005) Quality of Life of women with urinary incontinence: cross-cultural performance of 15 language versions of the I-QOL. Qual Life Res 14:1901–1913
Avery K, Donovan J, Peters TJ, Shaw C, Gotoh M, Abrams P (2004) ICIQ: a brief and robust measure for evaluating the symptoms and impact of urinary incontinence. Neurourol Urodyn 23:322–330
Twiss CO, Fischer MC, Nitti VW (2007) Comparison between reduction in 24-h pad weight, International Consultation on Incontinence-Short Form (ICIQ-SF) Score, International Prostate Symptom Score (IPSS), and Post-Operative Patient Global Impression of Improvement (PGI-I) score in patient evaluation after male perineal sling. Neurourol Urodyn 26:8–13
Dindo D, Demartines N, Clavien PA (2004) Classification of surgical complications: a new proposal with evaluation in a cohort of 6336 patients and results of a survey. Ann Surg 240:205–213
Kennelly MJ, McCammon KA, Knoll LD et al (2010) Two-year follow-up results in a multi-center, prospective trial of the AdVance male sling system for the treatment of postprostatectomy stress urinary incontinence. In: NEAUA 79th annual meeting abstracts; October 21–23, Providence, RI. Abstract #35. http://www.neaua.org/abstracts/2010/35.cgi
Cornu JN, Sèbe P, Ciofu C, Peyrat L, Cussenot O, Haab F (2011) Mid-term evaluation of the transobturator male sling for postprostatectomy incontinence: focus on prognostic factors. BJU Int 108:236–240
Bauer RM, Soljanik I, Füllhase C et al (2011) Mid-term results for the retroluminar transobturator sling suspension for stress urinary incontinence after postprostatectomy. BJU Int 108:94–98
Rehder P, Haab F, Cornu JN, Gozzi C, Bauer RM (2012) Treatment of postprostatectomy male urinary incontinence with the transobturator retroluminal repositioning sling suspension: 3-year follow-up. Eur Urol 62:140–145
Cornel EB, Elzevier HW, Putter H (2010) Can advance transobturator sling suspension cure male urinary postoperative stress incontinence? J Urol 183:1459–1463
Bauer RM, Mayer ME, Gratzke C et al (2009) Prospective evaluation of the functional sling suspension for male postprostatectomy stress urinary incontinence: results after 1 year. Eur Urol 56:928–933
Collado Serra A, Resel Folkersma L, Domínguez-Escrig JL, Gómez-Ferrer A, Rubio-Briones J, Solsona Narbón E (2013) AdVance/AdVance XP transobturator male slings: preoperative degree of incontinence as predictor of surgical outcome. Urology 81:1034–1039
Bauer RM, Mayer ME, May F et al (2010) Complications of the AdVance transobturator male sling in the treatment of male stress urinary incontinence. Urology 75:1494–1498
This study was sponsored by American Medical Systems.
The study has been approved by the appropriate ethics committee and has therefore been performed in accordance with the ethical standards laid down in the 1964 Declaration of Helsinki and its later amendments. All patients gave their informed consent prior to their inclusion in the study.
Conflict of interest
Ricarda M. Bauer declares lectures, consultancy work and participation in clinical trials for AMS. All other authors have no conflict of interest. All authors had full control of the primary data, and no agreement has limited their ability to fairly complete and publish the data.
About this article
Cite this article
Bauer, R.M., Kretschmer, A., Stief, C.G. et al. AdVance and AdVance XP slings for the treatment of post-prostatectomy incontinence. World J Urol 33, 145–150 (2015). https://doi.org/10.1007/s00345-014-1297-2
- AdVance sling
- AdVance XP sling
- Male incontinence
- Post-prostatectomy incontinence
- Quality of life