To assess visualization tasks using cone-beam CT (CBCT) compared to multi-detector CT (MDCT) for musculoskeletal extremity imaging.
Ten cadaveric hands and ten knees were examined using a dedicated CBCT prototype and a clinical multi-detector CT using nominal protocols (80kVp-108mAs for CBCT; 120kVp- 300mAs for MDCT). Soft tissue and bone visualization tasks were assessed by four radiologists using five-point satisfaction (for CBCT and MDCT individually) and five-point preference (side-by-side CBCT versus MDCT image quality comparison) rating tests. Ratings were analyzed using Kruskal–Wallis and Wilcoxon signed-rank tests, and observer agreement was assessed using the Kappa-statistic.
Knee CBCT images were rated “excellent” or “good” (median scores 5 and 4) for “bone” and “soft tissue” visualization tasks. Hand CBCT images were rated “excellent” or “adequate” (median scores 5 and 3) for “bone” and “soft tissue” visualization tasks. Preference tests rated CBCT equivalent or superior to MDCT for bone visualization and favoured the MDCT for soft tissue visualization tasks. Intraobserver agreement for CBCT satisfaction tests was fair to almost perfect (κ ~ 0.26–0.92), and interobserver agreement was fair to moderate (κ ~ 0.27–0.54).
CBCT provided excellent image quality for bone visualization and adequate image quality for soft tissue visualization tasks.
• CBCT provided adequate image quality for diagnostic tasks in extremity imaging.
• CBCT images were “excellent” for “bone” and “good/adequate” for “soft tissue” visualization tasks.
• CBCT image quality was equivalent/superior to MDCT for bone visualization tasks.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
Tax calculation will be finalised during checkout.
Cone-beam computed tomography
Multidetector computed tomography
Magnetic resonance imaging
Computed tomography dose index
Fayad LM, Corl F, Fishman EK (2009) Pediatric skeletal trauma: use of multiplanar reformatted and three-dimensional 64-row multidetector CT in the emergency department. Radiographics 29(1):135–50
Demehri S, Chalian M, Farahani SJ, Fishman EK, Fayad LM (2014) Detection and characterization of tendon abnormalities with multidetector computed tomography. J Comput Assist Tomogr 38(2):299–307
Scarfe WC, Farman AG, Sukovic P (2006) Clinical applications of cone-beam computed tomography in dental practice. J Can Dent Assoc 72(1):75–80
O”Connell A, Conover DL, Zhang Y et al (2010) Cone-beam CT for breast imaging: Radiation dose, breast coverage, and image quality. AJR Am J Roentgenol 195(2):496–509
Ingrosso G, Miceli R, Fedele D et al (2012) Cone-beam computed tomography in hypofractionated stereotactic radiotherapy for brain metastases. Radiat Oncol 7(1):54
Yu JJ, Kim GT, Choi YS et al (2012) Accuracy of a cone beam computed tomography-guided surgical stent for orthodontic mini-implant placement. Angle Orthod 82(2):275–83
Carrino JA, Al Muhit A, Zbijewski W, Thawait GK, Stayman JW, Packard N, Senn R, Yang D, Foos DH, Yorkston J, Siewerdsen JH (2014) Dedicated cone-beam CT system for extremity imaging. Radiology 270(3):816–24
Hirschmann A, Pfirrmann CW, Klammer G, Espinosa N, Buck FM (2014) Upright Cone CT of the hindfoot: Comparison of the non-weight-bearing with the upright weight-bearing position. Eur Radiol 24(3):553–8
Tuominen EK, Kankare J, Koskinen SK, Mattila KT (2013) Weight-bearing CT imaging of the lower extremity. AJR Am J Roentgenol 200(1):146–8
Suzuki T, Ito S, Handa S et al (2010) New low-field extremity MRI, compacTscan: comparison with whole-body 1.5 T conventional MRI. Mod Rheumatol 20(4):331–6
Choi JH, Fahrig R, Keil A et al (2013) Fiducial marker-based correction for involuntary motion in weight-bearing C-arm CT scanning of knees. Part I. Numerical model-based optimization. Med Phys 40(9):091905
Koskinen SK, Haapamäki VV, Salo J et al (2013) CT arthrography of the wrist using a novel, mobile, dedicated extremity cone-beam CT (CBCT). Skelet Radiol 42(5):649–57
Guggenberger R, Winklhofer S, Spiczak JV, Andreisek G, Alkadhi H (2013) In vitro high-resolution flat-panel computed tomographic arthrography for artificial cartilage defect detection: comparison with multidetector computed tomography. Invest Radiol 48(8):614–21
Finkenstaedt T, Morsbach F, Calcagni M, Vich M, Pfirrmann CW, Alkadhi H, Runge VM, Andreisek G, Guggenberger R (2014) Metallic artifacts from internal scaphoid fracture fixation screws: comparison between C-arm flat-panel, cone-beam, and multidetector computed tomography. Invest Radiol 49(8):532–9
Zbijewski W, De Jean P, Prakash P et al (2011) A dedicated cone-beam CT system for musculoskeletal extremities imaging: design, optimization, and initial performance characterization. Med Phys 38(8):4700–4713
Dixon RL, Anderson JA, Bakalyar DM, et al (2010) Comprehensive methodology for the evaluation of radiation dose in x-ray computed tomography. AAPM Task Group Report 111
Boone JM, Strauss KJ, Cody DD, et al (2011) Size-Specific Dose Estimates (SSDE) in Pediatric and Adult Body CT Examinations. AAPM Task Group Report 204
Kashani H, Gang JG, Shkumat NA et al (2009) Development of a high-performance dual-energy chest imaging system: initial investigation of diagnostic performance. Acad Radiol 16(4):464–476
Williams DB, Siewerdsen JH, Tward DJ et al (2007) Optimal kvp selection for dual-energy imaging of the chest: evaluation by task-specific observer preference tests. Med Phys 34(10):3916–3925
Bidgood WD Jr, Horii SC (1992) Introduction to the ACR-NEMA DICOM standard. Radiographics 12(2):345–355
Kruskal WH, Wallis WA (1952) Use of ranks in One-criterion variance analysis. J Am Stat Assoc 47(260):583–621
Zou KH, Carrino JA (2002) Comparison of accuracy and interreader agreement in side-by-side versus independent evaluations of MR imaging of the medial collateral ligament of the elbow. Acad Radiol 9(5):520–525
Wilcoxon F (1945) Individual Comparisons by Ranking Methods. Biom Bull 1(6):80–83
Cohen J (1960) A coefficient of agreement for nominal scales. Educ Psychol Meas 20(1):37–46
Cohen J (1968) Weighted kappa: nominal scale agreement with provision for scaled disagreement or partial credit. Psychol Bull 70(4):213–20
Mac Niocaill RF, Quinlan JF, Stapleton RD et al (2011) Inter- and intra-observer variability associated with the use of the Mirels” scoring system for metastatic bone lesions. Int Orthop 35(1):83–86
Landis JR, Koch GG (1977) The measurement of observer agreement for categorical data. Biometrics 33(1):159–174
Jarvik JG (2001) The research framework. AJR Am J Roentgenol 176(4):873–8
Sisniega A, Zbijewski W, Badal A et al (2013) Monte Carlo study of the effects of system geometry and antiscatter grids on cone-beam CT scatter distributions. Med Phys 40(5):051915
Erdogan H, Fessler JA (1999) Monotonic algorithms for transmission tomography. IEEE Trans Med Imaging 18(9):801–814
Muhit AA, Arora S, Ogawa M et al (2013) Peripheral quantitative CT (pQCT) using a dedicated extremity cone-beam CT scanner. SPIE Med Imaging: Biomed Appl Mol, Struct Func Imaging 29:867203–867207
Link TM (2012) Osteoporosis imaging: state of the art and advanced imaging. Radiology 263(1):3–17
Odgaard A (1997) Three-dimensional methods for quantification of cancellous bone architecture. Bone 20(4):315–28
Roos PG, Colbeth RE, Mollov I et al (2004) Multiple-gain-ranging readout method to extend the dynamic range of amorphous silicon flat-panel imagers. ProcSPIE Phys Med Imaging 5368(1):139–149
The scientific guarantor of this publication is Dr. Shadpour Demehri. The authors of this manuscript declare relationships with the following companies: Carestream Health, Inc. The research was supported by National Institutes of Health Grants 2R01-CA-112163 and R21-AR-062993 and academic-industry partnership with Carestream Health (Rochester NY). One of the authors has significant statistical expertise. Institutional Review Board approval was not required because this study does not involve any human subjects. All data was collected using cadavers, and state and institutional requirements were followed. Methodology: retrospective, observational, performed at one institution
Technical aspects of our CBCT scanner prototype illustrated in Fig. 1 were detailed previously . The scanner includes motorized rotation of an X-ray tube and flat-panel detector (FPD). The X-ray tube (XRS-125-7 K-P, Source-Ray, Ronkonkoma, NY) has a stationary anode with 0.5 mm focal spot, 0.875 kW maximum power, 10 mA maximum tube current, and 60–120 kVp range. The X-ray tube is pulsed up to 30 pulses/sec (nominally 25 pulses/sec in studies herein), with a pulse width of 10–30 ms. Total added filtration was 2 mm Al +0.2 mm Cu. A bowtie filter was not used in the current configuration. A custom anti-scatter grid (10:1 grid ratio, 40–70 cm focal range, Jungwon Precision Industries, South Korea) was attached to the FPD. The detector was a PaxScan 3030+ (Varian Medical Systems, Palo Alto, CA) with a 0.5-mm-thick CsI:Tl X-ray converter and 1536 × 1536 pixels at 0.194 mm pitch. The FPD was operated nominally at 25 frames/sec with 2 × 2 pixel binning (768 × 768 pixels at 0.388 mm pitch) and dynamic gain readout . The system can also be operated at full-resolution 1536 × 1536 pixel format, but at half the readout speed. The source-detector distance (SDD) was 550 mm, source-isocentre distance (SID) was 420 mm, magnification factor was 1.3, and the reconstruction field of view was 22 × 22 × 22 cm3. The source and detector traverse a circular orbit with angular extent 240° (exceeding the half-scan range of 180° + fan ≈ 212°). Nominal protocols acquired 490 projections with an imaging time of 20 s.
About this article
Cite this article
Demehri, S., Muhit, A., Zbijewski, W. et al. Assessment of image quality in soft tissue and bone visualization tasks for a dedicated extremity cone-beam CT system. Eur Radiol 25, 1742–1751 (2015). https://doi.org/10.1007/s00330-014-3546-6
- Cone-beam computed tomography