Assessment of image quality in soft tissue and bone visualization tasks for a dedicated extremity cone-beam CT system
- 762 Downloads
To assess visualization tasks using cone-beam CT (CBCT) compared to multi-detector CT (MDCT) for musculoskeletal extremity imaging.
Ten cadaveric hands and ten knees were examined using a dedicated CBCT prototype and a clinical multi-detector CT using nominal protocols (80kVp-108mAs for CBCT; 120kVp- 300mAs for MDCT). Soft tissue and bone visualization tasks were assessed by four radiologists using five-point satisfaction (for CBCT and MDCT individually) and five-point preference (side-by-side CBCT versus MDCT image quality comparison) rating tests. Ratings were analyzed using Kruskal–Wallis and Wilcoxon signed-rank tests, and observer agreement was assessed using the Kappa-statistic.
Knee CBCT images were rated “excellent” or “good” (median scores 5 and 4) for “bone” and “soft tissue” visualization tasks. Hand CBCT images were rated “excellent” or “adequate” (median scores 5 and 3) for “bone” and “soft tissue” visualization tasks. Preference tests rated CBCT equivalent or superior to MDCT for bone visualization and favoured the MDCT for soft tissue visualization tasks. Intraobserver agreement for CBCT satisfaction tests was fair to almost perfect (κ ~ 0.26–0.92), and interobserver agreement was fair to moderate (κ ~ 0.27–0.54).
CBCT provided excellent image quality for bone visualization and adequate image quality for soft tissue visualization tasks.
• CBCT provided adequate image quality for diagnostic tasks in extremity imaging.
• CBCT images were “excellent” for “bone” and “good/adequate” for “soft tissue” visualization tasks.
• CBCT image quality was equivalent/superior to MDCT for bone visualization tasks.
KeywordsCone-beam computed tomography Musculoskeletal Orthopedics Rheumatology Weight-bearing
Abbreviations and acronyms
Cone-beam computed tomography
Multidetector computed tomography
Magnetic resonance imaging
Computed tomography dose index
The scientific guarantor of this publication is Dr. Shadpour Demehri. The authors of this manuscript declare relationships with the following companies: Carestream Health, Inc. The research was supported by National Institutes of Health Grants 2R01-CA-112163 and R21-AR-062993 and academic-industry partnership with Carestream Health (Rochester NY). One of the authors has significant statistical expertise. Institutional Review Board approval was not required because this study does not involve any human subjects. All data was collected using cadavers, and state and institutional requirements were followed. Methodology: retrospective, observational, performed at one institution
- 14.Finkenstaedt T, Morsbach F, Calcagni M, Vich M, Pfirrmann CW, Alkadhi H, Runge VM, Andreisek G, Guggenberger R (2014) Metallic artifacts from internal scaphoid fracture fixation screws: comparison between C-arm flat-panel, cone-beam, and multidetector computed tomography. Invest Radiol 49(8):532–9CrossRefPubMedGoogle Scholar
- 16.Dixon RL, Anderson JA, Bakalyar DM, et al (2010) Comprehensive methodology for the evaluation of radiation dose in x-ray computed tomography. AAPM Task Group Report 111Google Scholar
- 17.Boone JM, Strauss KJ, Cody DD, et al (2011) Size-Specific Dose Estimates (SSDE) in Pediatric and Adult Body CT Examinations. AAPM Task Group Report 204Google Scholar
- 31.Muhit AA, Arora S, Ogawa M et al (2013) Peripheral quantitative CT (pQCT) using a dedicated extremity cone-beam CT scanner. SPIE Med Imaging: Biomed Appl Mol, Struct Func Imaging 29:867203–867207Google Scholar
- 34.Roos PG, Colbeth RE, Mollov I et al (2004) Multiple-gain-ranging readout method to extend the dynamic range of amorphous silicon flat-panel imagers. ProcSPIE Phys Med Imaging 5368(1):139–149Google Scholar