Evaluation of detection sensitivity in comprehensive two-dimensional liquid chromatography separations of an active pharmaceutical ingredient and its degradants
- 842 Downloads
In this paper, we describe the findings of a study aimed at assessing the detection sensitivity of comprehensive two-dimensional high-performance liquid chromatography (LCxLC) separation of a degraded active pharmaceutical ingredient (API) with UV absorption as the detection technique. Specifically, we have examined the impact of the volume and solvent composition (referred to as “interface conditions”) of fractions of first-dimension column effluent transferred to the second dimension for further separation on the ability to resolve and detect low-abundance compounds. Historically, LCxLC has been perceived as being inferior to 1D-LC from the point of view of detection sensitivity. In this work, we demonstrate that LCxLC is sufficiently sensitive to be useful in the pharmaceutical context where in general impurities present at 0.05 % (relative to the API concentration) should be quantified. Moreover, we find that this level of sensitivity is only attained under certain conditions: dilution of the first column effluent with weak solvent (water in this case) prior to injection into the second-dimension column is very beneficial because it promotes focusing of the analyte band in the second column, thereby improving the detection sensitivity of the LCxLC system; and, quantitation limits are also a strong function of peak location in the second-dimension separation window, where baseline disturbances near the dead time of the second column can limit reliable detection of low-abundance compounds.
KeywordsLiquid chromatography Comprehensive Two dimensional Pharmaceutical analysis Sensitivity Degradation
E.T. was supported by a gift from Agilent Technologies University Relations. D.S. was supported by a grant from the National Science Foundation (CHE-1213364) and a gift from Agilent Technologies University Relations. The 1290 Infinity 2D-LC system and HPLC columns used in this work were provided by Agilent Technologies. The LC Image software used to produce the LCxLC chromatograms shown here was provided by GC Image.
- 3.Vivó-Truyols G, van der Wal S, Schoenmakers PJ (2010) Comprehensive study on the optimization of online two-dimensional liquid chromatographic systems considering losses in theoretical peak capacity in first and second-dimensions: a Pareto-optimality approach. Anal Chem 82:8525–8536. doi: 10.1021/ac101420f CrossRefGoogle Scholar
- 12.Al Bakain R, Rivals I, Sassiat P, Thiébaut D, Hennion M-C, Euvrard G, Vial J (2011) Comparison of different statistical approaches to evaluate the orthogonality of chromatographic separations: application to reverse phase systems. J Chromatogr A 1218:2963–2975. doi: 10.1016/j.chroma.2011.03.031 CrossRefGoogle Scholar
- 13.Schure M (2008) In: Schure M, Cohen S (eds) Multidimensional liquid chromatography: theory and applications in industrial chemistry and the life sciences, Wiley-Interscience, HobokenGoogle Scholar
- 16.(2006) International Conference on Harmonization (ICH) Guidelines Q3A (R2). Impurities in new drug substances. International Conference on Harmonization. http://www.ich.org/fileadmin/Public_Web_Site/ICH_Products/Guidelines/Quality/Q3A_R2/Step4/Q3A_R2__Guideline.pdf. Accessed 24 July 2014
- 17.(2005) International Conference on Harmonization (ICH) Guidelines Q2(R1). Validation of analytical procedures: text and methodology. International Conference on Harmonization. http://www.ich.org/fileadmin/Public_Web_Site/ICH_Products/Guidelines/Quality/Q2_R1/Step4/Q2_R1__Guideline.pdf. Accessed 24 July 2014
- 18.Huidobro AL, Pruim P, Schoenmakers P, Barbas C (2008) Ultra rapid liquid chromatography as second dimension in a comprehensive two-dimensional method for the screening of pharmaceutical samples in stability and stress studies. J Chromatogr A 1190:182–190. doi: 10.1016/j.chroma.2008.02.114 CrossRefGoogle Scholar
- 19.Zhang K, Wang J, Tsang M, Wigman L, Chetwyn N (2013) Two-dimensional HPLC in pharmaceutical analysis. Am Pharm RevGoogle Scholar
- 21.(1996) International Conference on Harmonization (ICH) Guidelines Q1B. Stability testing: photostability testing of new drug substances and products. International Conference on Harmonization. http://www.fda.gov/downloads/Drugs/GuidanceComplianceRegulatoryInformation/Guidances/ucm073373.pdf. Accessed 24 July 2014.
- 23.Guiochon G, Colin H (1984) Narrow-bore and micro-bore columns in liquid chromatography. Microcolumn high perform, Liq. Chromatogr. Elsevier, Amsterdam, pp 6–9Google Scholar
- 32.Eghbali H, Sandra K, Tienpont B, Eeltink S, Sandra P, Desmet G (2012) Exploring the possibilities of cryogenic cooling in liquid chromatography for biological applications: a proof of principle. Anal Chem 84:2031–2037. doi: 10.1021/ac203252u
- 37.Stoll DR, Groskreutz SR (2013) Theory and practice of two-dimensional liquid chromatography separations involving the HILIC mode of separation. Hydrophilic Interact. Chromatogr. Guide Pract. Wiley, pp 265–305Google Scholar