Liquid injection field desorption/ionization of reactive transition metal complexes
- 210 Downloads
Liquid injection field desorption/ionization (LIFDI) has been applied to identify transition metal complexes that are highly reactive to air and moisture by mass spectrometry. The complexes of nickel and rhodium were supplied as dilute solutions (∼0.2 mg ml−1) in toluene, tetrahydrofuran or acetonitrile, and were applied onto the field desorption emitter inside the vacuum of the ion source under inert conditions by means of the injection capillary unique to the LIFDI set-up. LIFDI mass spectrometry on a double-focusing magnetic sector instrument provided spectra exhibiting intense molecular ion peaks for the species investigated or signals that could easily be related to the target compound by assuming neutral loss of the weakest-bound ligand. Eventually, byproducts of the synthesis or other components resulting from incomplete reactions or some degree of decomposition were also detected.
KeywordsLIFDI Field desorption Mass spectrometry Desorption/ionization Transition metal complexes Ligands
Generous support of this work by the Fonds der Chemischen Industrie, by BASF AG and by the Deutsche Forschungsgemeinschaft (SFB 623) is gratefully acknowledged.
- 2.Beckey HD, Schulten H-R (1975) Angew Chem 87:425–438Google Scholar
- 3.Beckey HD (1977) Principles of field desorption and field ionization mass spectrometry. Pergamon, OxfordGoogle Scholar
- 4.Schulten H-R (1978) Adv Mass Spectrom 7A:83–97Google Scholar
- 5.Prókai L (1990) Field desorption mass spectrometry. Marcel Dekker, New YorkGoogle Scholar
- 6.Inghram MG, Gomer R (1955) Z Naturforsch 10A:863–872Google Scholar
- 13.Linden HB (2001) In: Proc 49th ASMS Annual Conf Mass Spectrometry and Allied Topics, MPA 024, 27–31 May 2001, Chicago, ILGoogle Scholar
- 14.Linden HB (2002) In: Proc 50th ASMS Annual Conf Mass Spectrometry and Allied Topics, MPL 373, 2–6 June 2002, Orlando, ILGoogle Scholar
- 17.Dole RB (ed) (1997) Electrospray ionization mass spectrometry—fundamentals, instrumentation and applications. Wiley, Chichester, UKGoogle Scholar
- 18.Pramanik BN, Ganguly AK, Gross ML (eds) (2002) Applied electrospray mass spectrometry. Marcel Dekker, New YorkGoogle Scholar