Angiographic C-arm CT visualization of the Woven EndoBridge Cerebral Aneurysm Embolization Device (WEB): First Experience in an Animal Aneurysm Model
- First Online:
- 326 Downloads
Improvements in the imaging capabilities of angiographic C-arm computed tomography (CT) using flat detector angiographic (FD-CT) systems now provide a means for the in vivo visualization of devices used for the treatment of intracranial aneurysms. One such device, the WEB embolization device, is made of a braided mesh of nitinol wires, the size of which are near to the limits of visualization using conventional x-ray fluoroscopy. Our hypothesis was that the imaging of these implants C-arm CT would provide useful information regarding their positioning and deployment.
In five New Zealand white rabbits elastase induced aneurysms were created and subsequently treated using a WEB. Imaging was performed using digital subtraction angiography (DSA), X-ray imaging and two different Angiographic C-arm CT protocols. The images were evaluated by two neuroradiologists using an evaluation scale.
The mesh of the WEB was barely visible on the DSA or x-ray fluoroscopy images. Volume rendering technique (VRT) reconstruction and multiplanar reconstruction (MPR) of images done using the C-arm CT protocols clearly delineated the shape and structure of the device. Contrast-enhanced MPR and VRT reconstructions allowed assessment of the status of blood flow in the aneurysms. Beam hardening artifacts caused by platinum markers on the WEB were present.
In vivo C-arm CT imaging of the WEB is feasible and allows precise determination of the position and deployment status of the device. On contrast-enhanced images the occlusion status of aneurysms and the positioning of the WEB in relationship to the parent artery can be evaluated. C-arm-CT may serve as a minimal-invasive follow-up imaging modality.
KeywordsWEB device Flat-detector computed tomography Angiographic C-arm CT Endovascular treatment aneurysm
- 4.Lubicz B, Mine B, Collignon L, Brisbois D, Duckwiler G, Strother C. WEB device for endovascular treatment of wide-neck bifurcation aneurysms. AJNR Am J Neuroradiol. 2013 Jan 4. (Epub ahead of print)Google Scholar
- 6.Pfaff J, Struffert T, Gölitz P, Doerfler A. Angiographic CT for intraprocedural monitoring of complex neuroendovascular procedures. AJNR Am J Neuroradiol. 2012 Apr 5. (Epub ahead of print)Google Scholar
- 22.Struffert T, Ott S, Kowarschik M, Bender F, Adamek E, Engelhorn T, Gölitz P, Lang S, Strother CM, Doerfler A. Measurement of quantifiable parameters by time-density curves in the elastase-induced aneurysm model: first results in the comparison of a flow diverter and a conventional aneurysm stent. Eur Radiol. 2013;23:521–7.PubMedCrossRefGoogle Scholar