Wearable electroencephalograph system with preamplified electrodes
- 92 Downloads
KeywordsBioelectrical Impedance Analysis Sensor Electrode Main Amplifier Differential Operation Input Gain
These keywords were added by machine and not by the authors. This process is experimental and the keywords may be updated as the learning algorithm improves.
Unable to display preview. Download preview PDF.
- Barnea, O., andDeutsch, S. (1986): ‘Telemetry system for stimulation and monitoring of freely moving patients,’IEEE Trans.,BME-33, pp. 320–326Google Scholar
- Ebersole, J. S. (1989): ‘Ambulatory EEG monitoring’ (Raven Press, New York)Google Scholar
- Ginsberg, H. D., Yoshii, F., Vibulsresth, S.,et al. (1987): ‘Human task-specific somatosensory activation,’Neurol. 34, pp. 1301–1308Google Scholar
- MIL-E-36359 (1965): ‘Military specification electroencephalograph, 8 channels, 110 volt, 60 cycle’Google Scholar
- Pollak, V. (1974): ‘An equivalent diagram for the interface impedance of metal needle electrodes,’Med. Biol. Eng., pp. 454–459Google Scholar
- Roland, P. E., andFriberg, L. (1985): ‘Localiation of cortical areas activated by thinking,’J. Neurophysiol.,53, pp. 1219–1243Google Scholar
- Smit, H. W., Verton, K., andGrimbergen, A. (1987): ‘A low-cost multi-channel preamplifier for physiological signals,’IEEE Trans.,BME-34, p. 307Google Scholar
- Stores, G. (1986): ‘Comparison of video and ambulatory monitoring in the investigation of attacks in children.’. Proc. Fifth Int. Symposium on Ambulatory Monitoring, Cleup, Padua Padova, pp. 633–638Google Scholar
© IFMBE 1994