Transverse path of ultrasound waves in thick-walled cylinders
- 55 Downloads
The path is examined of a beam of ultrasound through the wall and central cavity of a series of thick-walled cylinders made of polymethylmethacrylate (PMMA) and brass used to simulate the cross-section of the cortical shaft of a long bone. Water is used as a coupling medium for the external ultrasound transducers and to fill the central cavity. Broadband pulse excitation is employed with two focused 2.5 MHz transducers mounted on a digital calliper. The transmission times of flight are measured on a 100 MHz digital cathode ray oscilloscope. Two distinct pulses of ultrasound were consistently observed at the receiver transducer, the first due to a nonlinear path only in the cylinder wall, and the second due to a linear path travelling directly between the transducers through the centre of the cylinder. The nonlinear path can be explained by refraction at the external surface of the cylinder and diffraction around the central cavity. The nonlinear path is influenced by the relative size of the central cavity. The proposed model for the ultrasound propagation in the cylinder wall gives an excellent correlation with the observed speed and attenuation. The technique can be applied to long bones and has considerable potential as a means of measuring bone quality in human and veterinary medicine.
KeywordsSonic path Speed Ultrasound Velocity
Unable to display preview. Download preview PDF.
- Abendschein, W., andHyatt, G. W. (1970): ‘Ultrasonics and selected physical properties of bone,’Clin. Orthop. Rel. Res.,69, pp. 294–301Google Scholar
- Carter, D. R., andHayes, W. C. (1977): ‘The compressive behaviour of bone as a two phase porous structure,’J. Bone Joint Surg.,59A, pp. 954–962Google Scholar
- Greenfield, M. A., Craven, J. D., Huddleston, A., Kehren, M. L., Wishko, D. S. andStern, R. (1981): ‘Measurement of the velocity of ultrasound in cortical bonein vivo,’Radiology,138, pp. 701–10Google Scholar
- Jeffcott, L. B., Buckingham, S. H. W., McCarthy, R. N., Cleeland, J. C., Scotti, E., andMcCartney, R. N. (1988): ‘Noninvasive measurement of bone: a review of clinical and research applications in the horse,’Eq. Vet. J. Supp.,6, pp. 71–79Google Scholar
- Langton, C. M., Riggs, C. M., andEvans, G. P. (1991): ‘Pathway of ultrasound waves in the equine third metacarpal bone,’J. Biomed. Eng.,13, pp 113–118Google Scholar
- Lees, S. (1986): ‘Sonic properties of mieralised tissues’in Greenleaf, J. F. (Ed.): ‘Tissue characterisation with ultrasound’ (CRC Press, Boca Raton) pp. 208–226Google Scholar
- McCarthy, R. N., Jeffcott, L. B., andMcCartney, R. N. (1991): ‘A method for determining the transmission speed of ultrasound in long bones’,J. Biomech., to be published.Google Scholar
- McCarthy, R. N., Jeffcott, L. B., andMcCartney, R. N. (1988): ‘Ultrasonic transmission velocity and single photon absorptiometric measurement of metacarpal strength: anin vitro study in the horse,’Eq. Vet. J. Supp.,6, pp. 80–87Google Scholar
- Pinkerton, J. M. M. (1947): ‘A pulse method for the measurement of ultrasonic absorption in liquids: results for water,’Nature,160, pp. 128–129Google Scholar
- Rossman, P. J. (1987): ‘Measurement of ultrasonic velocity and attenuation in the human os calcis and their relationships to photon absorptiometry bone mineral measurements,’ Masters, thesis, University of Winsconsin at MadisonGoogle Scholar