A microtomographic system for the nondestructive evaluation of bone architecture
- 959 Downloads
Microtomography (micro-computed-tomography, μ-CT) is a method to image and quantify trabecular bone. It has the capability to address the role of trabecular architecture on the mechanical properties of bone and to study trabecular bone remodeling. The system described in this work is based on a compact fan-beam type tomograph that can work in spiral scanning or multislice mode. An X-ray tube with a microfocus is used as a source, a CCD-array as a detector. Samples with diameters from a few millimeters to a maximum of 14 mm can be measured, typically, bone biopsies with a diameter of 8 mm and a length of approximately 10 mm are measured. Spatial resolution is 28 μm. Usually the volume of interest contains 4×4×4 mm3 and is represented in 14×14×14 μm3 voxels. 3D stereological indices are extracted according to the standard definitions used in histomorphometry. Triangular surface representation is effected with an extended marching cube algorithm and forms a convenient basis for finite element analysis. Microtomographic measurements may be employed to “calibrate” lower-dose, lower-resolution imagesin vivo as well as to nondestructively assess unprocessed surgical bone biopsy specimens. These specimens remain intact for mechanical or histological testing.
Key wordsMicrotomography Computed tomography Bone structure Bone strength Osteoporosis
Unable to display preview. Download preview PDF.
- 2.Parfitt AM (1983) The stereologic basis of bone histomorphometry. Theory of quantitative microscopy and reconstruction of the third dimension. In: Recker R (ed) Bone histomorphometry. Techniques and Interpretation. CRC Press, Boca Raton, pp 53–87Google Scholar
- 3.Vogel M, Hahn M, Pompesius-Kempa M, Delling G (1989) Trabecular microarchitecture of the human spine. In: HG Willert, FHW Heuck (eds) Neuere Ergebnisse in der Osteologie, vol 4. Springer Verlag Heidelberg, pp 449–455Google Scholar
- 7.Feldkamp LA, Goldstein SA, Parfitt AM, Jesion G, Kleerekoper M (1994) The direct examination of three-dimensional bone architecture in vitro by computed tomography. J Bone Miner Res 4:3–11Google Scholar
- 11.DeHoff RT, Aigeltinger EH, Craig KR (1972) Experimental determination of the topological properties of 3D-microstructures. J Microscopy 95:69Google Scholar
- 15.Lorensen WE, Cline HE (1987) Marching cubes: a high resolution 3D surface construction algorithm. Comput Graphics 21: 163–169Google Scholar
- 17.Whitehouse WJ (1974) The quantitative morphology of anisotropic trabecular bone. J Microscopy 101:153–168Google Scholar
- 20.Müller R, Koller B, Hildebrand T, Gionolini S, Rüegsegger P (1995) Resolution-dependency of microstructural properties of cancellous bone based on three-dimensional μ-tomography. In: Pedotti R, Rabischong P (eds) Abstracts 3rd European Conference on Engineering and Medicine, 1995, vol 112Google Scholar