Treatment of multiple organ failure by a totally implantable ventricular assist device
- 27 Downloads
Multiple organ failure (MOF) becomes an important problem during clinical use of the ventricular assist device (VAD). To improve the clinical record of the VAD, the function of other organs may be vitally important. For that reason, we have been developing a VAD system aiming at improving the function of other organs. Development of the vibrating flow pump (VFP), which can generate a unique flow pattern from 10 to 50Hz, is still ongoing in our institute. In order to evaluate brain blood flow and oxygen consumption, oxyhemoglobin was measured with NIRO (Hamamatsu Photo.) in healthy adult goats. Four healthy adult goats were anesthetized with halothane inhalation, and left thoracotomy was performed for left heart bypass. Oxyhemoglobin in the brain was measured with the recording of the hemodynamic variables during left heart assistance with the VFP system. During left heart bypass with the VFP system, the hemodynamic parameters kept within the normal range, and satisfactory pump output was easily obtained. Pump output was kept within 40%–50% bypass to evaluate the effect of high-frequency oscillating assist flow on brain blood flow with the same cardiac output. Interesting results were observed during the experiments. During 30Hz driving of the VFP left heart assistance, oxyhemoglobin suggested that brain blood flow was significantly increased compared with another drive frequency with the same total cardiac output. These results suggest that we can control brain blood flow with a totally implantable VAD system such as the VFP system, which can control the frequency of the blood flow.
Key wordsVentricular assist device (VAD) Vibrating flow pump (VFP) Totally implantable Brain blood flow
Unable to display preview. Download preview PDF.
- 5.Tatsumi E, Nakamura M, Masuzawa T, Taenaka Y, Sohn YS, Nishimura T, Nakata M, Nakatani T, Ohno T, Endo S, Takiura K, Takewa Y, Kakuta Y, Takano H. In vitro and in vivo evaluation of a left-right balancing capacity of an interatrial shunt in an electrohydraulic total artificial heart system. ASAIO J 1997;43:M619–625PubMedGoogle Scholar
- 6.Kung RTV, Yu RS, Ochs BD, Parnis SM, Macris MP, Frazier OH. Progress in the development of the ABIOMED total artificial heart. ASAIO J 1995:M245–248Google Scholar
- 9.Yambe T, Nitta S, Katahira Y, Sonobe T, Naganuma S, Kakinuma Y, Matsuzawa H, Tanaka M, Miura M, Satoh N, Mohri H, Yoshizawa M, Takeda H. Fundamental rhythm of the sympathetic nerve discharges in animals with total artificial hearts. ASAIO J 1992;39:91–95Google Scholar
- 10.Yambe T, Nitta S, Sonobe T, Naganuma S, Kakinuma Y, Izutsu K, Akiho H, Kobayashi S, Ohsawa N, Nanka S, Tanaka M, Fukuji T, Miura M, Uchida N, Sato N, Tabayashi K, Koide S, Abe K, Takeda H, Yoshizawa H. Deterministic chaos in the hemodynamics of an artificial heart. ASAIO J 1995;41:84–88PubMedGoogle Scholar
- 11.Yambe T, Nitta S, Sonobe T, Naganuma S, Kakinuma Y, Kobayashi S, Nanka S, Ohsawa N, Akiho H, Tanaka M, Fukuju T, Miura M, Uchida N, Sato N, Mohri H, Yoshizawa M, Koide S, Abe K, Takeda H. Origin of the rhythmical fluctuations in hemodynamic parameters in animal without natural heartbeat. Artif Organs 1993;1071–1021Google Scholar
- 12.Yambe T, Nitta S, Sonobe T, Naganuma S, Kakinuma Y, Kobayashi S, Nanka S, Ohsawa N, Akiho H, Tanaka M, Fukuju T, Miura M, Uchida N, Sato N, Mohri H, Koide S, Yoshizawa M, Abe K, Takeda H. Chaotic behavior of the hemodynamics with ventricular assist device. Int J Artif Organs 1995;18:17–21PubMedGoogle Scholar
- 13.Nojiri C, Kijima T, Maekawa J, Horiuchi K, Kido T, Sugiyama T, Mori T, Sugiura N, Asada T, Shimane H, Mishimura K, Ban T, Akamatsu T, Ozaki T, Ito H, Suzuki M, Akutsu T. More than 1 year continuous operation of a centrifugal pump with a magnetically suspended impeller. ASAIO J 1997;43:M548–552PubMedGoogle Scholar
- 15.Yambe T, Izutsu K, Hashimoto H, Yoshizawa M, Tanaka A, Abe K, Fukuju T, Tabayashi K, Takeda H, Shizuka K, Sonobe T, Naganuma S, Kobayashi S, Nanka S, Nitta S. Control of the pulmonary arterial resistances by the use of the oscillated assist flow. Artif Organs 1998;22:430–433PubMedCrossRefGoogle Scholar