Reconstruction of an iliac crest defect with a bioactive ceramic prosthesis
- 40 Downloads
Between June 1987 and December 1990, an iliac crest prosthesis made of bioactive apatite- and wollastonite-containing glass ceramic (A-W·GC) was used in 60 patients for the reconstruction of the iliac crest defect after harvesting autogenous tricortical iliac bone graft. The clinical results of this prosthesis were satisfactory. No patients felt spontaneous pain in the reconstructed area, and 93% of the patients had no tenderness there. In the radiological evaluation at the final follow-up, no apparent “radiolucent clear zone” was detected at the prosthesis-iliac bone junction in 98% of the patients. Excellent new bone formation between the prosthesis and the iliac crest was also noticed in 96% of the patients. The A-W·GC iliac crest prosthesis was beneficial for reconstruction of the iliac crest defect.
Key wordsBioactive ceramics Iliac crest prosthesis Ilium Reconstruction
Unable to display preview. Download preview PDF.
- 1.Asano S, Kaneda K, Abumi K, Hashimoto T (1989) Clinical experience of A-W containing glass ceramic spacer for reconstruction of iliac crest defect (in Japanese). The Hakkaido Journal of Orthopaedics and Traumatology 33: 61–66Google Scholar
- 3.Cockin J (1971) Autologous bone grafting-complications at the donor site. J Bone Joint Surg [Br] 53: 153Google Scholar
- 5.Hashimoto T, Kaneda K, Abumi K, Kurakami T, Shirado O, Takeda N, Takahashi H, Fujiya M (1987) An experience of use of ceramic implant for iliac bone defect (in Japanese). Jpn Soc Orthop Ceram Impl 7: 55–58Google Scholar
- 8.Kaneda K, Hashimoto T, Abumi K (1988) Reconstruction of the iliac crest with ceramic replacement. Orthop Trans 12: 633Google Scholar
- 13.Kokubo T, Shigematsu M, Nagashima Y, Tashiro M, Nakamura T, Yamamuro T, Higashi S (1982) Apatite- and wollastonite-containing glass-ceramics for prosthetic application. Bull Inst Chem Res Kyoto Univ 60: 260–268Google Scholar
- 14.Kokubo T (1989) Mechanical properties of a new type of glass-ceramic for prosthetic applications. In: Tsuruta T, Nakajima A (eds) Multiphase biomedical materials. VSP, Utrecht, pp 105–114Google Scholar
- 26.Satoh R, Furuya K, Shinomiya K, Ukegawa H, Satoh M, Okamoto A, Hasegawa S, Aoki H (1987) Clinical study on hydroxyapatite spacer for iliac bone defect (in Japanese). Orthop Surg Traumatol 30: 1439–1443Google Scholar
- 27.Takemitsu Y, Harada Y, Satoh Y, Shibata M, Suzuki S (1983) Ceramic implant for iliac bone defect (in Japanese). Jpn Soc Orthop Ceram Impl 3: 367–370Google Scholar
- 28.Yamamuro T (1990) Reconstruction of the iliac crest with bioactive glassceramic prosthesis. In: Yamamuro T, Hench LL, Wilson J (eds) Handbook of bioactive ceramics, Vol. I. CRC Press, Boca Raton, pp 335–342Google Scholar
- 29.Yamamuro T, Shikata J, Okumura H, Kitsugi T, Kakutani Y, Matsui T, Kokubo T (1990) Replacement of the lumbar vertebrae of sheep with ceramic prosthesis. J Bone Joint Surg [Br] 72: 889–893Google Scholar