Application of the nuclide identification system SHAMAN in monitoring the Comprehensive Test Ban Treaty
- 57 Downloads
SHAMAN is an expert system for qualitative and quantitative radionuclide identification in gamma spectrometry. SHAMAN requires as input the calibrations, peak search, and fitting results from reliable spectral analysis software, such as SAMPO. SHAMAN uses a comprehensive reference library with 2600 radionuclides and 80 000 gamma-lines, as well as a rule base consisting of sixty inference rules. Identification results are presented both via an interactive graphical interface and in the form of configurable text reports. An organization has been established for monitoring the recent Comprehensive Test Ban Treaty. For radionuclide monitoring, 80 stations will be set up around the world. Air-filter gammaspectra will be collected from these stations on a daily basis and they will need to be reliably analyzed with minimum turnaround time. SHAMAN is currently being evaluated within the prototype monitoring system as an automated radionuclide identifier, in parallel with existing radionuclide identification software. In air-filter monitoring, very low concentrations of radionuclides are measured from bulky sources in close geometry and with long counting time. In this case true coincidence summing and self-absorption become important factors. SHAMAN is able to take into account these complicated phenomena, and the results it produces have been found to be very reliable and accurate.
KeywordsRadionuclide Gamma Spectrometry Complicated Phenomenon Comprehensive Reference True Coincidence
Unable to display preview. Download preview PDF.
- 1.P. A. Aarnio, M. T. Nikkinen, J. T. Routti, J. Radioanal. Nucl. Chem., 193 (1995) 179.Google Scholar
- 2.P. A. Aarnio, T. T. Hakulinen, J. T. Routti, J. Radioanal. Nucl. Chem., 160 (1992) 245.Google Scholar
- 3.P. A. Aarnio, J. J. Ala-Heikkilä, T. T. Hakulinen, J. T. Routti, J. Radioanal. Nucl. Chem., 193 (1995) 219.Google Scholar
- 4.Preliminary User's Guide for Radionuclide Data Processing Software, Version 1.2. Pacific-Sierra Research Corporation, Arlington, 1996.Google Scholar
- 5.L. R. Mason, J. D. Bohner, W. C. Evans, J. B. Bohlin, D. L. Williams, K. K. Keksz, Radionuclide Monitoring Operations Report of the Prototype International Data Center-Fourth Quarter CY1996. Pacific-Sierra Research Corporation Technical Report 2698, Arlington 1997.Google Scholar
- 6.P. Harmon, D. King, Expert Systems, John Wiley & Sons, Inc. New York, 1985.Google Scholar
- 7.K. Sinkko, Licenciate's Thesis, Helsinki University 1981.Google Scholar
- 8.K. Heydorn, S. K. Rhee, ANS Transactions 60 (1989) 3.Google Scholar
- 9.J. K. Ousterhout, Tcl and the Tk Toolkit, Addison Wesley Publishing Company, Inc. Reading Massachusetts 1995.Google Scholar
- 10.J. J. Ala-Heikkilä, T. T. Hakulinen, P. A. Aarnio, H. I. K. Toivonen, L. R. Mason, W. C. Evans, J. B. Bohlin, CMR Technical Report CMR-97/05. Arlington 1997.Google Scholar
- 11.J. Bohlin, private communications, January 1997.Google Scholar