A reflex spectrographic technique for in-plane strain analysis
- 40 Downloads
A reflex camera that incorporates a light source and spectroscope for recording in-plane displacement fringes formed by reflective diffraction gratings is described and illustrated.
A technique for replicating reflective diffraction gratings on plane metallic surfaces is given in detail.
Typical fringe patterns of a mild-steel tension specimen containing a long shallow notch obtained using 6000-lines/inch gratings show the elastic deformation of the specimen.
The method should prove a useful supplement or alternative to reflective photoelasticity in elasto-plastic studies, particularly when strain distributions in the transition region are of interest.
KeywordsMechanical Engineer Light Source Fluid Dynamics Transition Region Elastic Deformation
angle of incidence
angle of emergence ofp-th order passing through grating G1 at point A
angle of reflection ofp, s-p order sequence at point B
angle of emergence ofp, s-p, r-s order sequence from grating G1 at point C
phase of emergent beam on leaving point C
fractional displacement of the rulings of the reference grating G1 at point C with respect to those on the model grating G2 at point N
wavelength of light
amplitude ofu component of emergent sequence
amplitude of emergent sequence
phase constant depending upon position of point C
fractional position of point A in ruling
phase constant depending upon order sequenceu
spectral order numbers
thickness of air gap
number defining order sequence
ordinate of point C
Unable to display preview. Download preview PDF.
- 1.Theocaris, P. S., “Moiré Fringes: A Powerful Measuring Device,”Appl. Mech Rec.,15,5,333–339 (1962).Google Scholar
- 2.Vinkier, A., andDechaene, R., “Use of the Moiré Effect to Measure Plastic Strains,”J. Basic Eng.,82,Series D,426–434 (1960).Google Scholar
- 3.Dantu, P., “Extension of the Moiré Method to Thermal Problems,”Experimental Mechanics,4,3,64–69 (1964).Google Scholar
- 4.Middleton, E., “Discussion on a “Method to Increase the Accuracy of the Moiré Method by Fu-Pen Chiang,”J. Eng. Mech. Div., Am. Soc. Civil Engrs.,91,EM5,227–232 (October1965).Google Scholar
- 5.Ross, B. E., Sciammarella, C. A., andSturgeon, P., “Basic Optical Law in the Interpretation of Moiré Patterns Applied to the Analysis of Strain, Part 2,”Experimental Mechanics,5,6,161–166 (1965).Google Scholar
- 6.Middleton, E., andJenkins, C. J., “A Moiré Method of Strain Analysis for Student Use,”Bull Mech. Eng. Ed.,5,257–267 (1966).Google Scholar
- 7.Low, I.A.B., “The Moiré Method Using Diffraction Gratings,” SESA Paper No. 1049, 1966 SESA Spring Meeting, Detroit (May 1966).Google Scholar
- 8.Middleton, E., “A Reflection Technique for the Survey of the Deflection of Flat Plates,” to be published in Experimental Mechanics.Google Scholar
- 9.Guild, J., “The Interference Systems of Crossed Diffrection Gratings,”Monographs on the Physics and Chemistry of Materials, Clarendon Press, Oxford (1956).Google Scholar
- 10.Hall, R. G. N., andSayce, L. A., “On the Production of Diffraction Gratings,” The Generation of Helical Rulings and the Preparation of Plane Gratings Therefrom,”Proc. Roy. Soc. London, 215, Series A, 536–550 (1952).Google Scholar