Response of foil strain gages to hydraulic pressure
- 56 Downloads
A series of tests to determine the effect that hydraulic pressure has on the output of electrical-resistance foil strain gages is described. Tests were conducted on Micro-Measurements gages, type EA-13-125BT-120 for pressures up to 20.7 MPa and frequencies up to 150 Hz, both for purely dilatational strain as well as various combinations of dilatational and distortional strains.
The results showed that gage performance due to the combined effects of hydrostatic pressure and distortion of the structure differed from the condition predicted when pressure and distortion were considered separately. Rate of pressurization (and, hence, strain rate) had no measurable effect within practical limits.
KeywordsMechanical Engineer Fluid Dynamics Combine Effect Hydrostatic Pressure Strain Gage
test-specimen external radius
test-specimen internal radius
Poisson's ratio with the stress applied parallel to the direction of extrusion
Poisson's ratio with the stress applied perpendicular to the direction of extrusion and the transverse strain also measured perpendicular to the direction of extrusion
Poisson's ratio with the stress applied perpendicular to the direction of extrusion and the transverse strain measured in the direction of extrusion
Unable to display preview. Download preview PDF.
- 1.Clough, W. R., Shank, M. E. andZaid, M., “The Behavior of SR-4 Wire-resistance Strain Gages on Certain Materials in the Presence of Hydrostatic Pressure,”Proc. of the SESA,X (2),167–176 (1953).Google Scholar
- 2.Majors, H., “Influence of Fluid Pressure on SR-4 Strain Gages,”Proc. of the SESA,XIII (1),13–24 (1955).Google Scholar
- 3.Guerard, J. P. andWeissmann, G. P., “Effect of Hydrostatic Pressure on SR-4 Strain Gages,”Proc. of the SESA,XVI (1),151–156 (1958).Google Scholar
- 4.Gerdeen, J. L., “Effect of Pressure on Small Foil Strain Gages,”Experimental Mechanics,3 (3),73–80 (Mar. 1963).Google Scholar
- 5.Milligan, R. V., “The Effect of High Pressure on Foil Strain Gages,”Experimental Mechanics,4 (2),25–36 (Feb. 1964).Google Scholar
- 6.Milligan, R. V., “The Effects of High Pressure on Foil Strain Gages on Convex and Concave Surfaces,”Experimental Mechanics,5 (2),59–64 (Feb. 1965).Google Scholar
- 7.Milligan, R. V., “The Gross Hydrostatic-pressure Effect as Related to Foil and Wire Strain Gages,”Experimental Mechanics,7 (2),67–74 (Feb. 1967).Google Scholar
- 8.Brace, W. F., “Effect of Pressure on Electric-resistance Strain Gages,”Experimental Mechanics,4 (7),212–216 (July 1964).Google Scholar
- 9.Kular, G. S., “Use of Foil Strain Gages at High Hydrostatic Pressure,”Experimental Mechanics,12 (7),311–316 (July 1972).Google Scholar
- 10.Foster, C. G., “Accurate Measurement of Poisson's Ratio in Small Samples,”Experimental Mechanics,16 (8),311–315 (August 1976).Google Scholar
- 11.Foster, C. G., “A Simple Instrumentation Gland for Use with Medium to High Hydraulic Pressures,” J. of Physics E: Scientific Instruments (June 1974).Google Scholar