Measuring small internal pressures along a tube during steady flow
- 43 Downloads
A tube that can accurately measure small strains and pressure profiles during flow of non-Newtonian paste explosives has been designed and strain gaged. Equiangular rosette strain gages were installed along the length of a 6-mm-diam, 0.46-m-long thin-walled aluminum tube. The rosettes were oriented in the classical stress-gage configuration to measure circumferential stress and, hence, internal pressure independent of other stresses.
The tube was static calibrated on a floating-piston pressure calibrator. Steady flow calibration was accomplished by extruding a viscous Newtonian silicone oil. Inlet pressure ranged from 0.52 to 2.1 MPa (75 to 300 psi). For the low-pressure 0.52-MPa silicone-oil extrusion, the full-scale strain levels varied from 6 to 53 μm/m. For all eight strain-gage stations, the maximum deviation from a linear pressure profile was equivalent to 0.5 μm/m. A pulsed-current-excitation signal-conditioning and digital data-acquisition system provided the necessary stability and precision to measure these unusually low-strain levels accurately.
KeywordsInternal Pressure Steady Flow Pressure Profile Inlet Pressure Circumferential Stress
Unable to display preview. Download preview PDF.
- 1.Tovey, F. Michael, “Material Properties in the Design of Transducer Flexures,” Proc. Western Regional Strain Gage Committee, SESA (Sept. 20–21, 1971).Google Scholar
- 2.Weinstein, Warren D., “Microperformance of Metals,”Machine Design,41 (28),174–181 (Dec. 11,1969).Google Scholar
- 3.Kern, Robert E., “The Stress Gage,”Proc. SESA,4 (1),124 (1946).Google Scholar
- 4.Perry, C.C. andLissner, H.R., The Strain Gage Primer, McGraw Hill, New York (1955).Google Scholar
- 5.Murray, W.M., “Some Simplifications in Rosette Analysis,”Proc. SESA,15 (2),39 (1958).Google Scholar
- 6.Flynn, Paul D., “Strain-gage Instrumentation for Ammunition Testing,”Experimental Mechanics,15 (5),192 (May 1975).Google Scholar
- 7.Dittbenner, Gerald R., “The Application of Strip Strain Gages for Measuring Residual Stresses in Beryllium,” Proc. Third SESA Int. Conf. Experimental Mechanics, Los Angeles, CA (May 13–18, 1973).Google Scholar
- 8.Freynik, H.S., Jr. andDittbenner, G.R., “Strain-gage-stability Measurements for Years at 75°C in Air,”Experimental Mechanics,16 (4),155 (Apr. 1976).Google Scholar
- 9.Liston, D.H., “Pulsed Strain Gage Excitation and Digital Data Acquisition—A Winning Combination,” Proc. Western Regional Strain Gage Committee, SESA (Aug. 20–21, 1973).Google Scholar