Reference standards for quantitative trace analysis of PCB's by GC. Technical PCB formulations for the calibration of ECD and MSD responses
- 55 Downloads
The composition of any technical PCB formulation can be determined directly by analyzing the PCB sample by gas chromatography with a flame ionization detector (GC-FID), provided the relative molecular masses of the components are known. The responses of electron capture and selected-ion monitoring, mass-spectra detectors can then be calibrated for individual PCB congeners by correlation of the chromatographic patterns with those of concentrated PCB samples obtained by GC-FID. This procedure, which uses a given technical PCB formulation as a secondary reference standard mixture, is to be preferred over existing calibration methods, when results with ±10% errors are acceptable because commercial PCB formulations cover the whole range of chlorination products.
Key WordsCapillary gas chromatography Polychlorinated biphenyls (PCBs) Quantitative analysis Detector response factors (FID, ECD, MS)
Unable to display preview. Download preview PDF.
- F. I. Onuska, K. A. Terry, HRC & CC9, 671 (1986).Google Scholar
- National Research Council of Canada, Institute for Marine Biosciences, Marine Analytical Chemistry Standards Program, Halifax Nova Scotia, Canada B3H3Z1, January 1985, Documentation Revised: January 1991.Google Scholar
- R. C. Dougherty, Anal. Chem.53, 625A (1981).Google Scholar