Reflectance pulse oximetry at the forehead of newborns: The influence of varying pressure on the probe
- 72 Downloads
Objective: Transmission pulse oximetry (TPO) is not a practical method of intrapartum fetal monitoring of arterial oxygen saturation. Reflectance pulse oximetry (RPO) requires a sensor applied to the skin of the fetal head and may be a useful technique. During labor, various degrees of pressure will be exerted on the RPO sensor. Previous studies have shown that moderate pressure on the sensor can improve the RPO signal. At increasing pressure, however, blood flow underneath the sensor will be occluded. This study examines the influence of pressure applied to the RPO sensor on the signal from the forehead of healthy newborns as a model for the fetal situation.Methods: After institutional approval, 12 healthy newborns were studied. The RPO probe was placed at the forehead. Pressure on the probe was increased stepwise from 0 to 80 mmHg, and the effect on the ratio between the relative changes of the red and infrared light intensities (R/IR, inversely related to oxygen saturation) and pulse sizes was evaluated. Additionally, the effect of firm pressure (>150 mmHg) on the probe was evaluated.Results: R/IR values remained virtually unchanged when pressure onto the probe was increased from 0 to 80 mmHg, although the standard deviation slightly decreased. The pulse size increased as pressure on the probe increased. During firm pressure on the probe (> 150 mmHg), plethysmographic signals remained detectable, but R/IR values markedly increased.Conclusions: In newborns, mild to moderate pressure on the probe has little influence on the RPO signal at the forehead. Even during firm pressure, RPO can be used to obtain pulsatile signals, that presumably derive from tissue underneath the skull, such as the cerebral circulation.
Key wordsReflectance pulse oximetry fetal monitoring newborn arterial oxygen saturation
Unable to display preview. Download preview PDF.
- 3.Johnson N. Development and potential of fetal pulse oximetry. Contemp Rev Obstet Gynaecol 1991; 3: 193–200Google Scholar
- 6.Graaff R, Dassel ACM, Aarnoudse JG, Zijlstra WG, Heida P, de Mul FFM, Koelink MH, Greve J. Biophysical aspects of reflection pulse oximetry. In: Lafeber HN, Aarnoudse JG, Jongsma HW, eds. Fetal and neonatal physiological measurements. Amsterdam: Elsevier, 1991: 129–134Google Scholar
- 14.König V, Ullrich GJ, Huch A, Huch R. Reflection pulse oximetry — experience in Zürich. In: Lafeber HN, Aarnoudse JG, Jongsma HW. Fetal and neonatal physiological measurements. Amsterdam: Elsevier, 1991: 111–117Google Scholar
- 22.Graaff R, Dassel ACM, Koelink MH, de Mul FFM, Aarnoudse JG, Zijlstra WG. Optical properties of human dermisin vitro andin vivo. Appl Opt 1993; 32: 435–447Google Scholar
- 23.Giacometti L. The anatomy of the human scalp. In: Montagna W, ed. Advances in biology of skin, vol VI. Oxford: Pergamon Press, 1965: 97–120Google Scholar
- 24.Ryan TJ. Development of the cutaneous circulation. In: Bralow L, ed. Fetal and neonatal physiology. Philadelphia: WB Saunders Company, 1992: 555–565Google Scholar
- 26.Holbrook KA 1982 A histological comparison of infant and adult skin. In: Maibach HI, Boisits EK, eds. Neonatal skin structure and function. New York: Marcel Dekker, 1982: 24–31Google Scholar
- 28.Jöbsis-van der Vliet FF, Piantadosi CA, Sylvia AL, Lucas SK, Keizer HH. Near-infrared monitoring of cerebral oxygen sufficiency. Neurol Res 1988; 10: 7–17.Google Scholar
- 31.Martin RJ, Okken A, Rubin D. Arterial oxygen tension during active and quiet sleep in the normal neonate. J Pediatr 1979; 942: 271–274.Google Scholar
- 32.Mok JYQ, McLaughlin FJ, Pintar M, Hak H, Amaro-Galvez R, Levison H. Transcutaneous monitoring of oxygenation: What is normal? J Pediatr 1986; 108: 165–371.Google Scholar