The development of a versatile field program for measuring tritium in “real-time”
- 32 Downloads
Robust sample handling and liquid scintillation counting (LSC) procedures have been developed to routinely monitor tritium in the field relative to the 20,000 pCi/L drinking water standard. This procedure allows tritium to be monitored hourly during 24 hour drilling operations at depths in the saturated zone potentially contaminated by sub-surface nuclear weapons testing at the Nevada Test Site. Using retrofitted, shock hardened, vibration damped counters and strict analytical protocols, tritium may be measured rapidly in the field under hostile conditions. Concentration standards and “dead” tritium backgrounds are prepared weekly in a central laboratory and delivered to remote drilling locations where they are recounted daily as a check on counter efficiency and calibration. Portable LSC counters are located in trailers and powered off a battery pack and line filter fed by mobile generator. The samples are typically groundwaters mixed with drilling fluids returned after circulation through a drill string. Fluids are aerated and de-foamed, filtered, mixed with scintillation cocktail and dark adapted before counting. “Real-time” monitoring affords drilling and field personnel early warning against intercepting down-gradient plumes of radioactivity. For routine operations, the tritium activity may not exceed a 10,000 pCi/L threshold.
KeywordsDrilling Tritium Liquid Scintillation Counting Drilling Fluid Drinking Water Standard
Unable to display preview. Download preview PDF.
- 1.J.I. DREVER, The Geochemistry of Natural Waters, 2nd edition, Prentice-Hall, (1988) 437p.Google Scholar
- 2.M.L. DAVISSON, J.M. KENNEALLY, D.K. SMITH, G.B. HUDSON, G.J. NIMZ, J.H. REGO, Lawrence Livermore National Laboratory Report UCRL-ID-116122, (1994) 43p.Google Scholar
- 3.G.J. NIMZ, J.L. THOMPSON, U.S. Department of Energy Report DOE/NV-346, UC-703 (1992) 17p.Google Scholar
- 4.D.K. SMITH, Position Paper by the Nuclear Chemistry Division of the Lawrence Livermore National Laboratory to the U.S. Department of Energy, Nevada Operations Office (1994) 10p.Google Scholar