Correlation of accuracy and sensitivity in photoelectric measuring equipment during automatic location of the middle of a line
- 13 Downloads
KeywordsPhysical Chemistry Analytical Chemistry Measuring Equipment Automatic Location Photoelectric Measuring
These keywords were added by machine and not by the authors. This process is experimental and the keywords may be updated as the learning algorithm improves.
Unable to display preview. Download preview PDF.
- 1.S. F. Korndorf, Photoelectric Measuring Installations in Mechanical Engineering [in Russian], Mashinostroenie, Moscow (1965).Google Scholar
- 2.A. V. Mironenko, Photoelectric Measuring Systems [in Russian], Énergiya, Moscow (1967).Google Scholar
- 3.S. P. Éskin, “Installation for Checking Electrical Measuring Instruments,” Author's Certificate No. 194948, Byull. Izobr., No. 9 (1967).Google Scholar
- 4.I. S. Clark and A. N. Cook, “A photoelectric microscope for the measurement of linear scales,” J. Scient. Instr.,33, No. 9 (1956).Google Scholar
- 5.M. G. Mikhailova, “On the optimal consitions of operation of scanning devices with photoelectric recording,” Izmeritel'. Tekh., No. 1 (1963).Google Scholar
- 6.M. L. Brzhezinskii, D. I. Zorin, and V. D. Sverdlichenko, Photometric photoelectric microscope. Proc. of the Institutes of the Committee of Standards, No. 78 (138) (1965).Google Scholar
- 7.S. P. Éskin and V. I. Pritsker, “Photoelectric indicator for automatic determination of the middle of lines in dynamic regime,” Izmeritel'. Tekh., No. 7 (1968).Google Scholar
© Consultants Bureau 1971