The analysis of data obtained with time to amplitude converter and multichannel analyser systems
- 101 Downloads
An exact general formula is derived relating the data obtained from t.a.c-m.c.a. systems to the input signal distribution; it is shown how the latter is modified by the measuring process. A general method of data correction is given, together with important approximations, which evaluates the input signal distribution from the data without any loss of signal. A simple and accurate method of system calibration based only on counting and not requiring the use of calibrated delay lines is also given.
Index HeadingsPositron annihilation Analysis of t.a.c.-m.c.a. data
Unable to display preview. Download preview PDF.
- 1.R. E. Bell, R. L. Graham, A. G. Petch: Canad. J. Phys.30, 35 (1952)Google Scholar