The influence of errors in precipitation measurements on the accuracy of the evaporation measurements performed by a class A evaporation pan
Since uncorrected precipitation measurements are used for calculating the evaporation measured by a class A pan, precipitation measurement errors are introduced into the evaporation results. In defining the total error of the evaporation measurement it is necessary to consider the fact that the class A pan orifice and the rain gauge orifice are at different elevations. This paper presents a model to estimate the average value for the total error of evaporation. It is also shown that the evaporation measured by a class A pan is smaller than the actual evaporation by between 5 and 20%.
KeywordsPrecipitation Climate Change Waste Water Evaporation Measurement Error
Unable to display preview. Download preview PDF.
- Allerup, P., Madsen, H., 1980: Accuracy of point precipitation measurements.Nordic Hydrology 11, 57–70.Google Scholar
- Fedorova, E. G., 1966: Calculation of wind speed at the level of the precipitation gauge considering the degree of station protection (in Russian).Trans. Voyeykov Main Geophys. Observ. 195, 63–68.Google Scholar
- Mendel, O., 1986: Correction of measured precipitation data and their application in Central and Eastern Europe. Proc. Correction of Prec. Meas. (Zürich, April 1985): 167–174.Google Scholar
- Sevruk, B., 1986: Correction of precipitation measurements: Swiss experience. Proc. Correction of Prec. Meas. (Zürich, April 1985): 187–196.Google Scholar
- Sevruk, B., (ed.) 1989: Proc. Int. Workshop on Precipitation Measurement (St. Moritz, Dec. 1989).Google Scholar