Single-photon emission tomographic quantification in spherical objects: effects of object size and background
- 49 Downloads
A method was set up for single-photon emission tomographic (SPET) quantification of radioactivity concentration in small anatomical structures. The method is based on the theoretical model proposed by Kessler et al. (J. Comput Assist Tomogr 1984; 8: 514–522) describing the effects of spatial resolution (partial volume effect and spillover) on the quantification of radioactivity concentration in small spherical objects. The model was validated here in SPET, by phantom experimental measurements, in relation to object size and source/background contrast. Good agreement was found between model-predicted and SPET-measured radioactivity concentration ratios in hot spots in hot background experiments. Accuracy of the method was assessed for comparison of model-corrected and true radioactivity concentration ratios and was found to be within 8.5% over the full range of object size (9.4–36.5 mm). The good agreement found indicates that the model can be used to correct for partial volume effect and spillover in specific clinical situations, when the anatomical structure under study can be approximated by a sphere of known size (e.g. neuroreceptor and tumour studies). The method was applied to a representative SPET monoclonal antibody patient study for the quantification of radioactivity concentration in ocular melanoma.
Key wordsSingle-photon emission tomography Partial volume effect Spillover Recovery coefficient
Unable to display preview. Download preview PDF.
- 4.Laurelle M, Wallace E, Seibyl JP, Baldwin RM, Zea-Ponce Y, Zoghbi SS, Neumeyer JL, Charney DS, Hoffer PB, Innis RB. Graphical, kinetic, and equilibrium analysis of in vivo [123I]β-CIT binding to dopamine transporters in healthy human subjects.J Cereb Blood Flow Metab 1994; 14: 982–994.PubMedGoogle Scholar
- 8.Genna S, Smith AP. The development of ASPECT, an annular single crystal brain camera for high efficiency SPECT.IEEE Trans Nucl Sci 1988; NS-35: 654–658.Google Scholar
- 9.Zito F, Savi A, Fazio F. CERASPECT: a brain-dedicated SPECT system. Performance evaluation and comparison with the rotating gamma camera.Phys Med Biol 1993; 38: 1433–1442.Google Scholar
- 11.Chang LT. A method for attenuation correction in radionuclide computed tomography.IEEE Trans Nucl Sci 1978; 25: 638–643.Google Scholar
- 12.Oppenheim BE. Scatter correction of SPECT.J Nucl Med 1984; 25: 928–929.Google Scholar
- 16.Bendriem B, Dewey SL, Schlyer DJ, Wolf AP, Volkow ND. Quantitation of the human basal ganglia with positron emission tomography: a phantom study of the effect of contrast and axial positioning.IEEE Trans Med Imaging 1991; 10: 216–222.Google Scholar
- 19.Muller-Gartner HW, Links JM, Prince JL, Bryan RN, McVeigh E, Leal JP, Davatzikos C, Frost JJ. Measurement of radiotracer concentration in brain gray matter using positron emission tomography: MRI-based correction for partial volume effects.J Cereb Blood Flow Metab 1992; 12: 571–583.PubMedGoogle Scholar