Primary thermometry with nanoscale tunnel junctions
- 28 Downloads
We have found current-voltage (I-V) and conductance (dI/dV) characteristics of arrays of nanoscale tunnel junctions between normal metal electrodes to exhibit suitable features for primary thermometry. The current through a uniform array depends on the ratio of the thermal energy kBT and the electrostatic charging energy E c of the islands between the junctions and is completely blocked by Coulomb repulsion at T = 0 and at small voltages eV/2 ≤ Ec. In the opposite limit, kBT ≫ Ec, the width of the conductance minimum scales linearly and universally with T and N, the number of tunnel junctions, and qualifies as a primary thermometer. The zero bias drop in the conductance is proportional to T−1 and can be used as a secondary thermometer. We will show with Monte Carlo simulations how background charge and nonuniformities of the array will affect the thermometer.
PACS numbers73.40.Gk 07.20.Dt 73.40.Rw
Unable to display preview. Download preview PDF.
- 1.D.V. Averin and K.K. Likharev, inMesoscopic Phenomena in Solids, edited by B.L. Altshuler, P.A. Lee, and R.A. Webb (Elsevier, Amsterdam, 1991), p. 173.Google Scholar
- 2.Single Charge Tunneling, Coulomb Blockade Phenomena in Nanostructures, edited by H. Grabert and M.H. Devoret (Plenum Press, New York, 1992).Google Scholar
- 3.J.P. Pekola, K.P. Hirvi, J.P. Kauppinen and M.A. Paalanen,Phys. Rev. Lett. 73, 2903 (1994).Google Scholar
- 4.N.S. Bakhvalov, G.S. Kazacha, K.K. Likharev, and S.I. Serdyukova,Sov. Phys. JETP 68, 581 (1989)[Zh. Eksp. Teor. Fiz. 95, 1010 (1989)].Google Scholar
- 5.D.V. Averin and A.A. Odintsov,Phys. Lett. A 140, 251 (1989).Google Scholar
- 6.D.V. Averin and Yu. V. Nazarov, ch. 6, p. 217 in —Ref. .Google Scholar
- 7.J.G. Simmons, ch. 10, p. 135 and J.M. Rowell, ch. 27, p. 385 inTunneling Phenomena in Solids, edited by E. Burnstein and S. Lundquist (Plenum, New York, 1969).Google Scholar