Quantitative angioscopy: A novel method of measurement of luminal dimensions during angioscopy with the use of a “lightwire”
- 31 Downloads
To determine the accuracy and reproducibility of luminal dimension measurements of a newly developed method of quantitative angioscopy.
A method was developed for quantitation of luminal dimensions during angioscopy, as variation in magnification with lens-object distance and ambiguity associated with identification of corresponding points about the circumference of a given discrete cross-section render subjective estimates unreliable. A transverse ring of fiberoptically transmitted light was emitted from a guidewire or its housing at a known distance from the distal end of an angioscope and discrete cross-sections of interest were observed as the ring of light was reflected from the luminal surface. Caliper measurement of the diameter of the light ring image (< 50 mW at 488/515 nm), obtained on angioscopic video recordings of cylindrical phantom vessels of known dimensions, was performed by three observers on five occasions.
The mean absolute difference between measured and known luminal diameter (n = 405 observations) was 65 μm±35 μm and the mean coefficient of variation was 4.2%, and the mean difference between measured and known areas (n = 195 observations) was 0.4 mm2, with a mean coefficient of variation of 6.5%.
By use of this new lightwire method, luminal dimensions can now be measuredin vitro with a high degree of accuracy and reproducibility during angioscopy.
Key wordsAngioscopy Angiography Coronary artery disease Atherosclerosis
Unable to display preview. Download preview PDF.
- 1.Spears JR, Marais HJ, Serur J, Pomerantzeff O, Geyer R, Weintraub R, Thurer R, Gerstin R, Paulin S, Grossman W (1983) In vivo coronary angioscopy. J Am Coll Cardiol 1:1311–1314Google Scholar
- 2.Spears JR, Spokojny AM, Grossman W (1985) Coronary angioscopy during cardiac catheterization. J Am Coll Cardiol 6:93–97Google Scholar
- 3.Grundfest WS, Litvack F, Sherman T, Caroll R, Lee M, Chaux A, Kass R, Matloff J, Berci G, Swan HJ, Morgenstern L, Forrester J (1985) Delineation of peripheral and coronary detail by intraoperative angioscopy. Ann Surg 202:394–400Google Scholar
- 4.Lee G, Garcia JM, Corso PJ, Chan MG, Rink JL, Pichard A, Lee KK, Reis RL, Mason DT (1986) Correlation of coronary angioscopic to angiographic findings in coronary artery disease. Am J Cardiol 58:238–241Google Scholar
- 5.Sherman CT, Litvack F, Grundfest WS, Lee M, Hickey A, Chaux A, Kass R, Blanche C, Matloff J, Morgenstern L, Ganz W, Swan HJC, Forrester J (1986) Coronary angioscopy in patients with unstable angina pectoris. N Engl J Med 315:912–919Google Scholar
- 6.Sanborn TA, Rygaard JA, Westbrook BM, Lazar HL, McCormick JR, Roberts AJ (1986) Intraoperative angioscopy of saphenous vein and coronary arteries. J Thorac Cardiovasc Surg 91:339–343Google Scholar
- 7.Konishi T, Inden M, Nakano T (1989) Clinical experience with percutaneous coronary angioscopy in cases with coronary artery disease. Angiology 40:18–23Google Scholar
- 8.Ramee SR, White CJ, Collins TJ, Mesa JE, Murgo JP (1991) Percutaneous angioscopy during coronary angioplasty using a steerable microangioscope. J Am Coll Cardiol 17:100–105Google Scholar
- 9.White CJ, Ramee SR, Mesa JE, Collins TJ (1991) Percutaneous coronary angioscopy in patients with restenosis after coronary angioplasty. J Am Coll Cardiol 17:46B–49BGoogle Scholar
- 10.Siegel RJ, Ariani M, Fishbein MC, Chae JS, Park JC, Maurer G, Forrester JS (1991) Histopathologic validation of angioscopy and intravascular ultrasound. Circulation 84:109–117Google Scholar
- 11.Crawford DW, Brooks SH, Barndt R Jr, Blankenhorn DH (1977) Measurement of atherosclerotic luminal irregularity and obstruction by radiographic densitometry. Invest Radiol 12:307–313Google Scholar
- 12.Cheong W-F, Prahl SA, Welch AJ (1990) A review of the optical properties of biological tissues. IEEE J Quant Elect 26:2166–2185Google Scholar
- 13.Spears JR, Sandor T (1986) Quantitation of coronary stenosis severity: Limitations of angiography and computerized information extraction. In: Reiber JHC, Serruys PW (eds). State of the art in quantitative arteriography. Martinus Nijhoff Publishers Amsterdam, pp 103–124Google Scholar
- 14.Brown BG, Bolson EL, Frimer M, Dodge HT (1977) Quantitative coronary arteriography: Estimation of dimensions, hemodynamic resistance, and atheroma mass of coronary artery lesions using the arteriogram and digital computation. Circulation 55:329–337Google Scholar
- 15.Crawford DW, Brooks SH, Selzer RH, Brandt R Jr, Beckenbach ES, Blankenhorn DH (1977) Computer densitometry for angiographic assessment of arterial cholesterol content and gross pathology in human atherosclerosis. J Lab Clin Med 89:378–392Google Scholar
- 16.Alderman EL, Berte LE, Harrison DC, Sanders W (1981) Quantitation of coronary artery dimensions using digital image processing. Diagn Radiol 314:273–277Google Scholar
- 17.Spears JR, Sandor T, Als AV, Malagold M, Markis JE, Grossman W, Serur JR, Paulin S (1983) Computerized image analysis for quantitative measurement of vessel diameter from cineangiograms. Circulation 68:453–461Google Scholar
- 18.Mancini GBJ, Simon SB, McGillem MJ, Le Free MT, Friedman HZ, Vogel RA (1987) Automated quantitative coronary arteriography: Morphologic and physiologic validation in vivo of a rapid digital angiographic method. Circulation 75:452–460Google Scholar
- 19.Selzer RH, Hagerty C, Azen SP, Siebes M, Lee P, Shircore A, Blankenhorn DH (1989) Precision and reproducibility of quantitative coronary angiography with applications to controlled clinical trials. J Clin Invest 83:520–526Google Scholar
- 20.Reiber JHC (1991) An overview of coronary quantitation techniques as of 1989. In: Reiber JHC, Serruys PW (eds): Quantitative coronary arteriography. Kluwer Academic Publishers, Dordrecht, pp 55–132Google Scholar
- 21.Spears JR, Sandor T, Baim DS, Paulin S (1983) The minimum error in estimating coronary luminal cross-sectional area from cineangiographic diameter measurements. Cath Cardiovasc Diagn 9:119–128Google Scholar
- 22.Serruys PW, Reiber JH, Wijns W,van den Brand M, Kooijman CJ, ten Katen HJ, Hugenholtz PG (1984) Assessment of percutaneous transluminal coronary angioplasty by quantitative coronary angiography: Diameter versus densitometric area measurements. Am J Cardiol 54:482–488Google Scholar
- 23.Spokojny AM, Serur JR, Skillman J, Spears JR (1986) Uptake of hematoporphyrin derivative by atheromatous plaques: Studies in human in vitro and rabbit in vivo. J Am Coll Cardiol 8:1387–1392Google Scholar