Laboratory and home comparison of wrist-activity monitors and polysomnography in middle-aged adults
Accurate measurement of time at lights out is essential for calculation of several measures of sleep in wrist-activity monitors. While some devices use subjective reporting of time of lights out from a sleep diary, others utilise an automated proprietary scoring algorithm to calculate time at lights out, thereby negating the need for a sleep diary. This study aimed to compare sleep measures from two such devices to polysomnography (PSG) measures (In laboratory) and against each other when worn at home (At home). Fifty middle-aged adults from the Raine Study underwent overnight PSG during which they wore an ActiGraph™ and a Readiband™. They also wore both devices at home for 7 nights. The Readiband uses an automated proprietary algorithm to determine time at lights out whereas the ActiGraph requires completion of a sleep diary noting this time. In laboratory, compared to PSG: Readiband underestimated time at lights out, sleep onset, and wake after sleep onset, overestimated sleep latency and duration (p < 0.001 for all); while ActiGraph underestimated sleep latency and wake after sleep onset and overestimated sleep efficiency and duration (p < 0.001 for all). Similar differences between devices were observed on the laboratory night and when at home. In conclusion, an automated algorithm such as the Readiband may be used in the same capacity as the ActiGraph for the collection of sleep measures including time at sleep onset, sleep duration and time at wake. However, Readiband and ActiGraph measures of sleep latency, efficiency and wake after sleep onset should be interpreted with caution.
KeywordsActigraphy Algorithm Validation Polysomnography Sleep Technology
We acknowledge and thank the Raine Study participants, their families Raine study team for cohort management and data collection. Many thanks to Fatigue Science, Vancouver, British Columbia for the supply of Readibands™.
Compliance with ethical standards
Conflict of interest
Ian C Dunican and John A Caldwell have previously undertaken consultancy work for Fatigue Science, but neither are currently engaged in any capacity with the company. The Raine Study 22-year follow-up was supported by NHMRC Project Grants 10277449, 1021858, 1031617 and 1044840. Core funding for cohort management was provided by the University of Western Australia, the Telethon Institute for Child Health Research, Raine Medical Research Foundation, University of Western Australia Faculty of Medicine, Dentistry and Health, Women’s and Infant’s Research Foundation, Edith Cowan University and Curtin University. Professors Straker and Eastwood were funded by National Health and Medical Research Council of Australia (NHMRC) Senior Research Fellowships (1019980, 1042341).
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards. Informed consent was obtained from all individual participants included in the study and ethical approval was obtained from the University of Western Australia Human Research Ethics Committee.
- 1.Hirshkowitz M. The history of polysomnography: tool of scientific discovery. In: Chokroverty S, Billiard M, editors. Sleep medicine. New York: Springer; 2015.Google Scholar
- 5.Lee SJ. chapter 41 actigraphy, what is actigraphy? Amercian Academy of Sleep Technologists Wolters Kluwer, Lippincott Williams & Wilins; 2012. p. 474.Google Scholar
- 14.Russell CA CJ, Arand D, Myers LJ, Wubbels P, Downs H. Controlling fatigue risk with the fatigue science ReadiBand actigraph and associated sleep/wake classification algorithms: use case and validation. Fatigue Science: Fatigue Science. In: Archinoetics L, editors. 700 Bishop St, Suite 2000, Honolulu, HI 96813, Fatigue Science BS, Suite 2000, Honolulu, HI 96813, Kettering Health Networks Sleep Disorders Ctr SB, Kettering, OH 45429, San Diego: Vivonoetics HC; 2006. p.92117.Google Scholar
- 15.Bowen J, Hinze A, Cunningham SJ, Parker R. Evaluating Low-cost Activity Trackers for Use in Large-scale Data Gathering of Forestry Workers. Proceedings of the Annual Meeting of the Australian Special Interest Group for Computer Human Interaction; Parkville, VIC, Australia. 2838741: ACM; 2015. p. 474–82.Google Scholar
- 18.Noor ZM, Smith AJ, Smith SS, Nissen LM. Feasibility and acceptability of wrist actigraph in assessing sleep quality and sleep quantity: a home-based pilot study in healthy volunteers. Health (London). 2013;05(08):10.Google Scholar
- 20.Johnsons R. Mitigating sleep loss: assessment of omega-3 fatty acids. Carlsbad: DTIC Document; 2011.Google Scholar
- 21.Fullagar HH, Duffield R, Skorski S, White D, Bloomfield J, Kolling S, et al. Sleep, travel and recovery responses of national footballers during and following long-haul international air travel. Int J Sports Physiol Perform. 2015.Google Scholar
- 25.Conrad Iber SA-I, Andrew L, Chesson, Stuart F, Quan. The AASM manual for the scoring of sleep and asscioted events—rules ,terminology and technical specifications. 1st ed. Westchester: Illinois American Academy of Sleep Medicine; 2007.Google Scholar
- 26.Berry RB, Budhiraja R, Gottlieb DJ, Gozal D, Iber C, Kapur VK, et al. Rules for scoring respiratory events in sleep: update of the 2007 AASM manual for the scoring of sleep and associated events. Deliberations of the sleep apnea definitions task force of the american academy of sleep medicine. J Clin Sleep Med. 2012;8(5):597–619.PubMedPubMedCentralGoogle Scholar
- 29.Readiband FDA, Approval FS. FDA approval for Readiband-K111514-SBV2. Honolulu: Federal Drug Administration; 2011. p. 6.Google Scholar
- 31.Team RDC. R: A language and environment for statistical computing. R Foundation for Statistical Computing Vienna, Austria. 2016. http://www.r-project.org/. Accessed 20 Feb 2017.
- 32.Mantua J, Gravel N, Spencer RM. Reliability of sleep measures from four personal health monitoring devices compared to research-based actigraphy and polysomnography. Sensors (Basel). 2016;16(5).Google Scholar
- 33.M.J.T. American Academy of Sleep Medicine: International classification of sleep disorders: diagnostic and coding 698. 2nd ed. Westchester: American Academy of Sleep Medicine; 2005.Google Scholar
- 45.Gupta L, Morgan K, Gilchrist S. Does elite sport degrade sleep quality? A systematic review. Sports Med. 2016.Google Scholar