Acoustic Field of Phased-Array Ultrasound Transducer with the Focus/Foci Shifting
- 4 Downloads
High-intensity focused ultrasound (HIFU) is becoming popular in the treatment of solid tumors because of its non-invasiveness with few complications. The acoustic field is of importance in evaluating the safe focus shifting distance and determining the treatment plan.
The propagation of finite-amplitude acoustic wave from a 331-element HIFU phased-array with focus steering along and transverse to the transducer axis and 4-foci shifting on the focal plane was simulated using the angular spectrum approach (ASA) with a marching second-order operator-splitting scheme. In addition, the acoustic field produced by a truncated asymmetric transesophageal HIFU annular array was also simulated, and the effects of driving frequency and the number of concentric rings were investigated.
Because of the nonlinear effects, the peak negative pressure is lower than that of peak positive pressure at the main lobe but has a larger beam size. However, the peak positive and negative pressures at the grating lobe are quite similar. The effects of the focus shifting distances on the main and grating lobe (both acoustic pressure and − 6 dB beam size) were evaluated. With the focus shifting axially away from the transducer surface, the main lobe has decreased acoustic pressure by ~ 1.9 fold and increased beam size by ~ 4.5 fold while the grating lobe has the increased acoustic pressure by ~ 1.8 fold. The focus shifting laterally leads to the reduced pressure at the main lobe by ~ 1.4 fold but the slight decrease at the grating lobe by ~ 1.1 fold. In comparison, the shifting of multi-foci has similar influences on the main lobe but increases the pressure at the grating lobe. Driving frequency of annular array is found to have greater influences on the peak pressure and beam size.
Our algorithm can simulate the acoustic field of phased-array in arbitrary shape and optimize the transducer design, and the focus shifting distance and strategy should be selected appropriately for the safe HIFU exposure.
KeywordsHigh-intensity focused ultrasound (HIFU) Phased-array Nonlinear wave propagation Angular spectrum algorithm (ASA) Focus shifting
- 1.Zhou, Y. (2015). Principles and applications of therapeutic ultrasound in healthcare. CRC Press. ISBN 1466510285.Google Scholar
- 4.Auboiroux, V., Dumont, E., Petrusca, L., Viallon, M., & Salomir, R. (2011). An MR-compliant phased-array HIFU transducer with augmented steering range, dedicated to abdominal thermotherapy. Physics in Medicine and Biology, 56(12), 3563. https://doi.org/10.1088/0031-9155/56/12/008.CrossRefGoogle Scholar
- 10.Kuznetsov, V. P. (1971). Equations of nonlinear acoustics. Soviet Physics Acoustics, 16(4), 467–470.Google Scholar
- 11.Zabolotskaya, E. A., & Khokhlov, R. V. (1969). Quasi-plane waves in the nonlinear acoustics of confined beams. Soviet Physics Acoustics, 15(1), 35–40.Google Scholar
- 16.Doinikov, A. A., Novell, A., Calmon, P., & Bouakaz, A. (2014). Simulations and measurements of 3-D ultrasonic fields radiated by phased-array transducers using the Westervelt equation. IEEE Transactions on Ultrasonics, Ferroelectrics, and Frequency Control, 61(9), 1470–1477. https://doi.org/10.1109/tuffc.2014.3061.CrossRefGoogle Scholar
- 17.Kreider, W., Yuldashev, P. V., Sapozhnikov, O. A., Farr, N., Partanen, A., Bailey, M. R., et al. (2013). Characterization of a multi-element clinical HIFU system using acoustic holography and nonlinear modeling. IEEE Transactions on Ultrasonics, Ferroelectrics, and Frequency Control, 60(8), 1683. https://doi.org/10.1109/tuffc.2013.2750.CrossRefGoogle Scholar
- 19.Seip, R., Sanghvi, N. T., Uchida, T., & Umemura, S.-I. (2001). Comparison of split-beam transducer geometries and excitation configurations for transrectal prostate HIFU treatments. In IEEE ultrasonics symposium (pp. 1343–1346). https://doi.org/10.1109/ultsym.2001.991969.
- 20.Constanciel, E., N’Djin, W. A., Bessiere, F., Chavrier, F., Grinberg, D., Vignot, A., et al. (2013). Design and evaluation of a transesophageal HIFU probe for ultrasound-guided cardiac ablation: simulation of a HIFU mini-maze procedure and preliminary ex vivo trials. IEEE Transactions on Ultrasonics, Ferroelectrics, and Frequency Control, 60(9), 1868–1883. https://doi.org/10.1109/tuffc.2013.2772.CrossRefGoogle Scholar
- 25.Ji, X., Bai, J.-F., Shen, G.-F., & Chen, Y.-Z. (2009). High-intensity focused ultrasound with large scale spherical phased array for the ablation of deep tumors. Journal of Zhejiang University SCIENCE B: Biomedicine and Biotechnology, 10(9), 639–647. https://doi.org/10.1631/jzus.b0920130.CrossRefGoogle Scholar
- 38.Dupenloup, F., Chapelon, J. Y., Cathignol, D. J., & Sapozhnikov, O. (1996). Reduction of the grating lobes of annular arrays used in focused ultrasound surgery. IEEE Transactions on Ultrasonics, Ferroelectrics, and Frequency Control, 43(6), 991–998. https://doi.org/10.1109/58.542044.CrossRefGoogle Scholar