Medical Devices: Classification and Analysis of Faults Leading to Harms
- 23 Downloads
Harms from medical devices are important, but have been much less well studied than adverse drug reactions. Information provided to device users is of variable quality.
Our aim was to define “medical device fault” and “adverse effect of a medical device”; to establish whether medical device faults arise in design, manufacture, or use; and to consider ways of mitigating the adverse effects of medical devices.
We analysed 100 consecutive faults reported by the US Food and Drug Administration (FDA) and 50 faults reported by the UK Medicines and Healthcare products Regulatory Agency (MHRA), and classified faults according to the point at which they occurred.
Nearly 70% of reported faults related to devices that entered the body. Over 70% arose at the design stage, a quarter of the faults were associated with manufacture, and less than 5% were primarily caused by faulty use.
We defined a medical device fault as an unintended failure in the design, manufacture, or use of a medical device that leads to, or has the potential to lead to, harm to the patient, and an adverse effect of a medical device as an unintended and appreciably harmful effect, caused by a medical device, which demonstrates a hazard of the device and may warrant preventive measures, or a change in the mode of use, or withdrawal of the device. Most faults that generate warnings arise from problems at the design stage, some arise at the manufacturing stage, and a few in usage. Careful assessment of the design of safety-critical devices in the light of previous problems may help to prevent repetition of errors. It would be helpful if, in addition to user manuals, manufacturers were required to produce Summaries of Device Characteristics (SDCs, “labels”) that contained a systematically presented set of information about a product.
We are grateful to the International Society of Pharmacovigilance (ISoP) for allowing us to organize a symposium on medical devices, held at the ISoP 2018 meeting in Geneva, on which this and a companion paper are based.
Compliance with Ethical Standards
This work was unfunded.
Conflict of interest
Neither Robin Ferner nor Jeffrey Aronson has any relevant financial conflict of interest.
- 3.European Medicines Agency. Medication errors. https://www.ema.europa.eu/en/human-regulatory/post-authorisation/pharmacovigilance/medication-errors. Accessed 25 Oct 2019.
- 5.Medicines and Healthcare products Regulatory Agency (MHRA). Alerts and recalls for drugs and medical devices. https://www.gov.uk/drug-device-alerts. Accessed 25 Oct 2019.
- 6.US Food and Drug Administration. 2018 medical device recalls. https://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm590900.htm. Accessed 25 Oct 2019.
- 7.US Food and Drug Administration. MAUDE—Manufacturer and User Facility Device Experience. https://www.accessdata.fda.gov/scripts/cdrh/cfdocs/cfMAUDE/search.CFM. Accessed 25 Oct 2019.
- 8.US Food and Drug Administration. Global Unique Device Identification Database. https://www.fda.gov/medical-devices/unique-device-identification-system-udi-system/global-unique-device-identification-database-gudid. Accessed 25 Oct 2019.
- 9.US Food and Drug Administration. Draeger Medical Systems, Inc. Jaundice Meter JM-103 and Jaundice Meter M-105 recalled due to misinterpretation of display messages for out of range values. https://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm611564.htm. Accessed 25 Oct 2019.
- 14.Anonymous. Blood glucose meters-recalled because of error reading. WHO Newslett 1998; 9/10: 16.Google Scholar
- 15.US Food and Drug Administration. Newport Medical Instruments Inc. Recalls Newport™ HT70 & Newport™ HT70 Plus ventilators due to a software issue which may cause an unexpected shutdown without alarm. https://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm556222.htm. Accessed 25 Oct 2019.
- 16.Medical Device Alert. MDA/2018/014. Issued: 02 May 2018 at 15:00. Valid until: May/2019. Infinity Acute Care System and M540 Patient Monitors software versions VG2.2-VG6.0—risk that alarms are not activated. https://assets.publishing.service.gov.uk/media/5ae99e7fed915d42f42b6104/MDA-2018-014_Final.pdf. Accessed 25 Oct 2019.
- 17.US Food and Drug Administration. Draeger Medical Systems, Inc. Jaundice Meter JM-103 and Jaundice Meter JM-105 recalled due to misinterpretation of display messages for out of range values. https://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm611564.htm. Accessed 25 Oct 2019.
- 18.US Food and Drug Administration. Terumo recalls Sarns™ TCM and TCM II Cooling and Heating Systems and HX2™ Temperature Management Systems due to revised cleaning instructions. https://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm606785.htm. Accessed 25 Oct 2019.
- 19.Medical Device Alert MDA/2018/023. Issued: 05 July 2018 at 13:00. Valid until: July 2019. Combur10 Test UX and Chemstrip 10 A test strips—risk of falsely low results when measuring test strips on the Urisys 1100 urine analyser. https://assets.publishing.service.gov.uk/media/5b3e0251e5274a377794502a/MDA-2018-023_Final.pdf. Accessed 25 Oct 2019.
- 20.US Food and Drug Administration. Zimmer Biomet, Inc. Recalls Spinal Fusion and Long Bone Stimulators due to lack of adequate validation and controls to ensure product cleanliness. https://www.fda.gov/medical-devices/medical-device-recalls/zimmer-biomet-inc-recalls-spinal-fusion-and-long-bone-stimulators-due-lack-adequate-validation-and. Accessed 25 Oct 2019.
- 21.Schatzmann M. Soul murder. London: Allen Lane; 1973.Google Scholar
- 22.House of Commons Select Committee on Patent Medicines. Section 16 Legal and official anomalies and curiosities. In: Report from the Select Committee on Patent Medicines. London: HMSO, 1914.Google Scholar
- 23.US Food and Drug Administration. Cook CloverSnare 4-Loop Vascular Retrieval Snare—snare tip may break during use. http://wayback.archive-it.org/7993/20170112083828/http://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm412778.htm. Accessed 26 Oct 2019.
- 24.US Food and Drug Administration. Edwards Lifesciences LLC. recalls Certitude Delivery System due to mold overflow defect which may obstruct blood flow. https://www.fda.gov/medical-devices/medical-device-recalls/edwards-lifesciences-llc-recalls-certitude-delivery-system-due-mold-overflow-defect-which-may. Accessed 26 Oct 2019.
- 25.MHRA. Various trauma guide wires—risk of infection due to packaging failure (MDA/2018/032). https://www.gov.uk/drug-device-alerts/various-trauma-guide-wires-risk-of-infection-due-to-packaging-failure-mda-2018-032. Accessed 26 Oct 2019.
- 26.MAUDE Adverse Event Report: Bayer Medical Care, Inc. Stellant CT syringe kit angiographic syringe kit. https://www.accessdata.fda.gov/scripts/cdrh/cfdocs/cfMAUDE/detail.cfm?mdrfoi__id=7878313&pc=DXT. Accessed 26 Oct 2019.
- 27.US Food and Drug Administration. MAUDE Adverse Event Report. Siemens Healthcare Diagnostics Inc. Advia Chemistry Creatinine_2 Reagent. https://www.accessdata.fda.gov/scripts/cdrh/cfdocs/cfmaude/detail.cfm?mdrfoi__id=5123970&pc=CGX. Accessed 26 Oct 2019.
- 28.US Food and Drug Administration. MAUDE Adverse Event Report: Oberdorf Synthes Produktions GMBH. Matrix lock screw Ø1.85 self-drill l6 TA screw, fixation, bone. https://www.accessdata.fda.gov/scripts/cdrh/cfdocs/cfMAUDE/detail.cfm?mdrfoi__id=7906393&pc=HWC. Accessed 26 Oct 2019.
- 29.Dilks K. Regulation 28 Report to Prevent Future Deaths. https://www.judiciary.uk/wp-content/uploads/2018/03/Sheila-Hynes-2017-0448_Redacted.pdf. Accessed 26 Oct 2019.
- 31.Medicines and Healthcare products Regulatory Agency (MHRA). SureSigns VS & VM patient monitors and viewing stations manufactured before May 2018: risk of batteries overheating or igniting (MDA/2018/031). https://www.gov.uk/drug-device-alerts/suresigns-vs-vm-patient-monitors-and-viewing-stations-manufactured-before-may-2018-risk-of-batteries-overheating-or-igniting-mda-2018-031. Accessed 26 Oct 2019.
- 32.US Food and Drug Administration. Unintended heating associated with the Monteris Medical NeuroBlate Probe: letter to health care providers. https://www.fda.gov/medicaldevices/safety/letterstohealthcareproviders/ucm602253.htm. Accessed 26 Oct 2019.
- 37.Regulation (EU) 2017/745 of the European Parliament and of the Council of 5 April 2017 on medical devices, amending Directive 2001/83/EC, Regulation (EC) No 178/2002 and Regulation (EC) No 1223/2009 and repealing Council Directives 90/385/EEC and 93/42/EEC. https://eur-lex.europa.eu/legal-content/EN/TXT/PDF/?uri=CELEX:32017R0745&from=EN. Accessed 26 Oct 2019.
- 38.International Conference on Harmonisation of technical requirements for registration of pharmaceuticals for human use. ICH harmonised tripartite guideline. Good manufacturing practice guide for active pharmaceutical ingredients. Q7. 10 November 2000. https://database.ich.org/sites/default/files/Q7_Guideline.pdf. Accessed 26 Oct 2019.
- 39.International Medical Device Regulators Forum. About IMDRF. http://www.imdrf.org/about/about.asp. Accessed 26 Oct 2019.
- 41.EUDAMED: European Database on Medical Devices. http://ec.europa.eu/idabc/en/document/2256/5637.html. Accessed 26 Oct 2019.