Functionalised microwave sensors for real-time monitoring of copper and zinc concentration in mining-impacted water
Microwave spectroscopy has been identified as a novel and inexpensive method for the monitoring of water pollutants. Integrating microwave sensors with developed coatings is a novel strategy to make the sensing system more specific for a target contaminant. This study describes the determination of copper and zinc concentration in water in both laboratory-prepared and acquired mine water samples from two abandoned mining areas in Wales, UK. Uncoated sensors immersed in samples spiked with 1.25 mg/L concentrations of copper and zinc, using the standard addition method, were able to quantify the concentration at 0.44 GHz with a strong linear correlation (R2 = 0.99) for the reflection coefficient magnitude (|S11|). Functionalised microwave sensors with l-cysteine, chitosan and bismuth zinc cobalt oxide-based coatings have shown improvement in the sensing performance. Specifically, the linear correlation at 0.91–1.00 GHz between |S11| and a polluted water sample spiked with Cu showed a higher (R2 = 0.98), sensitivity (1.65 ΔdB/mg/L) and quality factor (135) compared with uncoated sensors (R2 = 0.88, sensitivity of 0.82 ΔdB/mg/L and Q-factor 30.7). A Lorentzian peak fitting function was applied for performing advanced multiple peak analysis and identifying the changes in the resonant frequency peaks which are related to the change in metal ion content. This novel sensor platform offers the possibility of in situ monitoring of toxic metal concentrations in mining-impacted water, and multiple peak features, such as area, full width half maximum, centre and height of the peaks, have the possibility to offer higher specificity for similar toxic metals, as between copper and zinc ions.
KeywordsPlanar electromagnetic sensors Coatings Standard addition method Chelating polymers Lorentzian peak curve fitting Heavy metals
Metal pollution and monitoring challenges
Mining is an important contributor to global wealth (Younger et al. 2002). However, metal and mineral extraction generates huge quantities of waste rich in potentially toxic elements, including metals, that can impact the ecosystem and human health (Byrne et al. 2018; Wolkersdorfer 2008). The major mechanisms associated with the mobilisation of metal ions in mining areas are: (i) the oxidation and consequent hydrolysis of sulphide minerals in mine waters (Johnson 2003), (ii) the leaching from deposits of mine tailings (Perkins et al. 2016) and (iii) fluvial transport of mobilised metals from headwater catchments to coastal areas (Mayes et al. 2013).
Copper (Cu) and zinc (Zn) are among the most common toxic elements associated with mine wastes and have been widely documented to cause significant pollution of water resources (Environment Agency 2008). They are essential elements for living organisms, but high concentrations, especially with long-term exposure, can generate significant health problems, such as respiratory, gastrointestinal and neuronal disorders (Agency for Toxic Substances and Disease Registry (ATSDR) 2004). For this reason, the EU Water Framework Directive (EU WFD) and the US Environmental Protection Agency (US EPA) have developed environmental quality standards (EQS) that specify that the concentration of dissolved Cu and Zn in freshwater should not exceed 28–34 µg/L and 125–210 µg/L, respectively (UK Technical Advisory Group on the Water Framework Directive 2008; United States Environmental Protection Agency 1986). Globally, Cu and Zn concentrations in mine water range from < 0.01 to 10 mg/L (Byrne et al. 2012). In the UK, 9% of rivers in England and Wales are at risk of exceeding this threshold (Environment Agency 2008). For instance, Parys Mountain copper mine, in Anglesey, is the single largest contributor of Cu to the Irish Sea, with Cu concentrations in excess of 3–20 mg/L (Dean et al. 2013). Thousands of abandoned mining areas need to be monitored in more detail for better prioritisation of effective remedial actions (Environment Agency 2012).
The current standard methods for quantifying the concentration of toxic metals in water are laboratory-based and involve collected samples being analysed by inductively coupled plasma optical emission spectrometry (ICP-OES), inductively coupled plasma mass spectroscopy (ICP-MS), atomic absorption spectroscopy (AAS) and neutron activation analysis (NAA) (Bansod et al. 2017). Although these techniques are highly sensitive and versatile in terms of the simultaneous determination of multielements in complex matrices, they are not able to provide in situ, real-time and low-cost metal concentration data required by regulators and mining companies to monitor and protect water resources (Shukla et al. 2018).
There have been major advancements recently in technologies for in situ metal concentration monitoring, mostly based on the following technologies: voltammetry (Holmes et al. 2018; Wu et al. 2019), aptasensors (Varun and Daniel 2018), near infrared (Iqbal et al. 2017), fluorescence and colorimetric (Guo et al. 2019; Tan et al. 2016), colorimetric microfluidic (Xie et al. 2019), optical fibre-based sensors (Kopitzke and Geissinger 2014). Among existing commercial anodic stripping voltammetry (ASV)-based analysers, there are PDV6000ultra and Metalyser® Portable HM1000, developed by Trace2O and Modern Water, respectively, that offers portability, multimetal detection and relative fast measurement time (Barton et al. 2016). However, they require sample collection and pre-concentration, so they are not able to detect toxic metals in water directly for in situ measurements (Holmes et al. 2018). Consequently, they cannot be installed as part of a remote monitoring system (Cases-Utrera et al. 2015).
One of the biggest challenges is therefore to develop novel technologies able to detect toxic metals continuously to minimise the impact of mining by increasing the scientific understanding of transport processes and by providing early warning of pollution episodes (National Academies of Sciences and Medicine 2018).
Microwave technology and water
Microwave spectroscopy has emerged in recent years as a novel monitoring technique in the food industry (Mason et al. 2016), healthcare (Wang 2018), sports science (Greene et al. 2019), structural analysis (Zarifi et al. 2018a) and water quality control (Korostynska et al. 2014). This wide range of applications is due to the adaptability of the sensing structure, which can include resonant cavities, horn antennas, transmission lines, waveguides, flexible and planar resonant sensors, depending on the form of the material under test (MUT).
Current research has demonstrated the possibility to determine the presence and concentration of specific components and pollutants in water, e.g. lead ions (Pb2+) (Frau et al. 2018a), sugars (Harnsoongnoen et al. 2018), nitrates (Cashman et al. 2017), phosphate (Nag et al. 2019), antibiotics (Mason et al. 2018), silver materials (Ateeq et al. 2017), water in fuel (Andria et al. 2019), marine algae (Moejes et al. 2018), liquid mixtures, as acetone and water (Wei et al. 2018), ethanol and water (Salim et al. 2018; Wiltshire and Zarifi 2018).
The detection principle is based on the rotation and polarisation of the water molecules and other chemical compounds when an electromagnetic field is applied to a sample under test through a sensing structure. The reflected microwave signal varies with frequency due to the interaction of the electric field with the MUT because its conductivity and permittivity are frequency dependent. The electrical conductivity is the material’s ability to conduct a current, while the permittivity is a measure of how an electric field affects a dielectric medium. The permittivity (ε) is a complex value characterised by a real (ε′) and an imaginary (ε″) component (Korostynska et al. 2013). ε′ depends on how electrical energy is stored by the material, while ε’’ represents how this energy can be dissipated as heat (Jilani et al. 2012). The electromagnetic sensors output can be presented in terms of S-parameters. Measurements for a one-port configuration (e.g. reflection coefficient magnitude, as |S11|) provide a spectral response that depends on the composition and concentration of the MUT.
Some of the advantages of this technique compared with optical and electrochemical methods are (i) the immediate measurement response when a water sample is in contact with the EM field; (ii) its noninvasiveness, considering this is a field-propagation-based technique (Kölbl et al. 2018); (iii) its low power (1 mW) and measurement costs. Some of the disadvantages are related to the capability to detect every minor change in the MUT measurement, such as temperature and density (Al-Kizwini et al. 2013). Consequently, researching the causes of every change and assessing how to correct for them are fundamental steps for the application of this technology in the real world.
Between the numerous possible resonant structures, planar sensors have the potential to give high sensitivity and accuracy (Rahman et al. 2017). Moreover, they have the advantages of small size (from mm to cm), robustness and low-price fabrication. They are portable, light and practical for in situ and continuous monitoring, compared with resonant cavities and can be easily implemented with materials, as thin and thick films, microfluidic structures, etc. (Zarifi and Daneshmand 2016).
For a sensing platform to deliver high sensitivity (to detect low Cu concentration) and selectivity (to distinguish between similar metal ions), the integration of interface materials on planar sensors, such as polymers, nanotube, nanoparticles, nanorods, metamaterials, biomaterials, is a novel, attractive approach for microwave and impedance spectroscopy (Azmi et al. 2017; Ebrahimi et al. 2014; Zarifi et al. 2018b). The addition of materials on the sensing surfaces has been recognised as being able to improve resolution and specificity (Chen et al. 2012).
These functionalised electromagnetic (f-EM) sensors determine a more specific response thanks to the particular chemical or physical interactions between the materials and the target analyte under test. Screen-printing technology can be used to place thick films on planar EM sensors. The produced coatings have the advantage of being repeatable, reusable and can be prepared using a mixture of various materials, increasing the specific detection of compounds in complex water matrices (Frau et al. 2019b).
Polymers are able to improve the specificity, especially integrated with metal oxides (Graunke et al. 2018). Chelating polymers are able to interact with Cu ions, thanks to their hydroxyl and amine groups, which are reactive to Cu ions (Koneswaran and Narayanaswamy 2009; Ngah and Fatinathan 2008). Metal oxides have high surface area, reactive sites and are able to adsorb toxic metals (Ugwu and Igbokwe 2019).
The aims of this paper are: (1) to demonstrate the feasibility of using microwave spectroscopy to quantify changes in Cu and Zn concentration in water using the standard addition method; (2) to evaluate the sensitivity for Cu ion concentration using the f-EM sensors based on mixtures of l-cysteine, chitosan and bismuth cobalt zinc oxide; (3) to evaluate the selectivity for Cu and Zn as changes in spectral responses by applying a Lorentzian peak fitting function for multiple parameter peak analysis; (4) to measure mining-impacted water samples collected in stream water in the UK.
To achieve the investigation aims, this novel research was carried out between May 2018 and June 2019 in the engineering and chemical laboratories at Liverpool John Moores University (Liverpool, UK), where f-EM sensors and samples were prepared. Polluted samples were acquired from two mining areas in Wales (UK) during July 2018. These were then tested to evaluate the potential of this novel technology.
Materials and methods
Thick film preparation
The f-EM sensors (Fig. 1b) were cured in an oven at 150 °C for 1 h between each of the 6 prints, before soldering to an SMA connector. The final thickness of the film, measured using a surface profiler (Taylor Hobson, Form Talysurf 120), was 65 ± 5 µm. The connectors were then waterproofed using a thermoplastic adhesive, consisting of ethylene–vinyl acetate (EVA) and terpene-phenol resin (TPR) (internal part), and silicone (external part) and embedded in 50-mL centrifuge tube lids.
The coatings were characterised before and after their interaction with Cu-contaminated water samples using a scanning electron microscope (SEM, model FEI, Quanta 200) and an energy-dispersive spectroscopy (EDS, model INCA-X-act, Oxford Instruments), connected to the SEM.
Water sample preparation and collection
Cu and Zn solutions were prepared using copper and zinc standards for ICP (68921 and 18562 Sigma-Aldrich, in 2% nitric acid, prepared with high-purity Cu and Zn metal, HNO3 and water) by adding the required concentration in deionised water (DW). Simulated “unknown” samples with 3 (UNK3) and 0.2 (UNK0.2) mg/L of Cu and Zn, respectively, were prepared. Then, they were spiked using the standard addition method (as described in section Adapted standard addition method) to quantify the concentration of the spiked Cu and Zn samples in DW.
Four polluted samples (Frongoch Adit, Mill Race, Nant Ceunant and Parys Mountain Adit; respective acronyms: FA, MR, NC, PM) from two mining-impacted areas in Central (Wemyss mine) and North Wales (Parys Mountain) were collected and analysed. Rivers draining both mine areas have well-documented water quality problems and fail to meet EQS for Cu (Parys Mountain) and Zn (Parys Mountain and Wemyss) (Dean et al. 2013; Onnis et al. 2018).
Cu and Zn concentrations for these samples were analysed using an inductively coupled plasma mass spectrometry (ICP-MS, model 7900 Agilent Technologies). Electrical conductivity (EC), pH and temperature were measured using a multiparameter metre (model PCE-PHD 1, PCE Instruments). Temperature was also constantly monitored using a digital and a noncontact infrared thermometer (model TM-902C Lutron and 830-T2 Testo, respectively), and the room was kept at a constant temperature of 20 °C.
Adapted standard addition method
The standard addition method is a quantitative method for determining the concentration of an analyte in an unknown aqueous solution. The principle is based on the addition of the pollutant under test by multiple small and equal volumes to minimise the matrix effect which can interfere with the target analyte signal (Bader 1980). It is particularly important in the areas of instrumental methods, such as spectrophotometry, atomic absorption (AA) and electrochemistry, especially when complex matrices are analysed (Harris 2007). Using a linear calibration range, the response signal is directly proportional to the concentration (Bruce and Gill 1999). In this work, the standard addition method has been utilised with microwave spectroscopy for the first time.
The standard method was slightly modified, as the addition was performed continuously on the same sample, with the purpose of being able to evaluate the effective real-time change in the microwave responses related to the increase in the Cu content and to avoid small variations given by sample preparation. The change in |S11| response was monitored continuously. Once the resonant frequencies in the spectral response (between 0.01 and 3 GHz) were selected for Cu and Zn, a calibration curve was produced by using the |S11| signal for the known quantities of the chemical of interest. The ΔS11 of the blank response (the DW signal) was subtracted, and the linear regression was then plotted. The concentration of the unknown sample is determined using the slope and the intercept, which corresponds to the point at which the calibration plot crosses zero on the x-axis.
In this work, 40 mL of laboratory “unknown” samples, with concentrations of 3 (UNK3) and 0.2 (UNK0.2) mg/L of Cu and Zn, respectively, and an acquired sample (Frongoch Adit, FA) were spiked by adding 0.5 mL of 100 mg/L Cu or Zn solution which corresponded to an increase of 1.25 mg/L in metal concentration. DW samples were also spiked likewise with Cu, Zn and Cu + Zn (1:1) solutions. Other samples (FA + 0.2 mg/L of Cu and Zn) were prepared by dilution for measuring the decrease in concentration using the F-EM sensors.
The spectral responses for the polluted water samples consist of a combination of peaks and troughs, both defined as peaks in this work. For more accurate identification of a specific metal in water sample under test, the full spectral response was analysed by fitting the |S11| responses using peak functions. This enabled the determination of multiple features for each individual peak, such as area, centre and height of the peaks. Moreover, if the spectral response is convoluted, with peaks which are superimposed upon one another, curve fitting is the best way to isolate peaks and define their specific characteristic (Meier 2005).
Comparison of peak fitting function performances
0.96 ± 0.02
0.99 ± 0.01
0.97 ± 0.01
0.27 ± 0.08
0.06 ± 0.03
0.18 ± 0.08
Consequently, the |S11| responses for the tested samples were fitted with the Lorentzian peak fitting function and multiple peak analyses were performed.
Results and discussion
Uncoated sensors and standard addition method
Notably, the resonant peaks that were produced at frequencies < 0.5 GHz have a lower dB as the metal concentration is increasing (the reflection decreases) compared to the peaks at frequencies > 0.5 GHz, where inversely a higher dB is noticed as the concentration increases (the reflection increases), for both linear and polynomial correlations. For instance, the peak located at 0.44 GHz is − 4.2 dB for UNK3 + 1.25 mg/L of Cu and − 7.2 dB for UNK3 + 6.25 mg/L; at 0.76 GHz the reflection coefficient is − 11.9 dB for UNK3 + 1.25 mg/L of Cu and − 9.8 dB for UNK3 + 6.25 mg/L.
By plotting the linear correlation, the concentrations of Cu and Zn were determined, confirming the fact that microwaves are able to measure the concentration of Cu and Zn continuously using the reflection coefficient, |S11|. This value corresponds to the x-intercept of the linear correlation, resulting in 2.99 mg/L and 0.160 mg/L for UNK3 (Cu) and UNK0.2 (Zn) samples.
This study demonstrates the feasibility of the microwave response to detect the increase in metal content in water and to quantify its change. However, this experiment demonstrates the necessity of finding a solution for improving the specificity, as sensitivity and selectivity for metal ions detection, considering that the calibration curve is almost identical, demonstrating the need to find novel strategies for improving the discrimination between Cu and Zn. Although the concentration of these toxic metals was determined using a sensor immersed in water simulating the in situ monitoring, it is important to obtain a higher selectivity, considering that Zn and Cu have different toxicities in the environment.
The uncoated planar sensors used in this paper, specifically eight-pair gold IDE on PTFE substrates, have been shown to be able to distinguish changes in the cell concentration of Tetraselmis suecica (Moejes et al. 2018) at 4 GHz, and changes in Pb ions, detected at 2.5 GHz (Frau et al. 2018a). However, this sensing structure alone is not reliable for distinguishing such similar pollutants in water as Cu and Zn.
L-CyChBCZ coatings, standard addition method and field samples
More complex solutions were spiked using the standard addition method for evaluating the feasibility of using microwave rotational spectroscopy in the real environment by spiking a freshwater sample (FA) with Cu in the same manner as described in section Adapted standard addition method. These solutions were analysed using uncoated and coated gold IDE sensors with L-CyChBCZ-based films, for dipping in water.
Uncoated and coated sensors: a comparison
Moreover, the peaks produced at frequencies between 0.92–1.00 GHz and 1.8–2.3 GHz (indicated by the dashed violet arrows in Fig. 4b) were characterised by a bigger change due to the microwave signal reflection, with a higher dB variation (ΔdB) for Cu concentrations. This is probably caused by (i) a change in dielectric properties caused by the interaction of the Cu ions with the sensing material; (ii) the thickness of the coating itself (Chen et al. 2012).
Summary of statistical information for 3 identified peaks (at frequency < 1 GHz) using uncoated (UNC) and f-EM sensors based on L-CyCHBCZ for a mine water sample (Frongoch Adit, FA) spiked with Cu using the standard addition method
The linearity for peak 0 was higher (R2 = 0.97) using the uncoated sensor. Also, the sensitivity (indicated by a steeper gradient) and Q-factor for both peaks 0 and 1 are higher using the uncoated sensors, with a smaller coefficient of variation and LOD compared with the f-EM sensors. However, the efficacy of the f-EM sensor at peak 2 (highlighted in bold in Table 2) is much higher compared with the uncoated sensor, with a higher linearity (R2 = 0.98), sensitivity (1.651 dB changes for each mg/L change in Cu concentration) and Q-factor (135.5). Moreover, the low LOD (0.036 mg/L) demonstrates the prospect of detecting the Cu concentration in mine water just above the EQS established as the “safe limit” in freshwater (28–34 µg/L). Consequently, to obtain more precise measurements, to benefit from the good performance of each selected peak and to a more specific peak characterisation, can be advantageous to perform a multipeak analysis.
Lorentzian peak fitting function
Six distinct peaks were identified for each sample. ΔS11 that is related to the small change in response at 0.01 GHz is considered “peak 0” as it is not a wide-ranging peak, and it was not possible to analyse all its features. Through this novel strategy it is possible to not only identify and characterise all the peaks, including those that overlapped (e.g. peaks 3, 4 and 5, Fig. 5), but also to investigate more parameters than just the centre of the peak (xc) and its intensity (H), such as full width at half maximum (w) and the area of the peak (A). The fitting function converged and a Chi-square tolerance value of 1 × 10−6 was obtained.
Notably, the convoluted peak between 1.25 and 1.80 GHz encloses three distinct peaks which makes the apparent peak look irregular. Overlapping of individual peaks is common in vibrational spectroscopy such as middle infrared and Raman frequencies (Meier 2005). For microwave spectroscopy and f-EM planar IDE sensors this “phenomenon” mostly occurs at frequencies higher than 1 GHz.
FWHM, A and H present similar behaviours for the 6 peaks (Fig. 6b–d). Notably, peak 5, which was part of the convoluted peak, was not distinguished for concentrations of Cu higher than + 5 mg/L. Particularly, it is interesting as the values decrease for peak 1 located at 0.47 GHz when the Cu concentration increases and increase for peak 2 (0.92–1.00 GHz). Hence, peaks 2 and 6 drift to lower frequency and decrease the |S11| value when the Cu concentration increases. The analysis of the various parameters measured for DW (grey arrows show some example), compared with the FA samples and its Cu additions, will help to identify the changes in spectral response that are related to the matrix composition.
Cu vs Zn
The results demonstrate the capability of the proposed f-EM sensor based on L-CyChBCZ for selectivity sensing potential of increasing and decreasing of Cu and Zn in mining-impacted water. The responses for similar concentrations of these two metals show high repeatability and the potential to differentiate between these two pollutants by analysing multiple peaks identified which are related to changes to metal concentration. Consequently, by combining the responses for multiple peaks and various peak parameters, the selectivity can be improved, although more data are necessary for developing mathematical functions for programmed discrimination and quantification of the toxic metals under test.
The f-EM sensors based on L-CyChBCZ were able to detect changes in Cu and Zn concentration, especially in frequencies < 2.5 GHz. The detection of Cu using microwaves was also demonstrated by Puangngernmak and Chalermwisutkul (2014) using an open-ended coaxial structure, to be able to distinguish only between low (1 mg/L) and very high concentrations (100 and 1000 mg/L) and not changes in small concentration as is demonstrated in this paper.
SEM and EDS analysis
These changes reflect the variation in physical properties and the consequent change in complex permittivity due to the specific interaction between the Cu ions and the functional materials on the planar sensor, especially l-cysteine and chitosan that chelate with Cu ions. Consequently, the sensing performance was improved by a modification of the spectral response and the increment of specificity (Bernou et al. 2000). Notably, the EM waves interact with each part of the sensor and the sensing response will depend on the permittivity change of each component, such as the sensor itself, the coating, the coating thickness, the chelated metal ions on the coating and the sample under test (Frau et al. 2019a). The f-EM sensors are not only detecting the metals which are chelated on the L-CyChBCZ substrate, and which generate a variation in the spectral response, but also the overall change in the sample under test. Consequently, if there is an increase or decrease in metal concentration, the sensor would rapidly detect this change, despite the metals which chelate with the coating material.
The f-EM sensors demonstrated in this work seem to be reusable and recoverable. After washing, the f-EM returns to its baseline spectrum (defined as air spectrum) in less than 2 min (Fig. 8c). However, the SEM used was not sensitive enough for quantitatively determining the Cu concentration in the coatings as a function of the concentration of the sample or characterising the effective interaction between the sample and the materials. The Zn was also part of the coating materials, so it was difficult to quantify. Further work will be required using more sensitive techniques, such as X-ray photoelectron spectroscopy (XPS) analysis.
Feasibility of differentiating mining-water samples
Metal concentrations and physicochemical parameters for the four samples collected in two mining areas in Wales (UK)
The calibration equation obtained in section Cu vs Zn was tested using real water samples. Results show that peaks 1 and 2 can differentiate the pollution level, such as which sample is more or less polluted, but more data integration is necessary for a more specific quantification and qualification of toxic metals in mining-impacted waters. Specifically, FA shows Cu concentration of 1.13 mg/L and 1.21 mg/L at peaks 1 and 2, respectively; MR shows 1.05 and 3.8 mg/L of Cu at peaks 1 and 2; NR shows 1.04 and 0.22 mg/L of Cu at the same peaks; and PM shows 34.0 and 10.2 mg/L. The concentration of Zn was quantified at peaks 1 and 2, respectively, at each sample as follows: for FA 1.2 and 1.4 mg/L; for MR 1.2 and 2.7 mg/L; for NC 1.2 and 0.6 mg/L; and for PM 27.9 and 7.8 mg/L. Although the measured concentrations were not specific, the sensors were able to qualify the pollution level, as is notable comparing results with Table 3. This is because the linear correlation for Cu, Zn and Cu + Zn at peaks 1 and 2 is comparable; thus, they can interfere with each other. Even if the microwave spectrum generated by F-EM sensors which interact Cu and Zn samples is different in distinct parts of the EM spectrum, it is not sufficient to remove completely the interference generated by similar metals, such as Cu and Zn. Consequently, integrating more parameters at different peaks (3–6) can give the opportunity to improve the selective detection. Moreover, some other interferences were identified, which could be due to other materials dissolved in the mining-impacted waters.
This novel approach needs further investigation to understand the causes of every interference caused by the water matrix, and more data could give an opportunity to develop f-EM sensors for specific toxic metal detection.
Additional work is required to evaluate and correlate changes with the water composition to further understand the effect of real water samples on the coating and its interaction with the Cu and Zn ions. Particularly, it will be necessary i) to evaluate the different microwave response for free and complexed metals, ii) to qualify and quantify the interference in the spectral response due to other inorganic and/or organic material in water, iii) to measure and correct changes caused by temperature variations. More experiments and data are also necessary for integrated peak features on the linear correlation between metal ions and the reflection coefficient.
Multianalysis will also be performed using a combination of f-EM sensors based on different coating materials. The durability and degradability of the coating integrated onto the sensor also require further research. Finally, a sensing platform made by printed microwave circuit boards set at specific frequency ranges can be manufactured and integrated into the national network for real-time metal pollution monitoring.
This study demonstrates the feasibility of using microwaves and planar sensors for the analysis of Cu and Zn concentrations in water. Planar sensors immersed in water were able to quantify changes of Cu concentration by the standard addition method. F-EM sensors, functionalised using thick films based on chelating polymers and metal oxides, were used for improving the specificity for Cu ions in mine water samples that were spiked continuously. The coated sensors show an improvement for Cu ion detection, especially in the frequency range 0.92–1.00 GHz with an improvement in sensitivity (1.631 ΔdB/mg/L), higher Q-factor (135.5) and low LOD (0.036 mg/L), demonstrating the feasibility to detect Cu concentration just above the environmental quality standards for freshwater (28–34 µg/L). A Lorentzian peak fitting was used for analysing the spectral response to identify the parameters of the peaks, particularly those that are superimposed upon one another, including area and FWHM of the peaks. Detecting small changes in the spectral response by analysing more parameters for each peak as described was shown to improve the selectivity between Cu and Zn. More work is necessary for qualifying and quantifying the interferences caused by complex water matrices and similar pollutants. This research approach could provide a reliable specific f-EM sensor array for efficient water quality monitoring.
The authors gratefully acknowledge the support of Liverpool John Moores University, the Faculty of Engineering and Technology in conjunction with the Faculty of Science PhD Scholarship Programme which allowed this research to be undertaken. Special thanks go to Prof David Phipps, Emeritus Professor, at LJMU, for his valuable suggestions, and to Patrizia Onnis, PhD researcher, at LJMU, for her support during sample collection.
Compliance with ethical standards
Conflict of interest
The authors declare no conflict of interest.
- Agency for Toxic Substances and Disease Registry (ATSDR) (2004) Toxicological profile for Copper. Department of Health and Human Services, Public Health Service, Atlanta, GAGoogle Scholar
- Byrne P, Hudson-Edwards KA, Bird G, Macklin MG, Brewer PA, Williams RD, Jamieson HE (2018) Water quality impacts and river system recovery following the 2014 Mount Polley mine tailings dam spill. British Columbia. Can Appl Geochem 91:64–74. https://doi.org/10.1016/j.apgeochem.2018.01.012 CrossRefGoogle Scholar
- Environment Agency (2008) Abandoned mines and the water environment vol Science project SC030136. Environment Agency, BristolGoogle Scholar
- Environment Agency (2012) Prioritisation of abandoned non-coal mine impacts on the environment vol SC030136/R2 The national picture. Environment Agency, BristolGoogle Scholar
- Frau I, Wylie S, Byrne P, Cullen J, Korostynska O, Mason A (2018b) Continuous detection of copper and bromide in polluted water using f-EM sensors. In: 2018 12th international conference on sensing technology (ICST), pp 35–38. https://doi.org/10.1109/icsenst.2018.8603592
- Frau I, Wylie S, Cullen J, Korostynska O, Byrne P, Mason A (2019b) Microwaves and functional materials: a novel method to continuously detect metal ions in water. In: Mukhopadhyay SC, Jayasundera KP, Postolache OA (eds) Modern sensing technologies. Springer, Cham, pp 179–201. https://doi.org/10.1016/j.mseb.2019.114382 CrossRefGoogle Scholar
- Graunke T, Schmitt K, Busch S, Raible S, Wöllenstein J (2018) Towards an empirical model for the prediction of the selectivity of polymer. Membr Proc 2:979Google Scholar
- Greene J, Abdullah B, Cullen J, Korostynska O, Louis J, Mason A (2019) Non-invasive monitoring of glycogen in real-time using an electromagnetic sensor. In: Mukhopadhyay SC, Jayasundera KP, Postolache OA (eds) Modern sensing technologies. Springer, Cham, pp 1–15. https://doi.org/10.1007/978-3-319-99540-3_1 CrossRefGoogle Scholar
- Harris DC (2007) Quantitative chemical analysis, 7th edn. W.H. Freeman, New YorkGoogle Scholar
- Iqbal J, Yiping DU, Howari F, Bataineh M, Muhammad N, Rahim A (2017) Simultaneous enrichment and on-line detection of low-concentration copper, cobalt, and nickel ions in water by near-infrared diffuse reflectance spectroscopy combined with chemometrics. J AOAC Int 100:560–565. https://doi.org/10.5740/jaoacint.16-0224 CrossRefGoogle Scholar
- Jilani MT, Rehman MZU, Khan AM, Khan MT, Ali SM (2012) A brief review of measuring techniques for characterization of dielectric materials. Int J Inf Technol Electr Eng 1:1Google Scholar
- Kölbl F, Boulboul N, Commereuc M, Bourdel E (2018) A microstrip resonator based sensor for GHz characterization of in vitro cell culture. In: 2018 12th International conference on sensing technology (ICST), pp 319–323. https://doi.org/10.1109/icsenst.2018.8603622
- Onnis P, Byrne P, Hudson-Edwards KA, Stott T, Hunt C (2018) Source apportionment of trace metals over a range of stream flows using a multi-method tracer approach. In: Wolkersdorfer C, Sartz L, Weber A, Burgess J, Tremblay G (eds) 11th ICARD | IMWA | MWD Conference—“Risk to Opportunity”, Pretoria, South Africa, Tshwane University of Technology, pp 843–849Google Scholar
- UK Technical Advisory Group on the Water Framework Directive (2008) UK environmental standards and conditions (phase 1–SR1-2006)Google Scholar
- United States Environmental Protection Agency (1986) Quality criteria for water, vol 440. EPA, Washington, DC, p 20460Google Scholar
- Wolkersdorfer C (2008) Water management at abandoned flooded underground mines—fundamentals, tracer tests, modelling, water treatment. Springer, HeidelbergGoogle Scholar
- Wu W et al (2019) Simultaneous voltammetric determination of cadmium(II), lead(II), mercury(II), zinc(II), and copper(II) using a glassy carbon electrode modified with magnetite (Fe3O4) nanoparticles and fluorinated multiwalled carbon nanotubes. Microchimica Acta 186:97. https://doi.org/10.1007/s00604-018-3216-5 CrossRefGoogle Scholar
- Younger PL, Banwart SA, Hedin RS (2002) Hydrology, pollution, remediation. Kluwer Academic Publishers, DordrechtGoogle Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.